
/* 
 * Argyll Color Management System
 * Display callibrator.
 *
 * Author: Graeme W. Gill
 * Date:   14/10/2005
 *
 * Copyright 1996 - 2013 Graeme W. Gill
 * All rights reserved.
 *
 * This material is licenced under the GNU AFFERO GENERAL PUBLIC LICENSE Version 3 :-
 * see the License.txt file for licencing details.
 */

/* This program displays test patches, and takes readings from a display device */
/* in order to create a RAMDAC calibration curve (usually stored in the ICC vcgt tag) */

/* This is the third version of the program. */


/* TTBD

	Seems to get weird if brightness target is out of range of
	display during interactive adjustment ???

	Would be good to automaticall invoke -k0 if this display looks
	like it has black at zero and has a proportional characteristic
	from zero. 
 
	Would be nice to have option of using existing display profile
	as source of measurements (including VCGT curves) in a similar
	way to -dfake, to allow re-generating a new calibration with different
	parameters.

	Should shift to using xicc code for BT.1886 and target response
	curve, for consistency with collink etc.

	Add support for black recalibration using i1pro or munki.
	Setable timeout ? Need to allow placing instrument back
	on screen. Need this to properly handle ss anyway ?

	Calibrating the black point of a true power response
	device is very slow to converge - the jacobian is always
	underestimating the actual delta RGB needed because the
	slope is getting shallower and shallower. Need to
	be able to figure when to increase rgain in those circumstances,
	rather than reducing it ? 

	Dealing with noisy/inconsistent readings could probably
	be improved - the statistical information from a iteration
	series is being ignored. ie. do a linear regression/fit
	on all the values for a given target, and then
	at the end, use a weighted blend of the best solution
	and the fit. Weight by something like the number used
	for the fit. vs. 1.

	Try to improve calibration speed by using adaptive
	measurement set, rather than fixed resolution doubling ?
	(ie. just measure at troublesome points using a "divide in half"
	 strategy ?. Estimate error between measurement points and
	pick the next largest error.) 

	Add option to use L*u*v* DE's, as this is used in
	some video standards. They sometime use u*v* as
	a color tollerance too (see EBU  TECH 3320).

	Add a white point option that makes the target the
	closest temperature to the native one of the display :-
	ie. it moves the display to the closest point on the
	chosen locus to RGB 1,1,1.
     ie. should it do this if "-t" or "-T"
	with no specific temperature is chosen ?

	Change white point gamut clipping to be a measurement
	search rather than computing from primary XYZ ?

	Handling of white and black device clipping is not so good.
	White clipping isn't characterized very well due to sparse sampling,
	and moncurve tends to smooth over the clip inflection point,
	making it innacurate. This particularly hurts the black point
	accuracy, leading to raised or crushed blacks.

	Add bell at end of calibration ?

	Add option to plot graph of native and calibrated RGB ?

	Add a "delta E" number to the interactive adjustments,
	so the significance of the error can be judged ?

	Need to add flare measure/subtract, to improve
	projector calibration ? - need to add to dispread too.

	Instead of measuring/matching output at 50% device input as
	measure of gamma, what about inverting it - measure/match device
    values at 50% perceptual (18%) output value ?
	[ Hmm. Current method is OK because a good perceptual
	  display gives about 18% output at 50% device input.]


	The verify (-z) may not be being done correctly.
	Like update, shouldn't it read the .cal file to set what's
	being calibrated aganist ? (This would fix missing ambient value too!)

	What about the "Read the base test set" - aren't
	there numbers then used to tweak the black aim point
	in "Figure out the black point target" - Yes they are !!
	Verify probably shouldn't work this way.

	Add DICOM Part 14 GSDF support:

		* Add absolute DICOM function target to dispcal.
		* Add 20% grey background full screen option + 10% patch recommendation
		* Add "include Glare" option for contact instruments to dispsup.c
		* Add DICOM mode black point hue handling (? what policy ?)
		* Add DICOM stats report (JND dE + mean + SD) to verify ??
 */

#ifdef __MINGW32__
# define WINVER 0x0500
#endif

#include <stdio.h>
#include <stdlib.h>
#include <math.h>
#include <sys/types.h>
#include <time.h>
#include <string.h>
#include <stdarg.h>
#if defined (NT)
#include <conio.h>
#endif
#include "copyright.h"
#include "aconfig.h"
#include "numlib.h"
#include "xicc.h"
#include "xspect.h"
#include "xcolorants.h"
#include "cgats.h"
#include "insttypes.h"
#include "conv.h"
#include "icoms.h"
#include "inst.h"
#include "ccmx.h"
#include "ccss.h"
#include "dispwin.h"
#include "ui.h"
#include "ccast.h"
#include "dispsup.h"
#include "rspl.h"
#include "moncurve.h"
#include "targen.h"
#include "ofps.h"
#include "icc.h"
#include "sort.h"
#include "instappsup.h"
#ifdef ENABLE_USB
# include "spyd2.h"
#endif

#undef DEBUG
#undef DEBUG_OFFSET			/* Keep test window out of the way */
#undef DEBUG_PLOT			/* Plot curve each time around */
#undef CHECK_MODEL			/* Do readings to check the accuracy of our model */
#undef SHOW_WINDOW_ONFAKE	/* Display a test window up for a fake device */

#undef DEBUG_MEAS_RES		/* Debug just VideoLUT resolution code */

/* Invoke with -dfake for testing with a fake device. */
/* Will use a fake.icm/.icc profile if present, or a built in fake */
/* device behaviour if not. */

#define COMPORT 1			/* Default com port 1..4 */
#define OPTIMIZE_MODEL		/* Adjust model for best fit */
#define REFINE_GAIN 0.90	/* Refinement correction damping/gain */
#define VER_RES 100			/* Verification resolution */
#define NEUTRAL_BLEND_RATE 4.0		/* Default rate of transition for -k factor < 1.0 (power) */
#define ADJ_JACOBIAN		/* Adjust the Jacobian predictor matrix each time */
#define JAC_COMP_FACT 0.4	/* Amount to compound Jacobian correction */
#define JAC_COR_FACT 0.4	/* Amount to damp Jacobian by (to filter noise) */
#define REMEAS_JACOBIAN		/* Re-measure Jacobian if it is a poor predictor */
#define MOD_DIST_POW 1.6	/* Power used to distribute test samples for model building */
#define REFN_DIST_POW 1.6	/* Power used to distribute test samples for grey axis refinement */
#define CHECK_DIST_POW 1.6	/* Power used to distribute test samples for grey axis checking */
#define THRESH_SCALE_POW 0.5 /* Amount to loosen threshold for first itterations */
#define ADJ_THRESH			/* Adjust threshold to be half a step on final pass */
#define MIN_THRESH 0.25	 	/* Minimum stopping threshold to allow in ADJ_THRESH */
#define POWERR_THR 0.05		/* Point near black to start weighting +ve error */
#define POWERR_WEIGHT 999.0	/* Weight to give +ve delta E at black */
#define POWERR_WEIGHT_POW 4.0	/* Curve to plend from equal weight to +ve extra weight */
#define CAL_RES 256			/* Resolution of calibration table to produce. */
#define CLIP				/* Clip RGB during refinement */
#define RDAC_SMOOTH 0.3		/* RAMDAC curve fitting smoothness */
#define MEAS_RES			/* Measure the RAMNDAC entry size */ 

#if defined(DEBUG_PLOT) || defined(DEBUG) || defined(DEBUG_MEAS_RES)
#include "plot.h"
#endif

#if defined(DEBUG)
# define DBG(xxx) fprintf xxx ;
# define DEBUG_MEAS_RES
#else
# define DBG(xxx) 
#endif	/* DEBUG */

#define dbgo stderr

/* - - - - - - - - - - - - - - - - - - - - - - - - - - - - */
/* Sample points used in initial device model optimisation */

typedef struct {
	double dev[3];		/* Device values */
	double lab[3];		/* Read value */
	double w;			/* Weighting */
} optref;

/* - - - - - - - - - - - - - - - - - - - */
/* device RGB inverse solution code */

/* Selected transfer curve */
typedef enum {
	gt_power     = 0,	/* A simple power */
	gt_Lab       = 1,	/* The L* curve */
	gt_sRGB      = 2,	/* The sRGB curve */
	gt_Rec709    = 3,	/* REC 709 video standard */
	gt_SMPTE240M = 4	/* SMTPE 240M video standard */
} gammatype;

/* Context for calibration solution */
typedef struct {
	double wh[3];		/* White absolute XYZ value */
	double bk[3];		/* Black absolute XYZ value */

	/* Target model */
	gammatype gammat;	/* Transfer curve type */
	double egamma;		/* Effective Gamma target */
	double oofff;		/* proportion of output offset vs input offset (default 1.0) */
	double gioff;		/* Gamma curve input zero offset */
	double gooff;		/* Target output offset (normalised to Y max of 1.0) */
	int nat;			/* Flag - nz if native white target */
	double nbrate;		/* Neutral blend weight (power) */
	int bkhack;			/* Flag - nz if black is hacked to be device zero */

	/* Viewing  conditions adjustment */
	int vc;				/* Flag, nz to enable viewing conditions adjustment */
	icxcam *svc;		/* Source viewing conditions */
	icxcam *dvc;		/* Destination viewing conditions */
	double vn0, vn1;	/* Normalisation values */

	double nwh[3];		/* Target white normalised XYZ value (Y = 1.0) */
	double twh[3];		/* Target white absolute XYZ value */
	double twYxy[3];	/* Target white Yxy (informational) */
	icmXYZNumber twN;	/* Same as above as XYZNumber */

	double tbk[3];		/* Target black point color */
	icmXYZNumber tbN;	/* Same as above as XYZNumber */

	/* Device model */
	double fm[3][3];	/* Forward, aprox. linear RGB -> XYZ */
	double bm[3][3];	/* Backwards, aprox. XYZ -> linear RGB */
	mcv *dcvs[3];		/* Device RGB channel to linearised RGB curves */
						/* These are always normalized to map 1.0 to 1.0 */

	/* Current state */
	mcv *rdac[3];		/* Current RGB to RGB ramdac curves */

	double xyz[3];		/* Target xyz value */

	/* optimisation information */
	int np;				/* Total number of optimisation parameters */
	int co[3];			/* Offset in the parameters to each curve offset */
	int nc[3];			/* Number of for each curve */
	int nrp;			/* Total number of reference points */
	optref *rp;			/* reference points */
	double *dtin_iv;	/* Temporary array :- dp for input curves */
} calx;

/* - - - - - - - - - - - - - - - - - - - */
/* Ideal target curve definitions */

/* Convert ideal device (0..1) to target Y value (0..1) */
static double dev2Y(calx *x, double egamma, double vv) {

	switch(x->gammat) {
		case gt_power: {
			vv = pow(vv, egamma);
			break;
		}
		case gt_Lab: {
			vv = icmL2Y(vv * 100.0);
			break;
		}
		case gt_sRGB: {
			if (vv <= 0.03928)
				vv = vv/12.92;
			else
				vv = pow((0.055 + vv)/1.055, 2.4);
			break;
		}
		case gt_Rec709: {
			if (vv <= 0.081)
				vv = vv/4.5;
			else
				vv = pow((0.099 + vv)/1.099, 1.0/0.45);
			break;
		}
		case gt_SMPTE240M: {
			if (vv <= 0.0913)
				vv = vv/4.0;
			else
				vv = pow((0.1115 + vv)/1.1115, 1.0/0.45);
			break;
		} 
		default:
			error("Unknown gamma type");
	}
	return vv;
}

/* Convert target Y value (0..1) to ideal device (0..1) */
static double Y2dev(calx *x, double egamma, double vv) {

	switch(x->gammat) {
		case gt_power: {
			vv = pow(vv, 1.0/egamma);
			break;
		}
		case gt_Lab: {
			vv = icmY2L(vv) * 0.01;
			break;
		}
		case gt_sRGB: {
			if (vv <= 0.00304)
				vv = vv * 12.92;
			else
				vv = pow(vv, 1.0/2.4) * 1.055 - 0.055;
			break;
		}
		case gt_Rec709: {
			if (vv <= 0.018)
				vv = vv * 4.5;
			else
				vv = pow(vv, 0.45) * 1.099 - 0.099;
			break;
		}
		case gt_SMPTE240M: {
			if (vv <= 0.0228)
				vv = vv * 4.0;
			else
				vv = pow(vv, 0.45) * 1.1115 - 0.1115;
			break;
		} 
		default:
			error("Unknown gamma type");
	}
	return vv;
}

/* - - - - - - - - - - - - - - - - - - - */
/* Compute a viewing environment Y transform */

static double view_xform(calx *x, double in) {
	double out = in;

	if (x->vc != 0) {
		double xyz[3], Jab[3];

		xyz[0] = in * x->nwh[0];				/* Compute value on neutral axis */
		xyz[1] = in * x->nwh[1];
		xyz[2] = in * x->nwh[2];
		x->svc->XYZ_to_cam(x->svc, Jab, xyz);
		x->dvc->cam_to_XYZ(x->dvc, xyz, Jab);

		out = xyz[1] * x->vn1 + x->vn0;			/* Apply scaling factors */
	}
	return out;
}

/* - - - - - - - - - - - - - - - - - - - */

/* Info for optimization */
typedef struct {
	double thyr;		/* 50% input target */
	double roo;			/* 0% input target */
} gam_fits;

/* gamma + input offset function handed to powell() */
static double gam_fit(void *dd, double *v) {
	gam_fits *gf = (gam_fits *)dd;
	double gamma = v[0];
	double ioff = v[1];
	double rv = 0.0;
	double tt;

	if (gamma < 0.0) {
		rv += 100.0 * -gamma;
		gamma = 0.0;
	}
	if (ioff < 0.0) {
		rv += 100.0 * -ioff;
		ioff = 0.0;
	} else if (ioff > 0.999) {
		rv += 100.0 * (ioff - 0.999);
		ioff = 0.999;
	}
	tt = gf->roo - pow(ioff, gamma);
	rv += tt * tt;
	tt = gf->thyr - pow(0.5 + (1.0 - 0.5) * ioff, gamma);
	rv += tt * tt;
	
//printf("~1 gam_fit %f %f returning %f\n",ioff,gamma,rv);
	return rv;
}


/* Given the advertised gamma and the output offset, compute the */
/* effective gamma and input offset needed. */
/* Return the expected output value for 50% input. */
/* (It's assumed that gooff is normalised the target brightness) */
static double tech_gamma(
	calx *x,
	double *pegamma,		/* return effective gamma needed */
	double *pooff,			/* return output offset needed */
	double *pioff,			/* return input offset needed */
	double egamma,			/* effective gamma needed (> 0.0 if valid, overrides gamma) */
	double gamma,			/* advertised gamma needed */
	double tooff			/* Total output offset needed */
) {
	int i;
	double rv;
	double gooff = 0.0;		/* The output offset applied */
	double gioff = 0.0;		/* The input offset applied */
	double roo;				/* Remaining output offset accounted for by input offset */

	/* Compute the output offset that will be applied */
	gooff = tooff * x->oofff;
	roo = (tooff - gooff)/(1.0 - gooff);

//printf("~1 gooff = %f, roo = %f\n",gooff,roo);

	/* Now compute the input offset that will be needed */
	if (x->gammat == gt_power && egamma <= 0.0) {
		gam_fits gf;
		double op[2], sa[2], rv;

		gf.thyr = pow(0.5, gamma);					/* Advetised 50% target */
		gf.thyr = (gf.thyr - gooff)/(1.0 - gooff);	/* Target before gooff is added */
		gf.roo = roo;

		op[0] = gamma;
		op[1] = pow(roo, 1.0/gamma);
		sa[0] = 0.1;
		sa[1] = 0.01;

		if (powell(&rv, 2, op, sa, 1e-6, 500, gam_fit, (void *)&gf, NULL, NULL) != 0)
			warning("Computing effective gamma and input offset is inaccurate");

		if (rv > 1e-5) {
			warning("Computing effective gamma and input offset is inaccurate (%f)",rv);
		}
		egamma = op[0];
		gioff = op[1];  

//printf("~1 Result gioff %f, gooff %f, egamma %f\n",gioff, gooff, egamma);
//printf("~1 Verify 0.0 in -> out = %f, tooff = %f\n",gooff + dev2Y(x, egamma, gioff) * (1.0 - gooff),tooff);
//printf("~1 Verify 0.5 out = %f, target %f\n",gooff + dev2Y(x, egamma, gioff + 0.5 * (1.0 - gioff)) * (1.0 - gooff), pow(0.5, gamma));

	} else {
		gioff = Y2dev(x, egamma, roo);
//printf("~1 Result gioff %f, gooff %f\n",gioff, gooff);
//printf("~1 Verify 0.0 in -> out = %f, tooff = %f\n",gooff + dev2Y(x, egamma, gioff) * (1.0 - gooff),tooff);
	}

	/* Compute the 50% output value */
	rv = gooff + dev2Y(x, egamma, gioff + 0.5 * (1.0 - gioff)) * (1.0 - gooff);

	if (pegamma != NULL)
		*pegamma = egamma;
	if (pooff != NULL)
		*pooff = gooff;
	if (pioff != NULL)
		*pioff = gioff;
	return rv;
}

/* Compute approximate advertised gamma from black/50% grey/white readings, */
/* (assumes a zero based gamma curve shape) */
static double pop_gamma(double bY, double gY, double wY) {
	int i;
	double grat, brat, gioff, gvv, gamma;

	grat = gY/wY;
	brat = bY/wY;
	
	gamma = log(grat) / log(0.5);
	return gamma;
}

/* - - - - - - - - - - - - - - - - - - - */

/* Return the xyz that is predicted by our aproximate device model */
/* by the given device RGB. */
static void fwddev(calx *x, double xyz[3], double rgb[3]) {
	double lrgb[3];
	int j;

//printf("~1 fwddev called with rgb %f %f %f\n",rgb[0],rgb[1],rgb[2]);

	/* Convert device RGB into linear light RGB via curves */
	for (j = 0; j < 3; j++)
		lrgb[j] = x->dcvs[j]->interp(x->dcvs[j], rgb[j]);

//printf("~1 fwddev got linear RGB %f %f %f\n",lrgb[0],lrgb[1],lrgb[2]);

	/* Convert linear light RGB into XYZ via the matrix */
	icmMulBy3x3(xyz, x->fm, lrgb);

//printf("~1 fwddev got final xyz %f %f %f\n",xyz[0],xyz[1],xyz[2]);
}

/* Return the closest device RGB predicted by our aprox. device model */
/* to generate the given xyz. */
static void invdev(calx *x, double rgb[3], double xyz[3]) {
	double lrgb[3];
	int j;

//printf("~1 invdev called with xyz %f %f %f\n",xyz[0],xyz[1],xyz[2]);

	/* Convert XYZ to linear light RGB via the inverse matrix */
	icmMulBy3x3(lrgb, x->bm, xyz);
//printf("~1 invdev; lin light rgb = %f %f %f\n",lrgb[0],lrgb[1],lrgb[2]);

	/* Convert linear light RGB to device RGB via inverse curves */
	for (j = 0; j < 3; j++) {
		lrgb[j] = x->dcvs[j]->inv_interp(x->dcvs[j], lrgb[j]);
		if (lrgb[j] < 0.0) {
#ifdef CLIP
			lrgb[j] = 0.0;
#endif
		} else if (lrgb[j] > 1.0) {
#ifdef CLIP
			lrgb[j] = 1.0;
#endif
		}
	}
//printf("~1 invdev; inverse curves rgb = %f %f %f\n",lrgb[0],lrgb[1],lrgb[2]);
	if (rgb != NULL) {
		rgb[0] = lrgb[0];
		rgb[1] = lrgb[1];
		rgb[2] = lrgb[2];
	}
}

/* Return the closest linear device RGB predicted by our aprox. device matrix */
/* to generate the given xyz. */
/* Return > 0 if clipped */
static double invlindev(calx *x, double rgb[3], double xyz[3]) {
	double lrgb[3];
	double clip = 0.0;
	int j;

//printf("~1 invlindev called with xyz %f %f %f\n",xyz[0],xyz[1],xyz[2]);

	/* Convert XYZ to linear light RGB via the inverse matrix */
	icmMulBy3x3(lrgb, x->bm, xyz);
//printf("~1 invlindev; lin light rgb = %f %f %f\n",lrgb[0],lrgb[1],lrgb[2]);

	/* Check for out of gamut */
	for (j = 0; j < 3; j++) {
		if (lrgb[j] < 0.0) {
			if (-lrgb[j] > clip)
				clip = -lrgb[j];
			lrgb[j] = 0.0;
		} else if (lrgb[j] > 1.0) {
			if ((lrgb[j]-1.0) > clip)
				clip = (lrgb[j]-1.0);
			lrgb[j] = 1.0;
		}
	}
//printf("~1 invlindev; clipped rgb = %f %f %f, clip = %f \n",lrgb[0],lrgb[1],lrgb[2],clip);
	if (rgb != NULL) {
		rgb[0] = lrgb[0];
		rgb[1] = lrgb[1];
		rgb[2] = lrgb[2];
	}
	return clip;
}

/* Overall optimisation support */

/* Set the optimsation parameter number and offset values in calx, */
/* and return an array filled in with the current parameters. */
/* Allocate temporary arrays */
static double *dev_get_params(calx *x) {
	double *p, *tp;
	int i, j;

	x->np = 9;
	for (i = 0; i < 3; i++)
		x->np += x->dcvs[i]->luord;

	if ((p = (double *)malloc(x->np * sizeof(double))) == NULL)
		error("dev_params malloc failed");
	
	tp = p;
	
	for (i = 0; i < 3; i++)
		for (j = 0; j < 3; j++)
			*tp++ = x->fm[i][j];

	for (i = 0; i < 3; i++) {
		x->co[i] = tp - p;				/* Offset to start */
		for (j = 0; j < x->dcvs[i]->luord; j++)
			*tp++ = x->dcvs[i]->pms[j];
		x->nc[i] = (tp - p) - x->co[i];	/* Number */
	}
		
	if ((x->dtin_iv = (double *)malloc(x->np * sizeof(double))) == NULL)
		error("dev_params malloc failed");

	return p;
}

/* Given a set of parameters, put them back into the model */
/* Normalize them so that the curve maximum is 1.0 too. */
static void dev_put_params(calx *x, double *p) {
	int i, j;
	double scale[3];

	for (i = 0; i < 3; i++)
		for (j = 0; j < 3; j++)
			x->fm[i][j] = *p++;
	for (i = 0; i < 3; i++)
		for (j = 0; j < x->dcvs[i]->luord; j++)
			x->dcvs[i]->pms[j] = *p++;

	/* Figure out how we have to scale the curves */
	for (j = 0; j < 3; j++) {
		scale[j] = x->dcvs[j]->interp(x->dcvs[j], 1.0);
		x->dcvs[j]->force_scale(x->dcvs[j], 1.0);
	}

	/* Scale the matrix to compensate */
	for (i = 0; i < 3; i++)
		for (j = 0; j < 3; j++)
			x->fm[i][j] *= scale[j];
}

/* Device model optimisation function handed to powell() */
static double dev_opt_func(void *edata, double *v) {
	calx *x = (calx *)edata;
	int i, j;
	double tw = 0.0;
	double rv, smv;

#ifdef NEVER
	printf("params =");
	for (i = 0; i < x->np; i++)
		printf(" %f",v[i]);
	printf("\n");
#endif

	/* For all our data points */
	rv = 0.0;
	for (i = 0; i < x->nrp; i++) {
		double lrgb[3];		/* Linear light RGB */
		double xyz[3], lab[3];
		double de;

		/* Convert through device curves */
		for (j = 0; j < 3; j++)
			lrgb[j] = x->dcvs[j]->interp_p(x->dcvs[j], v + x->co[j], x->rp[i].dev[j]);

		/* Convert linear light RGB into XYZ via the matrix */
		icxMulBy3x3Parm(xyz, v, lrgb);

		/* Convert to Lab */
		icmXYZ2Lab(&x->twN, lab, xyz);
	
		/* Compute delta E squared */
		de = icmCIE94sq(lab, x->rp[i].lab) * x->rp[i].w;
#ifdef NEVER
	printf("point %d DE %f, Lab is %f %f %f, should be %f %f %f\n",
i, sqrt(de), lab[0], lab[1], lab[2], x->rp[i].lab[0], x->rp[i].lab[1], x->rp[i].lab[2]);
#endif
		rv += de;
		tw += x->rp[i].w;
	}

	/* Normalise error to be a weighted average delta E squared and scale smoothing */
	rv /= (tw * 5.0);

	/* Sum with shaper parameters squared, to */
	/* minimise unsconstrained "wiggles" */
	smv = 0.0;
	for (j = 0; j < 3; j++)
		smv += x->dcvs[j]->shweight_p(x->dcvs[j], v + x->co[j], 1.0);
	rv += smv;

#ifdef NEVER
	printf("rv = %f (%f)\n",rv, smv);
#endif
	return rv;
}


/* Device model optimisation function handed to conjgrad() */
static double dev_dopt_func(void *edata, double *dv, double *v) {
	calx *x = (calx *)edata;
	int i, j, k;
	int f, ee, ff, jj;
	double tw = 0.0;
	double rv, smv;

	double dmato_mv[3][9];		/* Del in mat out due to del in matrix param vals */
	double dmato_tin[3][3];		/* Del in mat out due to del in matrix input values */
	double dout_lab[3][3];		/* Del in out due to XYZ to Lab conversion */
	double de_dout[2][3];		/* Del in delta E due to input Lab values */

#ifdef NEVER
	printf("params =");
	for (i = 0; i < x->np; i++)
		printf(" %f",v[i]);
	printf("\n");
#endif

	/* Zero the accumulated partial derivatives */
	for (i = 0; i < x->np; i++)
		dv[i] = 0.0;

	/* For all our data points */
	rv = 0.0;
	for (i = 0; i < x->nrp; i++) {
		double lrgb[3];		/* Linear light RGB */
		double xyz[3], lab[3];

		/* Apply the input channel curves */
		for (j = 0; j < 3; j++)
			lrgb[j] = x->dcvs[j]->dinterp_p(x->dcvs[j], v + x->co[j],
			                         x->dtin_iv + x->co[j], x->rp[i].dev[j]);

		/* Convert linear light RGB into XYZ via the matrix */
		icxdpdiMulBy3x3Parm(xyz, dmato_mv, dmato_tin, v, lrgb);
		
		/* Convert to Lab */
		icxdXYZ2Lab(&x->twN, lab, dout_lab, xyz);
	
		/* Compute delta E squared */
//printf("~1 point %d: Lab is %f %f %f, should be %f %f %f\n",
//i, lab[0], lab[1], lab[2], x->rp[i].lab[0], x->rp[i].lab[1], x->rp[i].lab[2]);
		rv += icxdCIE94sq(de_dout, lab, x->rp[i].lab) * x->rp[i].w;
		de_dout[0][0] *= x->rp[i].w;
		de_dout[0][1] *= x->rp[i].w;
		de_dout[0][2] *= x->rp[i].w;
		tw += x->rp[i].w;

		/* Compute and accumulate partial difference values for each parameter value */

		/* Input channel curves */
		for (ee = 0; ee < 3; ee++) {				/* Parameter input chanel */
			for (k = 0; k < x->nc[ee]; k++) {		/* Param within channel */
				double vv = 0.0;
				jj = x->co[ee] + k;					/* Overall input curve param */

				for (ff = 0; ff < 3; ff++) {		/* Lab channels */
					for (f = 0; f < 3; f++) {		/* XYZ channels */
						vv += de_dout[0][ff] * dout_lab[ff][f]
						    * dmato_tin[f][ee] * x->dtin_iv[jj];
					}
				}
				dv[jj] += vv;
			}
		}

		/* Matrix parameters */
		for (k = 0; k < 9; k++) {				/* Matrix parameter */
			double vv = 0.0;

			for (ff = 0; ff < 3; ff++) {		/* Lab channels */
				for (f = 0; f < 3; f++) {		/* XYZ channels */
					vv += de_dout[0][ff] * dout_lab[ff][f]
					    * dmato_mv[f][k];
				}
			}
			dv[k] += vv;
		}
	}

	/* Normalise error to be a weighted average delta E squared and scale smoothing */
	rv /= (tw * 1200.0);
	for (i = 0; i < x->np; i++)
		dv[i] /= (tw * 900.0);

	/* Sum with shaper parameters squared, to */
	/* minimise unsconstrained "wiggles" */
	smv = 0.0;
	for (j = 0; j < 3; j++)
		smv += x->dcvs[j]->dshweight_p(x->dcvs[j], v + x->co[j], x->dtin_iv + x->co[j], 1.0);
	rv += smv;

#ifdef NEVER
	printf("drv = %f (%f)\n",rv, smv);
#endif
	return rv;
}

#ifdef NEVER
/* Check partial derivative function within dev_opt_func() using powell() */

static double dev_opt_func(void *edata, double *v) {
	calx *x = (calx *)edata;
	int i;
	double dv[2000];
	double rv, drv;
	double trv;
	
	rv = dev_opt_func_(edata, v);
	drv = dev_dopt_func(edata, dv, v);

	if (fabs(rv - drv) > 1e-6) {
		printf("######## RV MISMATCH is %f should be %f ########\n",drv, rv);
		exit(0);
	}

	/* Check each parameter delta */
	for (i = 0; i < x->np; i++) {
		double del;

		v[i] += 1e-7;
		trv = dev_opt_func_(edata, v);
		v[i] -= 1e-7;
		
		/* Check that del is correct */
		del = (trv - rv)/1e-7;
		if (fabs(dv[i] - del) > 1.0) {
//printf("~1 del = %f from (trv %f - rv %f)/0.1\n",del,trv,rv);
			printf("######## EXCESSIVE at v[%d] is %f should be %f ########\n",i,dv[i],del);
			exit(0);
		}
	}
	return rv;
}
#endif

/* =================================================================== */

/* White point brightness optimization function handed to powell. */
/* Maximize brigtness while staying within gamut */
static double wp_opt_func(void *edata, double *v) {
	calx *x = (calx *)edata;
	double wxyz[3], rgb[3];
	int j;
	double rv = 0.0;

	wxyz[0] = v[0] * x->twh[0];
	wxyz[1] = v[0] * x->twh[1];
	wxyz[2] = v[0] * x->twh[2];

//printf("~1 wp_opt_func got scale %f, xyz = %f %f %f\n",
//v[0],wxyz[0],wxyz[1],wxyz[2]);

	if ((rv = invlindev(x, rgb, wxyz)) > 0.0) {		/* Out of gamut */
		rv *= 1e5;
//printf("~1 out of gamut %f %f %f returning %f\n", rgb[0], rgb[1], rgb[2], rv);
		return rv;
	}
	/* Maximize scale factor */
	if (v[0] < 0.00001)
		rv = 1.0/0.00001;
	else
		rv = 1.0/v[0];
	
//printf("~1 %f %f %f returning %f\n", rgb[0], rgb[1], rgb[2], rv);
	return rv;
}

/* =================================================================== */
/* Structure to save aproximate model readings in */
typedef struct {
	double v;			/* Input value */
	double xyz[3];		/* Reading */
} sxyz;

/* ------------------------------------------------------------------- */
#if defined(__APPLE__) && defined(__POWERPC__)

/* Workaround for a ppc gcc 3.3 optimiser bug... */
/* It seems to cause a segmentation fault instead of */
/* converting an integer loop index into a float, */
/* when there are sufficient variables in play. */
static int gcc_bug_fix(int i) {
	static int nn;
	nn += i;
	return nn;
}
#endif	/* APPLE */


/* =================================================================== */
/* Calibration sample point support. This allows the successive */
/* refinement of our neutral sample points */

/* A sample point */
typedef struct {
	double v;			/* Desired input value */
	double rgb[3];		/* Input value through calibration curves */
	double tXYZ[3];		/* Target XYZ */
	double XYZ[3];		/* Read XYZ */
	double deXYZ[3];	/* Delta XYZ wanted to target */
	double _de;			/* Non-weighted Delta Lab */
	double de;			/* Weightd Delta Lab to neutral target */
	double dc;			/* Weightd Delta XYZ to neutral target */
	double peqde;		/* Weightd Delta Lab to last pass equivalent point value */
	double hde;			/* Weightd Hybrid de composed of de and peqde */

	double prgb[3];		/* Previous measured RGB */
	double pXYZ[3];		/* Previous measured XYZ */
	double pdXYZ[3];	/* Delta XYZ intended from previous measure */
	double pdrgb[3];	/* Delta rgb made to previous to acorrect XYZ */

	double dXYZ[3];		/* Actual delta XYZ resulting from previous delta rgb */

	double j[3][3];		/* Aproximate Jacobian (del RGB -> XYZ) */
	double ij[3][3];	/* Aproximate inverse Jacobian (del XYZ-> del RGB) */
	double fb_ij[3][3];	/* Copy of initial inverse Jacobian, used as a fallback */
} csp;


/* All the sample points */
typedef struct {
	int no;				/* Number of samples */
	int _no;			/* Allocation */
	csp *s;				/* List of samples */
} csamp;

static void free_alloc_csamp(csamp *p) {
	if (p->s != NULL)
		free(p->s);
	p->s = NULL;
}

/* Initialise v values */
static void init_csamp_v(csamp *p, calx *x, int psrand) {
	int i, j;
	sobol *so = NULL;

	if (psrand != 0) {	/* Use pseudo random distribution for verification */
		if ((so = new_sobol(1)) == NULL)
			error("New sobol failed");
	}
	
	/* Generate the sample points */
	for (i = 0; i < p->no; i++) {
		double vv;

#if defined(__APPLE__) && defined(__POWERPC__)
		gcc_bug_fix(i);
#endif
		if (so != NULL) {
			if (i == 0)
				vv = 1.0;
			else if (i == 1)
				vv = 0.0;
			else
				so->next(so, &vv);
		} else
			vv = i/(p->no - 1.0);
		vv = pow(vv, REFN_DIST_POW);	/* Skew sample points to be slightly perceptual */
		p->s[i].v = vv;
	}

	if (so != NULL) {
		/* Sort it so white is last */
#define HEAP_COMPARE(A,B) (A.v < B.v) 
	HEAPSORT(csp,p->s,p->no)
#undef HEAP_COMPARE
		so->del(so);
	}
}

/* Initialise txyz values from v values */
static void init_csamp_txyz(csamp *p, calx *x, int fixdev, int verb) {
	int i, j;
	double tbL[3];		/* tbk as Lab */

	if (verb >= 3)
		printf("init_csamp_txyz:\n");

	/* Convert target black from XYZ to Lab here, */
	/* in case twN has changed at some point. */
	icmXYZ2Lab(&x->twN, tbL, x->tbk);

	/* Set the sample points targets */
	for (i = 0; i < p->no; i++) {
		double y, vv;
		double XYZ[3];			/* Existing XYZ value */
		double Lab[3];
		double bl;

		vv = p->s[i].v;

		/* Compute target relative Y value for this device input. */
		/* We allow for any input and/or output offset */
		y = x->gooff + dev2Y(x, x->egamma, x->gioff + vv * (1.0 - x->gioff)) * (1.0 - x->gooff);

		/* Add viewing environment transform */
		y = view_xform(x, y);

		/* Convert Y to L* */
		Lab[0] = icmY2L(y);
		Lab[1] = Lab[2] = 0.0;	/* Target is neutral */

		/* Compute blended neutral target a* b* */
		bl = pow((1.0 - vv), x->nbrate);		/* Crossover near the black */
		Lab[1] = (1.0 - bl) * 0.0 + bl * tbL[1];
		Lab[2] = (1.0 - bl) * 0.0 + bl * tbL[2];

		icmAry2Ary(XYZ, p->s[i].tXYZ);				/* Save the existing values */
		icmLab2XYZ(&x->twN, p->s[i].tXYZ, Lab);		/* New XYZ Value to aim for */

		if (verb >= 3) {
			printf("%d: target XYZ %.4f %.4f %.4f, Lab %.3f %.3f %.3f\n",i, p->s[i].tXYZ[0],p->s[i].tXYZ[1],p->s[i].tXYZ[2], Lab[0],Lab[1],Lab[2]);
		}
	}
}


/* Allocate the sample points and initialise them with the */
/* target device and XYZ values, and first cut device values. */
static void init_csamp(csamp *p, calx *x, int doupdate, int verify, int psrand, int no, int verb) {
	int i, j;
	
	p->_no = p->no = no;

	if ((p->s = (csp *)calloc(p->_no, sizeof(csp))) == NULL)
		error("csamp malloc failed");

	/* Compute v and txyz */
	init_csamp_v(p, x, psrand);
	init_csamp_txyz(p, x, 0, verb);

	/* Generate the sample points */
	for (i = 0; i < no; i++) {
		double dd, vv;

#if defined(__APPLE__) && defined(__POWERPC__)
		gcc_bug_fix(i);
#endif
		vv = p->s[i].v;

		if (verify == 2) {		/* Verifying through installed curve */
			/* Make RGB values the input value */
			p->s[i].rgb[0] = p->s[i].rgb[1] = p->s[i].rgb[2] = vv;

		} else if (doupdate) {	/* Start or verify through current cal curves */
			for (j = 0; j < 3; j++) {
				p->s[i].rgb[j] = x->rdac[j]->interp(x->rdac[j], vv);
#ifdef CLIP
				if (p->s[i].rgb[j] < 0.0)
					p->s[i].rgb[j] = 0.0;
				else if (p->s[i].rgb[j] > 1.0)
					p->s[i].rgb[j] = 1.0;
#endif
			}
		} else {		/* we have model */
			/* Lookup an initial device RGB for that target by inverting */
			/* the approximate forward device model */
			p->s[i].rgb[0] = p->s[i].rgb[1] = p->s[i].rgb[2] = vv;
			invdev(x, p->s[i].rgb, p->s[i].tXYZ);
		}
		/* Force white to be native if native flag set */
		if (x->nat && i == (no-1)) {
//printf("~1 Forcing white rgb to be 1,1,1\n");
			p->s[i].rgb[0] = p->s[i].rgb[1] = p->s[i].rgb[2] = 1.0;
		}
		
//printf("~1 Inital point %d rgb %f %f %f\n",i,p->s[i].rgb[0],p->s[i].rgb[1],p->s[i].rgb[2]);

		/* Compute the approximate inverse Jacobian at this point */
		/* by taking the partial derivatives wrt to each device */
		/* channel of our aproximate forward model */
		if (verify != 2) {
			double refXYZ[3], delXYZ[3];
			fwddev(x, refXYZ, p->s[i].rgb);
			if (vv < 0.5)
				dd = 0.02;
			else
				dd = -0.02;
			/* Matrix organization is J[XYZ][RGB] for del RGB->del XYZ*/
			for (j = 0; j < 3; j++) {
				p->s[i].rgb[j] += dd;
				fwddev(x, delXYZ, p->s[i].rgb);
				p->s[i].j[0][j] = (delXYZ[0] - refXYZ[0]) / dd;
				p->s[i].j[1][j] = (delXYZ[1] - refXYZ[1]) / dd;
				p->s[i].j[2][j] = (delXYZ[2] - refXYZ[2]) / dd;
				p->s[i].rgb[j] -= dd;
			}
			if (icmInverse3x3(p->s[i].ij, p->s[i].j)) {
				error("dispcal: inverting Jacobian failed (1)");
			}
			/* Make a copy of this Jacobian in case we get an invert failure later */
			icmCpy3x3(p->s[i].fb_ij, p->s[i].ij);
		}
	}
}

/* Return a linear XYZ interpolation */
static void csamp_interp(csamp *p, double xyz[3], double v) {
	int i, j;
	double b;

	if (p->no < 2)
		error("Calling csamp_interp with less than two existing samples");

	/* Locate the pair surrounding our input value */
	for (i = 0; i < (p->no-1); i++) {
		if (v >= p->s[i].v && v <= p->s[i+1].v)
			break;
	}
	if (i >= (p->no-1))
		error("csamp_interp out of range");
	
	b = (v - p->s[i].v)/(p->s[i+1].v - p->s[i].v);

	for (j = 0; j < 3; j++) {
		xyz[j] = b * p->s[i+1].XYZ[j] + (1.0 - b) * p->s[i].XYZ[j];
	}
}

/* Re-initialise a CSP with a new number of points. */
/* Interpolate the device values and jacobian. */
/* Set the current rgb from the current RAMDAC curves if not verifying */
static void reinit_csamp(csamp *p, calx *x, int verify, int psrand, int no, int verb) {
	csp *os;			/* Old list of samples */
	int ono;			/* Old number of samples */
	int i, j, k, m;
	
	if (no == p->no)
		return;			/* Nothing has changed */

	os = p->s;			/* Save the existing per point information */
	ono = p->no;

	init_csamp(p, x, 0, 2, psrand, no, verb);

	p->_no = p->no = no;

	/* Interpolate the current device values */
	for (i = 0; i < no; i++) {
		double vv, b;

		vv = p->s[i].v;

		/* Locate the pair surrounding our target value */
		for (j = 0; j < ono-1; j++) {
			if (vv >= os[j].v && vv <= os[j+1].v)
				break;
		}
		if (j >= (ono-1))
			error("csamp interp. out of range");
		
		b = (vv - os[j].v)/(os[j+1].v - os[j].v);
	
		for (k = 0; k < 3; k++) {
			if (verify == 2) {

				p->s[i].rgb[k] = b * os[j+1].rgb[k] + (1.0 - b) * os[j].rgb[k];

			} else {	/* Lookup rgb from current calibration curves */
				for (m = 0; m < 3; m++) {
					p->s[i].rgb[m] = x->rdac[m]->interp(x->rdac[m], vv);
#ifdef CLIP
					if (p->s[i].rgb[m] < 0.0)
						p->s[i].rgb[m] = 0.0;
					else if (p->s[i].rgb[m] > 1.0)
						p->s[i].rgb[m] = 1.0;
#endif
				}
			}
			p->s[i].XYZ[k] = b * os[j+1].XYZ[k] + (1.0 - b) * os[j].XYZ[k];
			p->s[i].deXYZ[k] = b * os[j+1].deXYZ[k] + (1.0 - b) * os[j].deXYZ[k];
			p->s[i].pXYZ[k] = b * os[j+1].pXYZ[k] + (1.0 - b) * os[j].pXYZ[k];
			p->s[i].pdrgb[k] = b * os[j+1].pdrgb[k] + (1.0 - b) * os[j].pdrgb[k];
			p->s[i].dXYZ[k] = b * os[j+1].dXYZ[k] + (1.0 - b) * os[j].dXYZ[k];
#ifdef INTERP_JAC
			for (m = 0; m < 3; m++) 
				p->s[i].j[k][m] = b * os[j+1].j[k][m] + (1.0 - b) * os[j].j[k][m];
#endif

		}
#ifndef INTERP_JAC
		/* Create a Jacobian at this location from our forward model */
		{
			double dd, refXYZ[3], delXYZ[3];
			fwddev(x, refXYZ, p->s[i].rgb);
			if (vv < 0.5)
				dd = 0.02;
			else
				dd = -0.02;
			/* Matrix organization is J[XYZ][RGB] for del RGB->del XYZ*/
			for (j = 0; j < 3; j++) {
				p->s[i].rgb[j] += dd;
				fwddev(x, delXYZ, p->s[i].rgb);
				p->s[i].j[0][j] = (delXYZ[0] - refXYZ[0]) / dd;
				p->s[i].j[1][j] = (delXYZ[1] - refXYZ[1]) / dd;
				p->s[i].j[2][j] = (delXYZ[2] - refXYZ[2]) / dd;
				p->s[i].rgb[j] -= dd;
			}
		}
#endif
		if (icmInverse3x3(p->s[i].ij, p->s[i].j)) {
			error("dispcal: inverting Jacobian failed (2)");
		}
		/* Make a copy of this Jacobian in case we get an invert failure later */
		icmCpy3x3(p->s[i].fb_ij, p->s[i].ij);

		/* Compute expected delta XYZ using new Jacobian */
		icmMulBy3x3(p->s[i].pdXYZ, p->s[i].j, p->s[i].pdrgb);

		p->s[i]._de = p->s[i].de = b * os[j+1].de + (1.0 - b) * os[j].de;
		p->s[i].dc = b * os[j+1].dc + (1.0 - b) * os[j].dc;
	}

	free(os);
}

/* - - - - - - - - - - - - - - - - - - - - - - - - - - - - - */

#ifdef NEVER
/* Do a linear interp of the ramdac */
static void interp_ramdac(double cal[CAL_RES][3], double drgb[3], double rgb[3]) {
	int i, j;
	int gres = CAL_RES;
	double w;

	/* For r,g & b */
	for (j = 0; j < 3; j++) {
		int mi, gres_1 = gres-1;
		double t, vv = rgb[j];
		t = gres * vv;
		mi = (int)floor(t);			/* Grid coordinate */
		if (mi < 0)					/* Limit to valid cube base index range */
			mi = 0;
		else if (mi >= gres_1)
			mi = gres_1-1;
		w = t - (double)mi;	 		/* 1.0 - weight */

		drgb[j] = (1.0 - w) * cal[mi][j] + w * cal[mi+1][j];
	}
}
#endif	/* NEVER */

/* Given an XYZ, compute the color temperature and the delta E 2K to the locus */
static double comp_ct(
	double *de,		/* If non-NULL, return CIEDE2000 to locus */
	double lxyz[3],	/* If non-NULL, return normalised XYZ on locus */
	int plank,		/* NZ if Plankian locus, 0 if Daylight locus */
	int dovct,		/* NZ if visual match, 0 if traditional correlation */
	icxObserverType obType,		/* If not default, set a custom observer */
	xspect custObserver[3],		/* If obType = icxOT_custom */
	double xyz[3]	/* Color to match */
) {
	double ct_xyz[3];		/* XYZ on locus */
	double nxyz[3];			/* Normalised input color */
	double ct, ctde;		/* Color temperature & delta E to Black Body locus */
	icmXYZNumber wN;
	

	if (obType == icxOT_default)
		obType = icxOT_CIE_1931_2;

	if ((ct = icx_XYZ2ill_ct(ct_xyz, plank != 0 ? icxIT_Ptemp : icxIT_Dtemp,
	                         obType, custObserver, xyz, NULL, dovct)) < 0)
		error("Got bad color temperature conversion\n");

	if (de != NULL) {
		icmAry2XYZ(wN, ct_xyz);
		icmAry2Ary(nxyz, xyz);
		nxyz[0] /= xyz[1];
		nxyz[2] /= xyz[1];
		nxyz[1] /= xyz[1];
		ctde = icmXYZCIE2K(&wN, nxyz, ct_xyz);
		*de = ctde;
	}
	if (lxyz != NULL) {
		icmAry2Ary(lxyz, ct_xyz);
	}
	return ct;
}
	
/* =================================================================== */

/* Return the normal Delta E given two XYZ values, but */
/* exagerate the L* error if act L* > targ L* by a factor of fact */
extern ICCLIB_API double bwXYZLabDE(icmXYZNumber *w, double *targ, double *act, double fact) {
	double targlab[3], actlab[3], rv;

	icmXYZ2Lab(w, targlab, targ);
	icmXYZ2Lab(w, actlab, act);
	if (actlab[0] > targlab[0])
		actlab[0] = targlab[0] + fact * (actlab[0] - targlab[0]);
	rv = icmLabDE(targlab, actlab);
	return rv;
}
/* =================================================================== */

#ifdef MEAS_RES

#define NVAL 240			/* Nominal measurement value */

/* return the estimated RAMDAC precision. Return 0 if not certain */
static int comp_ramdac_prec(
	int base,		/* Base quantization to test for */
	col *ttt		/* Measurement values */
) {
	int i, j;
	double min, max;
	double val[17];
	double meas[17];

	double scale;
	double targ[17];
	double score[5];
	double bits[5];

	double bcor, bcor2;
	int bbits = 0, rbits;

	int verb = 0;

	/* Extract the measurements */
	for (i = 0; i < 17; i++) {
		val[i] = (double)i;
		meas[i] = ttt[i].XYZ[1];
	}
#ifdef DEBUG_MEAS_RES
	fprintf(dbgo,"raw measurements:\n");
	do_plot(val, meas, NULL, NULL, 17);
#endif

	/* Determine min & max, and normalize the values */
	min = 1e9, max = -1e9;
	for (i = 0; i < 17; i++) {
		if (meas[i] < min)
			min = meas[i];
		if (meas[i] > max)
			max = meas[i];
	}
	for (i = 0; i < 17; i++) {
		meas[i] = (meas[i] - min)/(max - min);
//printf("meas[%d] = %f\n",i,meas[i]);
	}

	/* Create score for each hypothesis */
	scale = 1.0;
	for (j = 0; j < 5; j++) {		/* Res 8, 9, 10, 11, 12 bits */
		int k;
		int step = 1 << (4 - j);	/* Step 16, 8, 4, 2, 1 */
		double v = 0.0;
		double merr;
		int off;
		double oscore;		/* Offset score */

		bits[j] = 8.0 + j;
		score[j] = 1e38;

		/* Try possible offsets */
		for (off = 0; off < step; off++) {
			int ik = off;		/* Initial k */

			/* Create the target response */
			for (i = 0; i < 17;) {
				int ii;			/* Actual count of loop */
				for (ii = 0, k = ik; k < step && (i+k) < 17; k++, ii++) {
					targ[i + ii] = v; 
//printf("j %d: targ[%d] = %f\n",j,i+k,v);
				}
				v += step/16.0;
				i += ii;
				ik = 0;
			}

			/* Tweak it for typical display non-linearity */
			min = 1e9, max = -1e9;
			for (i = 0; i < 17; i++) {
				targ[i] = pow((NVAL + targ[i]/16.0)/255.0, 2.2);
				if (targ[i] < min)
					min = targ[i];
				if (targ[i] > max)
					max = targ[i];
			}
			for (i = 0; i < 17; i++)
				targ[i] = (targ[i] - min)/(max - min);
			
			/* Try and make fit a little better */
			/* with a crude optimisation */
			for (k = 0; k < 50; k++) {

				merr = 0.0;
				for (i = 0; i < 17; i++)
					merr += targ[i] - meas[i];
				merr /= 17.0;
		
				for (i = 0; i < 17; i++) {
					targ[i] *= (1.0 + 0.5 * merr);
					targ[i] -= 0.5 * merr;
//				targ[i] -= merr;
				}
			}

			oscore = 0.0;
			for (i = 0; i < 17; i++) {
				double tt = targ[i] - meas[i];
				tt *= tt;
				oscore += tt;
			}

#ifdef DEBUG_MEAS_RES
			printf("%d bits %d offset score %f\n",8+j,off,oscore);
			do_plot(val, meas, targ, NULL, 17);
#endif
			/* keep best score from offsets */
			if (oscore < score[j])
				score[j] = oscore;
		}

		score[j] *= scale;
		scale *= 1.7;			/* De-weight higher bit depth slightly */
								/* Correct if overall graph is symetrical */
								/* around minimum value ? */

	}

	/* Locate the best and second best scores */
	bcor = bcor2 = 1e8;
	bbits = 0;

	for (j = 0; j < 5; j++) {
		if (score[j] < bcor) {
			bcor2 = bcor;
			bcor = score[j];
			bbits = 8+j;
		} else if (score[j] < bcor2)
			bcor2 = score[j];
	}

	rbits = bbits;

	/* Don't pick anything if it's not reasonably certain */
	if (bcor2/bcor < 1.3
	 || (bcor2/bcor < 2.1 && bcor > 0.15)
	) {
#ifdef DEBUG_MEAS_RES
		printf("bcor2/bcor < 1.3 %d\n",bcor2/bcor < 1.3);
		printf("bcor2/bcor < 2.1 %d && bcor > 0.15 %d -> %d\n", bcor2/bcor < 2.1, bcor > 0.15, bcor2/bcor < 2.1 && bcor > 0.15);
#endif
		rbits = 0;
	}

#ifdef DEBUG_MEAS_RES
	printf("Win score %f by cor %f, ratio %f\n",bcor, bcor2 - bcor, bcor2/bcor);
	printf("Best %d, returning %d bits\n",bbits,rbits);
	do_plot10(bits, score, NULL, NULL, NULL, NULL, NULL, NULL, NULL, NULL, NULL, 5, 1);
#endif

	return rbits;
}

static int meas_ramdac_prec(int base, disprd *dr) {
	col ttt[17];
	int rv, n, rbits;

	/* setup the measurements */
	for (n = 0; n < 17; n++) {
		int ii;
		ii = (NVAL * 256);
		ii += n * 16;
		ttt[n].r = ttt[n].g = ttt[n].b = ii/(double)(65535);
	}

	if ((rv = dr->read(dr, ttt, 17, 1, 17, 1, 0, instNoClamp, 0)) != 0) {
		warning("display read failed with '%s'\n",disprd_err(rv));
		return 0;
	}

	rbits = comp_ramdac_prec(base, ttt);

	/* If it failed, try the average to two sets of measurements */
	if (rbits == 0) {
		col t2[17];
		for (n = 0; n < 17; n++)
			t2[n] = ttt[n];

		if ((rv = dr->read(dr, t2, 17, 1, 17, 1, 0, instNoClamp, 0)) != 0) {
			warning("display read failed with '%s'\n",disprd_err(rv));
			return 0;
		}

		for (n = 0; n < 17; n++)
			ttt[n].XYZ[1] += t2[n].XYZ[1];
	
		rbits = comp_ramdac_prec(base, ttt);

		/* As a last resort, try just the second measurement */
		if (rbits == 0)
			rbits = comp_ramdac_prec(base, t2);
	}

	return rbits;
}

#undef NVAL

#endif /* MEAS_RES */

/* =================================================================== */

/* Default gamma */
double g_def_gamma = 2.4;

/*

  Flags used:

         ABCDEFGHIJKLMNOPQRSTUVWXYZ
  upper  .................. . .... 
  lower  .......   . ...... .... ..

*/

/* Flag = 0x0000 = default */
/* Flag & 0x0001 = list ChromeCast's */
void usage(int flag, char *diag, ...) {
	int i;
	disppath **dp;
	icompaths *icmps;
	inst2_capability cap2 = inst2_none;

	fprintf(stderr,"Calibrate a Display, Version %s\n",ARGYLL_VERSION_STR);
	fprintf(stderr,"Author: Graeme W. Gill, licensed under the AGPL Version 3\n");
	if (diag != NULL) {
		va_list args;
		fprintf(stderr,"Diagnostic: ");
		va_start(args, diag);
		vfprintf(stderr, diag, args);
		va_end(args);
		fprintf(stderr,"\n");
	}
	fprintf(stderr,"usage: dispcal [options] outfile\n");
	fprintf(stderr," -v [n]               Verbose mode\n");
#if defined(UNIX_X11)
	fprintf(stderr," -display displayname Choose X11 display name\n");
	fprintf(stderr," -d n[,m]             Choose the display n from the following list (default 1)\n");
	fprintf(stderr,"                      Optionally choose different display m for VideoLUT access\n"); 
#else
	fprintf(stderr," -d n                 Choose the display from the following list (default 1)\n");
#endif
	dp = get_displays();
	if (dp == NULL || dp[0] == NULL)
		fprintf(stderr,"    ** No displays found **\n");
	else {
		int i;
		for (i = 0; ; i++) {
			if (dp[i] == NULL)
				break;
			fprintf(stderr,"    %d = '%s'\n",i+1,dp[i]->description);
		}
	}
	free_disppaths(dp);
	fprintf(stderr," -dweb[:port]         Display via a web server at port (default 8080)\n");
	fprintf(stderr," -dcc[:n]             Display via n'th ChromeCast (default 1, ? for list)\n");
	if (flag & 0x001) {
		ccast_id **ids;
		if ((ids = get_ccids()) == NULL) {
			fprintf(stderr,"    ** Error discovering ChromeCasts **\n");
		} else {
			if (ids[0] == NULL)
				fprintf(stderr,"    ** No ChromeCasts found **\n");
			else {
				int i;
				for (i = 0; ids[i] != NULL; i++)
					fprintf(stderr,"    %d = '%s'\n",i+1,ids[i]->name);
				free_ccids(ids);
			}
		}
	}
#ifdef NT
	fprintf(stderr," -d madvr             Display via MadVR Video Renderer\n");
#endif
	fprintf(stderr," -d dummy             Dummy (non-existant, invisible) display\n");
//	fprintf(stderr," -d fake              Use a fake (ICC profile) display device for testing, fake%s if present\n",ICC_FILE_EXT);
	fprintf(stderr," -c listno            Choose instrument from the following list (default %d)\n",COMPORT);
	if ((icmps = new_icompaths(g_log)) != NULL) {
		icompath **paths;
		if ((paths = icmps->paths) != NULL) {
			int i;
			for (i = 0; ; i++) {
				if (paths[i] == NULL)
					break;
				if ((paths[i]->dtype == instSpyder1 && setup_spyd2(0) == 0)
				 || (paths[i]->dtype == instSpyder2 && setup_spyd2(1) == 0))
					fprintf(stderr,"    %d = '%s' !! Disabled - no firmware !!\n",i+1,paths[i]->name);
				else
					fprintf(stderr,"    %d = '%s'\n",i+1,paths[i]->name);
			}
		} else
			fprintf(stderr,"    ** No ports found **\n");
	}
	fprintf(stderr," -r                   Report on the calibrated display then exit\n");
	fprintf(stderr," -R                   Report on the uncalibrated display then exit\n");
	fprintf(stderr," -m                   Skip adjustment of the monitor controls\n");
	fprintf(stderr," -o [profile%s]       Create fast matrix/shaper profile [different filename to outfile%s]\n",ICC_FILE_EXT,ICC_FILE_EXT);
	fprintf(stderr," -O \"description\"     Fast ICC Profile Description string (Default \"outfile\")\n");
	fprintf(stderr," -u                   Update previous calibration and (if -o used) ICC profile VideoLUTs\n");
	fprintf(stderr," -q [vlmh]            Quality - Very Low, Low, Medium (def), High\n");
//	fprintf(stderr," -q [vfmsu]           Speed - Very Fast, Fast, Medium (def), Slow, Ultra Slow\n");
	fprintf(stderr," -p                   Use telephoto mode (ie. for a projector, if available)\n");
	fprintf(stderr," -a                   Use ambient mode (ie. for a projector, if available)\n");
	cap2 = inst_show_disptype_options(stderr, " -y                   ", icmps, 0, 0);
	fprintf(stderr," -t [temp]            White Daylight locus target, optional target temperaturee in deg. K (deflt.)\n");
	fprintf(stderr," -T [temp]            White Black Body locus target, optional target temperaturee in deg. K\n");
	fprintf(stderr," -w x,y        	      Set the target white point as chromaticity coordinates\n");
#ifdef NEVER	/* Not worth confusing people about this ? */
	fprintf(stderr," -L                   Show CCT/CDT rather than VCT/VDT during native white point adjustment\n");
#endif
	fprintf(stderr," -b bright            Set the target white brightness in cd/m^2\n");
	fprintf(stderr," -g gamma             Set the target response curve advertised gamma (Def. %3.1f)\n",g_def_gamma);
	fprintf(stderr,"                      Use \"-gl\" for L*a*b* curve\n");
	fprintf(stderr,"                      Use \"-gs\" for sRGB curve\n");
	fprintf(stderr,"                      Use \"-g709\" for REC 709 curve (should use -a ambient as well!)\n");
	fprintf(stderr,"                      Use \"-g240\" for SMPTE 240M curve (should use -a ambient as well!)\n");
	fprintf(stderr,"                      Use \"-G2.4 -f0\" for BT.1886\n"); 
	fprintf(stderr," -G gamma             Set the target response curve actual technical gamma\n");
	fprintf(stderr," -f [degree]          Amount of black level accounted for with output offset (default all output offset)\n");
	fprintf(stderr," -a ambient           Use viewing condition adjustment for ambient in Lux\n");
	fprintf(stderr," -k factor            Amount to correct black hue, 0 = none, 1 = full, Default = Automatic\n");
	fprintf(stderr," -A rate              Rate of blending from neutral to black point. Default %.1f\n",NEUTRAL_BLEND_RATE);
	fprintf(stderr," -b                   Use forced black point hack\n");
	fprintf(stderr," -B blkbright         Set the target black brightness in cd/m^2\n");
	fprintf(stderr," -e [n]               Run n verify passes on final curves\n");
	fprintf(stderr," -z                   Run only verify pass on installed calibration curves\n");
	fprintf(stderr," -P ho,vo,ss[,vs]     Position test window and scale it\n");
	fprintf(stderr,"                      ho,vi: 0.0 = left/top, 0.5 = center, 1.0 = right/bottom etc.\n");
	fprintf(stderr,"                      ss: 0.5 = half, 1.0 = normal, 2.0 = double etc.\n");
	fprintf(stderr," -F                   Fill whole screen with black background\n");
#if defined(UNIX_X11)
	fprintf(stderr," -n                   Don't set override redirect on test window\n");
#endif
	fprintf(stderr," -E                   Encode the test values for video range 16..235/255\n");
	fprintf(stderr," -K                   Run instrument calibration first (used rarely)\n");
	fprintf(stderr," -N                   Disable initial calibration of instrument if possible\n");
	fprintf(stderr," -H                   Use high resolution spectrum mode (if available)\n");
//	fprintf(stderr," -V                   Use adaptive measurement mode (if available)\n");
	if (cap2 & inst2_ccmx)
		fprintf(stderr," -X file.ccmx         Apply Colorimeter Correction Matrix\n");
	if (cap2 & inst2_ccss) {
		fprintf(stderr," -X file.ccss         Use Colorimeter Calibration Spectral Samples for calibration\n");
		fprintf(stderr," -Q observ            Choose CIE Observer for spectrometer or CCSS colorimeter data:\n");
		fprintf(stderr,"                      1931_2 (def), 1964_10, 2012_2, 2012_10, S&B 1955_2, shaw, J&V 1978_2, 1964_10c or file.cmf\n");
	}
	fprintf(stderr," -I b|w               Drift compensation, Black: -Ib, White: -Iw, Both: -Ibw\n");
	fprintf(stderr," -Y R:rate            Override measured refresh rate with rate Hz\n");
	fprintf(stderr," -Y A                 Use non-adaptive integration time mode (if available).\n");
	fprintf(stderr," -Y p                 Don't wait for the instrument to be placed on the display\n");
	fprintf(stderr," -C \"command\"         Invoke shell \"command\" each time a color is set\n");
	fprintf(stderr," -M \"command\"         Invoke shell \"command\" each time a color is measured\n");
	fprintf(stderr," -W n|h|x             Override serial port flow control: n = none, h = HW, x = Xon/Xoff\n");
	fprintf(stderr," -D [level]           Print debug diagnostics to stderr\n");
	fprintf(stderr," inoutfile            Base name for created or updated .cal and %s output files\n",ICC_FILE_EXT);
	if (icmps != NULL)
		icmps->del(icmps);
	exit(1);
}

int main(int argc, char *argv[]) {
	int i, j, k;
	int fa, nfa, mfa;					/* current argument we're looking at */
	disppath *disp = NULL;				/* Display being used */
	double hpatscale = 1.0, vpatscale = 1.0;	/* scale factor for test patch size */
	double ho = 0.0, vo = 0.0;			/* Test window offsets, -1.0 to 1.0 */
	double icalmax = 1.0;				/* Scale inst. cal. test values by this (0.0 .. 1.0) */
	int out_tvenc = 0;					/* 1 to use RGB Video Level encoding */
	int fullscreen = 0;            		/* NZ if whole screen should be filled with black */
	int verb = 0;
	int debug = 0;
	int fake = 0;						/* Use the fake device for testing */
	int override = 1;					/* Override redirect on X11 */
	int docalib = 0;					/* Do a manual instrument calibration */
	int doreport = 0;					/* 1 = Report the current uncalibrated display response */
										/* 2 = Report the current calibrated display response */
	int docontrols = 1;					/* Do adjustment of the display controls */
	int doprofile = 0;					/* Create/update ICC profile */
	icmTV iccver = ICMTV_DEFAULT;		/* ICC profile version to create */
	char *profDesc = NULL;				/* Created profile description string */
	char *copyright = NULL;				/* Copyright string */
	char *deviceMfgDesc = NULL;			/* Device manufacturer string */
	char *modelDesc = NULL;				/* Device model description string */
	int doupdate = 0;				    /* Do an update rather than a fresh calbration */
	int comport = COMPORT;				/* COM port used */
	icompaths *icmps = NULL;
	icompath *ipath = NULL;
	flow_control fc = fc_nc;			/* Default flow control */
	int ditype = 0;						/* Display type selection charater(s) */
	int tele = 0;						/* nz if telephoto mode */
	int ambient = 0;					/* nz if ambient mode */
	int nocal = 0;						/* Disable auto calibration */
	int noplace = 0;					/* Disable initial user placement check */
	int highres = 0;					/* Use high res mode if available */
	double refrate = 0.0;				/* 0.0 = default, > 0.0 = override refresh rate */ 
	int nadaptive = 0;					/* Use non-adaptive mode if available */
	int bdrift = 0;						/* Flag, nz for black drift compensation */
	int wdrift = 0;						/* Flag, nz for white drift compensation */
	double temp = 0.0;					/* Color temperature (0 = native) */
	int planckian = 0;					/* 0 = Daylight, 1 = Planckian color locus */
	int dovct = 1;						/* Show VXT rather than CXT for adjusting white point */
	double wpx = 0.0, wpy = 0.0;		/* White point xy (native) */
	double tbright = 0.0;				/* Target white brightness ( 0.0 == max)  */
	double gamma = 0.0;					/* Advertised Gamma target */
	double egamma = 0.0;				/* Effective Gamma target, NZ if set */
	double ambientl = 0.0;				/* NZ if viewing cond. adjustment to be used (Lux) */
	double bkcorrect = -1.0;			/* Level of black point correction, < 0 = auto */ 
	int bkhack = 0;
	double bkbright = 0.0;				/* Target black brightness ( 0.0 == min)  */
	int quality = -99;					/* Quality level, -2 = v, -1 = l, 0 = m, 1 = h, 2 = u */
	int isteps = 22;					/* Initial measurement steps/3 (medium) */
	int rsteps = 64;					/* Refinement measurement steps (medium) */
	double errthr = 1.5;				/* Error threshold for refinement steps (medium) */
	int thrfail = 0;					/* Set to NZ if failed to meet threshold target */
	double failerr = 0.0;				/* Delta E of worst failed target */
	int mxits = 3;						/* maximum iterations (medium) */
	int mxrpts = 12;					/* maximum repeats (medium) */
	int verify = 0;						/* Do a verify after last refinement, 2 = do only verify. */
	int nver = 0;						/* Number of verify passes after refinement */
	int webdisp = 0;					/* NZ for web display, == port number */
	int ccdisp = 0;			 			/* NZ for ChromeCast, == list index */
	ccast_id **ccids = NULL;
	ccast_id *ccid = NULL;
#ifdef NT
	int madvrdisp = 0;					/* NZ for madvr display */
#endif
	int dummydisp = 0;					/* NZ for dummy display */
	char *ccallout = NULL;				/* Change color Shell callout */
	char *mcallout = NULL;				/* Measure color Shell callout */
	char outname[MAXNAMEL+1] = { 0 };	/* Output cgats file base name */
	char iccoutname[MAXNAMEL+1] = { 0 };/* Output icc file base name */
	char ccxxname[MAXNAMEL+1] = "\000";  /* CCMX or CCSS file name */
	ccmx *cmx = NULL;					/* Colorimeter Correction Matrix */
	ccss *ccs = NULL;					/* Colorimeter Calibration Spectral Samples */
	int spec = 0;						/* Want spectral data from instrument */
	icxObserverType obType = icxOT_default;
	xspect custObserver[3];				/* If obType = icxOT_custom */
	
	disprd *dr = NULL;					/* Display patch read object */
	csamp asgrey;						/* Main calibration loop test points */
	double dispLum = 0.0;				/* Display luminence reading */
	int it;								/* verify & refine iteration */
	int rv;
	int fitord = 30;					/* More seems to make curves smoother */
	int native = 3;						/* X0 = use current per channel calibration curve */
										/* X1 = set native linear output and use ramdac high prec */
										/* 0X = use current color management cLut (MadVR) */
										/* 1X = disable color management cLUT (MadVR) */
	int noramdac = 0;					/* Will be set to nz if can't set ramdac */
	int nocm = 0;						/* Will be set to nz if can't set color managament */
	int errc;							/* Return value from new_disprd() */
	calx x;								/* Context for calibration solution */


	set_exe_path(argv[0]);				/* Set global exe_path and error_program */
	check_if_not_interactive();

#if defined(UNIX_APPLE)
	{
		SInt32 MacMajVers, MacMinVers, MacBFVers;

		/* Hmm. Maybe this should actually be 1.72 ?? */
		g_def_gamma = 1.8;

#if __MAC_OS_X_VERSION_MAX_ALLOWED >= 1040
		/* "Gestalt(gestaltSystemVersion, &MacVers)" is supported for early */
		/* systems, but was deprecated in 10.8 and causes warnings in 10.10. */
		/* gestaltSystemVersionMajor etc. isn't supported on older systems. */
		/* kCFCoreFoundationVersionNumber has been available since 10.0 */

		/* OS X 10.6+ uses a nominal gamma of 2.2 */
		if (floor(kCFCoreFoundationVersionNumber) >= kCFCoreFoundationVersionNumber10_6) {
			g_def_gamma = 2.4;
		}
#endif	/* >= 1040 */
	}
#else
	g_def_gamma = 2.4;		/* Typical CRT gamma */
#endif
	gamma = g_def_gamma;

	x.gammat = gt_power ;				/* Default gamma type */
	x.egamma = 0.0;						/* Default effective gamma none */
	x.oofff = 1.0;						/* Default is all output ofset */
	x.vc = 0;							/* No viewing conditions adjustment */
	x.svc = NULL;
	x.dvc = NULL;
	x.nbrate = NEUTRAL_BLEND_RATE;		/* Rate of blending from black point to neutral axis */

#ifdef DEBUG_OFFSET
	ho = 0.8;
	vo = -0.8;
#endif

#if defined(DEBUG) || defined(DEBUG_OFFSET) || defined(DEBUG_PLOT)
	printf("!!!!!! Debug turned on !!!!!!\n");
#endif


	if (argc <= 1)
		usage(0,"Too few arguments");

	/* Process the arguments */
	mfa = 1;        /* Minimum final arguments */
	for (fa = 1;fa < argc;fa++) {
		nfa = fa;					/* skip to nfa if next argument is used */
		if (argv[fa][0] == '-')		/* Look for any flags */
			{
			char *na = NULL;		/* next argument after flag, null if none */

			if (argv[fa][2] != '\000')
				na = &argv[fa][2];		/* next is directly after flag */
			else {
				if ((fa+1+mfa) < argc) {
					if (argv[fa+1][0] != '-')
						{
						nfa = fa + 1;
						na = argv[nfa];		/* next is seperate non-flag argument */
					}
				}
			}

			if (argv[fa][1] == '?' || argv[fa][1] == '-') {
				usage(0,"Usage requested");

			} else if (argv[fa][1] == 'v') {
				verb = 1;
				if (na != NULL && na[0] >= '0' && na[0] <= '9') {
					verb = atoi(na);
					fa = nfa;
				}
				g_log->verb = verb;

			/* Display number */
			} else if (argv[fa][1] == 'd') {
				if (na != NULL
				 && (strncmp(na,"web",3) == 0
				  || strncmp(na,"WEB",3) == 0)) {
					webdisp = 8080;
					if (na[3] == ':') {
						webdisp = atoi(na+4);
						if (webdisp == 0 || webdisp > 65535)
							usage(0,"Web port number must be in range 1..65535");
					}
					fa = nfa;
				} else if (na != NULL && (
				    strncmp(na,"cc",2) == 0
				 || strncmp(na,"CC",2) == 0)) {
					ccdisp = 1;
					if (na[2] == ':') {
						if (na[3] < '0' || na[3] > '9')
							usage(0x0001,"Available ChromeCasts");

						ccdisp = atoi(na+3);
						if (ccdisp <= 0)
							usage(0,"ChromeCast number must be in range 1..N");
					}
					fa = nfa;
#ifdef NT
				} else if (na != NULL && strncmp(na,"madvr",5) == 0
				 || strncmp(na,"MADVR",5) == 0) {
					madvrdisp = 1;
					fa = nfa;
#endif
				} else if ((na != NULL && strncmp(na,"dummy",5) == 0)
				 || strncmp(na,"DUMMY",5) == 0) {
					dummydisp = 1;
					fa = nfa;
				} else {
#if defined(UNIX_X11)
					int ix, iv;

					if (strcmp(&argv[fa][2], "isplay") == 0 || strcmp(&argv[fa][2], "ISPLAY") == 0) {
						if (++fa >= argc || argv[fa][0] == '-') usage(0,"Parameter expected following -display");
						setenv("DISPLAY", argv[fa], 1);
					} else {
						if (na == NULL) usage(0,"Parameter expected following -d");
						fa = nfa;
						if (strcmp(na,"fake") == 0) {
							fake = 1;
						} else {
							if (sscanf(na, "%d,%d",&ix,&iv) != 2) {
								ix = atoi(na);
								iv = 0;
							}
							if (disp != NULL)
								free_a_disppath(disp);
							if ((disp = get_a_display(ix-1)) == NULL)
								usage(0,"-d parameter %d out of range",ix);
							if (iv > 0)
								disp->rscreen = iv-1;
						}
					}
#else
					int ix;
					if (na == NULL) usage(0,"Parameter expected following -d");
					fa = nfa;
					if (strcmp(na,"fake") == 0) {
						fake = 1;
					} else {
						ix = atoi(na);
						if (disp != NULL)
							free_a_disppath(disp);
						if ((disp = get_a_display(ix-1)) == NULL)
							usage(0,"-d parameter %d out of range",ix);
					}
#endif
				}

			} else if (argv[fa][1] == 'E') {
				out_tvenc = 1;

			} else if (argv[fa][1] == 'K') {
				docalib = 1;

			} else if (argv[fa][1] == 'N') {
				nocal = 1;

			/* High res mode */
			} else if (argv[fa][1] == 'H') {
				highres = 1;

			/* Adaptive mode - now default, so flag is deprecated */
			} else if (argv[fa][1] == 'V') {
				warning("dispcal -V flag is deprecated");

			/* Colorimeter Correction Matrix */
			/* or Colorimeter Calibration Spectral Samples */
			} else if (argv[fa][1] == 'X') {
				int ix;
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected following -X");
				strncpy(ccxxname,na,MAXNAMEL-1); ccxxname[MAXNAMEL-1] = '\000';

			/* Drift Compensation */
			} else if (argv[fa][1] == 'I') {
				fa = nfa;
				if (na == NULL || na[0] == '\000') usage(0,"Parameter expected after -I");
				for (i=0; ; i++) {
					if (na[i] == '\000')
						break;
					if (na[i] == 'b' || na[i] == 'B')
						bdrift = 1;
					else if (na[i] == 'w' || na[i] == 'W')
						wdrift = 1;
					else
						usage(0,"-I parameter '%c' not recognised",na[i]);
				}

			/* Spectral Observer type */
			} else if (argv[fa][1] == 'Q') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expecte after -Q");
				if (strcmp(na, "1931_2") == 0) {			/* Classic 2 degree */
					obType = icxOT_CIE_1931_2;
				} else if (strcmp(na, "1964_10") == 0) {	/* Classic 10 degree */
					obType = icxOT_CIE_1964_10;
				} else if (strcmp(na, "2012_2") == 0) {		/* Latest 2 degree */
					obType = icxOT_CIE_2012_2;
				} else if (strcmp(na, "2012_10") == 0) {	/* Latest 10 degree */
					obType = icxOT_CIE_2012_10;
				} else if (strcmp(na, "1964_10c") == 0) {	/* 10 degree corrected */
					obType = icxOT_CIE_1964_10c;
				} else if (strcmp(na, "1955_2") == 0) {		/* Stiles and Burch 1955 2 degree */
					obType = icxOT_Stiles_Burch_2;
				} else if (strcmp(na, "1978_2") == 0) {		/* Judd and Voss 1978 2 degree */
					obType = icxOT_Judd_Voss_2;
				} else if (strcmp(na, "shaw") == 0) {		/* Shaw and Fairchilds 1997 2 degree */
					obType = icxOT_Shaw_Fairchild_2;
				} else {	/* Assume it's a filename */
					obType = icxOT_custom;
					if (read_cmf(custObserver, na) != 0)
						usage(0,"Failed to read custom observer CMF from -Q file '%s'",na);
				}

			/* Change color callout */
			} else if (argv[fa][1] == 'C') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected after -C");
				ccallout = na;

			/* Measure color callout */
			} else if (argv[fa][1] == 'M') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected after -M");
				mcallout = na;

			/* Serial port flow control */
			} else if (argv[fa][1] == 'W') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected following -W");
				if (na[0] == 'n' || na[0] == 'N')
					fc = fc_None;
				else if (na[0] == 'h' || na[0] == 'H')
					fc = fc_Hardware;
				else if (na[0] == 'x' || na[0] == 'X')
					fc = fc_XonXOff;
				else
					usage(0,"-W parameter '%c' not recognised",na[0]);

			/* Debug coms */
			} else if (argv[fa][1] == 'D') {
				debug = 1;
				if (na != NULL && na[0] >= '0' && na[0] <= '9') {
					debug = atoi(na);
					fa = nfa;
				}
				g_log->debug = debug;
				callback_ddebug = 1;		/* dispwin global */

			/* Black point correction amount */
			} else if (argv[fa][1] == 'k') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected following -k");
				bkcorrect = atof(na);
				if (bkcorrect < 0.0 || bkcorrect > 1.0) usage(0,"-k parameter must be between 0.0 and 1.0");
			/* Neutral blend rate (power) */
			} else if (argv[fa][1] == 'A') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected following -A");
				x.nbrate = atof(na);
				if (x.nbrate < 0.05 || x.nbrate > 20.0) usage(0,"-A parameter must be between 0.05 and 20.0");
			/* Black brightness */
			} else if (argv[fa][1] == 'B') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected after -B");
				bkbright = atof(na);
				if (bkbright <= 0.0 || bkbright > 100000.0) usage(0,"-B parameter %f out of range",bkbright);

			/* Number of verify passes */
			} else if (argv[fa][1] == 'e') {
				if (verify == 0)
					verify = 1;
				nver = 1;
				if (na != NULL && na[0] >= '0' && na[0] <= '9') {
					nver = atoi(na);
					fa = nfa;
				}

			} else if (argv[fa][1] == 'z') {
				verify = 2;
				if (nver == 0)
					nver = 1;
				mfa = 0;

#if defined(UNIX_X11)
			} else if (argv[fa][1] == 'n') {
				override = 0;
#endif /* UNIX */
			/* COM port  */
			} else if (argv[fa][1] == 'c') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected following -c");
				comport = atoi(na);
				if (comport < 1 || comport > 50) usage(0,"-c parameter %d out of range",comport);

			/* Telephoto */
			} else if (argv[fa][1] == 'p') {
				tele = 1;
				
			} else if (argv[fa][1] == 'r' || argv[fa][1] == 'R') {
				if (argv[fa][1] == 'R')
					doreport = 1;		/* raw */
				else
					doreport = 2;		/* Calibrated */
				mfa = 0;

			} else if (argv[fa][1] == 'm') {
				docontrols = 0;

			/* Output/update ICC profile [optional different name] */
			} else if (argv[fa][1] == 'o') {
				doprofile = 1;

				if (na != NULL) {	/* Found an optional icc profile name */
					fa = nfa;
					strncpy(iccoutname,na,MAXNAMEL); iccoutname[MAXNAMEL] = '\000';
				}

			/* Fast Profile Description */
			} else if (argv[fa][1] == 'O') {
				fa = nfa;
				if (na == NULL) usage(0,"Expect argument to profile description flag -O");
				profDesc = na;


			/* Update calibration and (optionally) profile */
			} else if (argv[fa][1] == 'u') {
				doupdate = 1;
				docontrols = 0;

			/* Speed/Quality */
			} else if (argv[fa][1] == 'q') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected following -q");
    			switch (na[0]) {
					case 'L':			/* Test value */
						quality = -3;
						break;
					case 'v':			/* very fast */
						quality = -2;
						break;
					case 'f':			/* fast */
					case 'l':
						quality = -1;
						break;
					case 'm':			/* medium */
					case 'M':
						quality = 0;
						break;
					case 's':			/* slow */
					case 'h':
					case 'H':
						quality = 1;
						break;
					case 'u':			/* ultra slow */
					case 'U':
						quality = 2;
						break;
					default:
						usage(0,"-q parameter '%c' not recognised",na[0]);
				}

			/* Display type */
			} else if (argv[fa][1] == 'y') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected after -y");
				ditype = na[0];
				if (ditype == '_' && na[1] != '\000')
					ditype = ditype << 8 | na[1];

			/* Daylight color temperature */
			} else if (argv[fa][1] == 't' || argv[fa][1] == 'T') {
				if (argv[fa][1] == 'T')
					planckian = 1;
				else
					planckian = 0;
				if (na != NULL) {
					fa = nfa;
					temp = atof(na);
					if (temp < 1000.0 || temp > 15000.0) usage(0,"-%c parameter %f out of range",argv[fa][1], temp);
				}

			/* White point as x, y */
			} else if (argv[fa][1] == 'w') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected after -w");
				if (sscanf(na, " %lf,%lf ", &wpx, &wpy) != 2)
					usage(0,"-w parameter '%s' not recognised",na);

			/* Show CXT rather than VXT when adjusting native white point */
			} else if (argv[fa][1] == 'L') {
				dovct = 0;

			/* Black point hack/White brightness */
			} else if (argv[fa][1] == 'b') {
				if (na == NULL) {
					bkhack = 1;
				} else {
					fa = nfa;
					/* if (na == NULL) usage(0,"Parameter expected after -b"); */
					tbright = atof(na);
					if (tbright <= 0.0 || tbright > 100000.0) usage(0,"-b parameter %f out of range",tbright);
				}

			/* Target transfer curve */
			} else if (argv[fa][1] == 'g') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected after -g");
				if ((na[0] == 'l' || na[0] == 'L') && na[1] == '\000')
					x.gammat = gt_Lab;
				else if ((na[0] == 's' || na[0] == 'S') && na[1] == '\000')
					x.gammat = gt_sRGB;
				else if (strcmp(na, "709") == 0)
					x.gammat = gt_Rec709;
				else if (strcmp(na, "240") == 0)
					x.gammat = gt_SMPTE240M;
				else {
					gamma = atof(na);
					if (gamma <= 0.0 || gamma > 10.0) usage(0,"-g parameter %f out of range",gamma);
					x.gammat = gt_power;
				}

			/* Effective gamma power */
			} else if (argv[fa][1] == 'G') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected after -G");
				egamma = atof(na);
				if (egamma <= 0.0 || egamma > 10.0) usage(0,"-G parameter %f out of range",egamma);
				x.gammat = gt_power;

			/* Degree of output offset */
			} else if (argv[fa][1] == 'f') {
				fa = nfa;
				if (na == NULL) {
					x.oofff = 0.0;
				} else {
					x.oofff = atof(na);
					if (x.oofff < 0.0 || x.oofff > 1.0)
						usage(0,"-f parameter %f out of range",x.oofff);
				}

			/* Ambient mode or Ambient light level */
			} else if (argv[fa][1] == 'a') {
				fa = nfa;
				if (na == NULL) {
					ambient = 1;
				} else {
					ambientl = atof(na);
					if (ambientl < 0.0)
						usage(0,"-a parameter %f out of range",ambientl);
				}

			/* Test patch offset and size */
			} else if (argv[fa][1] == 'P') {
				fa = nfa;
				if (na == NULL) usage(0,"Parameter expected after -P");
				if (sscanf(na, " %lf,%lf,%lf,%lf ", &ho, &vo, &hpatscale, &vpatscale) == 4) {
					;
				} else if (sscanf(na, " %lf,%lf,%lf ", &ho, &vo, &hpatscale) == 3) {
					vpatscale = hpatscale;
				} else {
					usage(0,"-P parameter '%s' not recognised",na);
				}
				if (ho < 0.0 || ho > 1.0
				 || vo < 0.0 || vo > 1.0
				 || hpatscale <= 0.0 || hpatscale > 50.0
				 || vpatscale <= 0.0 || vpatscale > 50.0)
					usage(0,"-P parameters %f %f %f %f out of range",ho,vo,hpatscale,vpatscale);
				ho = 2.0 * ho - 1.0;
				vo = 2.0 * vo - 1.0;

			/* Full screen black background */
			} else if (argv[fa][1] == 'F') {
				fullscreen = 1;

			/* Extra flags */
			} else if (argv[fa][1] == 'Y') {
				if (na == NULL)
					usage(0,"Flag '-Y' expects extra flag");
			
				if (na[0] == 'R') {
					if (na[1] != ':')
						usage(0,"-Y R:rate syntax incorrect");
					refrate = atof(na+2);
					if (refrate < 5.0 || refrate > 150.0)
						usage(0,"-Y R:rate %f Hz not in valid range",refrate);
				} else if (na[0] == 'A') {
					nadaptive = 1;
				} else if (na[0] == 'p') {
					noplace = 1;
				} else {
					usage(0,"Flag '-Y %c' not recognised",na[0]);
				}
				fa = nfa;

			} else 
				usage(0,"Flag '-%c' not recognised",argv[fa][1]);
		} else
			break;
	}



	if (bkhack && bkbright > 0.0) {
		error("Can't use -b black point hack and set target black brightness");
	}

	if (bkhack && bkcorrect != 0.0) {
		if (bkcorrect > 0.0)
			warning("Due to -b flag, -k factor will be set to 0.0");
		bkcorrect = 0.0;
	}

	/* No explicit display has been set */
	if (
#ifndef SHOW_WINDOW_ONFAKE
	!fake 
#endif
#ifdef NT
	 && madvrdisp == 0
#endif
	 && dummydisp == 0
	 && webdisp == 0
	 && ccdisp == 0
	 && disp == NULL) {
		int ix = 0;
#if defined(UNIX_X11)
		char *dn, *pp;

		if ((dn = getenv("DISPLAY")) != NULL) {
			if ((pp = strrchr(dn, ':')) != NULL) {
				if ((pp = strchr(pp, '.')) != NULL) {
					if (pp[1] != '\000')
						ix = atoi(pp+1);
				}
			}
		}
#endif
		if ((disp = get_a_display(ix)) == NULL)
			error("Unable to open the default display");
	}

	/* See if there is an environment variable ccxx */
	if (ccxxname[0] == '\000') {
		char *na;
		if ((na = getenv("ARGYLL_COLMTER_CAL_SPEC_SET")) != NULL) {
			strncpy(ccxxname,na,MAXNAMEL-1); ccxxname[MAXNAMEL-1] = '\000';

		} else if ((na = getenv("ARGYLL_COLMTER_COR_MATRIX")) != NULL) {
			strncpy(ccxxname,na,MAXNAMEL-1); ccxxname[MAXNAMEL-1] = '\000';
		}
	}

	/* Load up CCMX or CCSS */
	if (ccxxname[0] != '\000') {
		if ((cmx = new_ccmx()) == NULL
		  || cmx->read_ccmx(cmx, ccxxname)) {
			if (cmx != NULL) {
				cmx->del(cmx);
				cmx = NULL;
			}
			
			/* CCMX failed, try CCSS */
			if ((ccs = new_ccss()) == NULL
			  || ccs->read_ccss(ccs, ccxxname)) {
				if (ccs != NULL) {
					ccs->del(ccs);
					ccs = NULL;
					error("Reading CCMX/CCSS File '%s' failed\n", ccxxname);
				}
			}
		}
	}

	if (fake)
		comport = FAKE_DEVICE_PORT;
	if ((icmps = new_icompaths(g_log)) == NULL)
		error("Finding instrument paths failed");
	if ((ipath = icmps->get_path(icmps, comport)) == NULL)
		error("No instrument at port %d",comport);

	/* If we've requested ChromeCast, look it up */
	if (ccdisp) {
		if ((ccids = get_ccids()) == NULL)
			error("discovering ChromeCasts failed");
		if (ccids[0] == NULL)
			error("There are no ChromeCasts to use\n");
		for (i = 0; ccids[i] != NULL; i++)
			;
		if (ccdisp < 1 || ccdisp > i)
			error("Chosen ChromeCasts (%d) is outside list (1..%d)\n",ccdisp,i);
		ccid = ccids[ccdisp-1];
	}


	if (docalib) {
		if ((rv = disprd_calibration(ipath, fc, ditype, -1, 0, tele, ambient, nadaptive, nocal, disp,
		                             webdisp, ccid,
#ifdef NT
			                         madvrdisp,
#endif
			                         dummydisp, icalmax, out_tvenc, fullscreen, override,
			                         100.0 * hpatscale, 100.0 * vpatscale,
			                         ho, vo, g_log)) != 0) {
			error("docalibration failed with return value %d\n",rv);
		}
	}

	if (verify != 2 && doreport == 0) {
		/* Get the file name argument */
		if (fa >= argc || argv[fa][0] == '-') usage(0,"Output filname parameter not found");
		strncpy(outname,argv[fa],MAXNAMEL-4); outname[MAXNAMEL-4] = '\000';
		strcat(outname,".cal");
		if (iccoutname[0] == '\000') {
			strncpy(iccoutname,argv[fa++],MAXNAMEL-4); iccoutname[MAXNAMEL-4] = '\000';
			strcat(iccoutname,ICC_FILE_EXT);
		}
	}

	if (verify == 2) {
		if (doupdate)
			warning("Update flag ignored because we're doing a verify only");
		doupdate = 0;
		docontrols = 0;
	}

	if (doreport != 0) {
		if (verify == 2)
			warning("Verify flag ignored because we're doing a report only");
		verify = 0;
		nver = 0;
	}

	/* Normally calibrate against native response */
	if (verify == 2 || doreport == 2)
		native = 0;	/* But measure current calibrated & CM response for verify or report calibrated */ 

	/* Get ready to do some readings */
	if ((dr = new_disprd(&errc, ipath, fc, ditype, -1, 0, tele, ambient, nadaptive, nocal, noplace,
	                     highres, refrate, native, &noramdac, &nocm, NULL, 0,
		                 disp, icalmax, out_tvenc, fullscreen, override, webdisp, ccid,
#ifdef NT
		                 madvrdisp,
#endif
		                 dummydisp, ccallout, mcallout, 0,
	                     100.0 * hpatscale, 100.0 * vpatscale, ho, vo,
	                     ccs != NULL ? ccs->dtech : cmx != NULL ? cmx->dtech : disptech_unknown,
	                     cmx != NULL ? cmx->cc_cbid : 0,
	                     cmx != NULL ? cmx->matrix : NULL,
	                     ccs != NULL ? ccs->samples : NULL, ccs != NULL ? ccs->no_samp : 0,
	                     spec, obType, custObserver, bdrift, wdrift,
	                     "fake" ICC_FILE_EXT, g_log)) == NULL)
		error("new_disprd() failed with '%s'\n",disprd_err(errc));

	if ((native & 1) && noramdac) {
		warning("Unable to access to VideoLUTs so can't be sure colors are native");
		if (doprofile)
			warning("Profile will reflect the as-is display response and not contain a 'vcgt' tag");
		native &= ~1;

		if (doupdate && doprofile)
			error("Can't update a profile that doesn't use the 'vcgt' tag for calibration");
	}

	if (icmps != NULL) {
		icmps->del(icmps);
		icmps = NULL;
	}
	if (cmx != NULL) {
		cmx->del(cmx);
		cmx = NULL;
	}
	if (ccs != NULL) {
		ccs->del(ccs);
		ccs = NULL;
	}

	/* - - - - - - - - - - - - - - - - - - - - - - - - - - - - */
	if (doreport) {
		col tcols[3] = {	/* Base set of test colors */
			{ 0.0, 0.0, 0.0 },
			{ 0.5, 0.5, 0.5 },
			{ 1.0, 1.0, 1.0 }
		};
		double cct, cct_de;		/* Color temperatures and DE 2K */
		double cdt, cdt_de;
		double vct, vct_de;
		double vdt, vdt_de;
		double cgamma, w[3], wp[2];
		int sigbits = 0;	/* Number of significant bits in VideoLUT/display/instrument */

		if ((rv = dr->read(dr, tcols, 3, 1, 3, 1, 0, instClamp, 0)) != 0) {
			dr->del(dr);
			error("display read failed with '%s'\n",disprd_err(rv));
		} 

//printf("~1 Got black = %f, half = %f, white = %f\n",tcols[0].XYZ[1],tcols[1].XYZ[1],tcols[2].XYZ[1]);
		/* Normalised XYZ white point */
		w[0] = tcols[2].XYZ[0]/tcols[2].XYZ[1];
		w[1] = tcols[2].XYZ[1]/tcols[2].XYZ[1];
		w[2] = tcols[2].XYZ[2]/tcols[2].XYZ[1];

		/* White point chromaticity coordinates */
		wp[0] = w[0]/(w[0] + w[1] + w[2]);
		wp[1] = w[1]/(w[0] + w[1] + w[2]);

		cct = comp_ct(&cct_de, NULL, 1, 0, obType, custObserver, w); /* Compute CCT */
		cdt = comp_ct(&cdt_de, NULL, 0, 0, obType, custObserver, w); /* Compute CDT */
		vct = comp_ct(&vct_de, NULL, 1, 1, obType, custObserver, w); /* Compute VCT */
		vdt = comp_ct(&vdt_de, NULL, 0, 1, obType, custObserver, w); /* Compute VDT */

		/* Compute advertised current gamma - use the gross curve shape for robustness */
		cgamma = pop_gamma(tcols[0].XYZ[1], tcols[1].XYZ[1], tcols[2].XYZ[1]);

#ifdef MEAS_RES
#ifdef NEVER		// Use new code
		/* See if we can detect what sort of precision the LUT entries */
		/* have. Our ability to detect this may be limited by the instrument */
		/* (ie. Huey and Spyder 2) */
		if (doreport == 1) {
#define MAX_RES_SAMPS 24
			col ttt[MAX_RES_SAMPS];
			int res_samps = 9;
			double a0, a1, a2, dd;
			int n;
			int issig = 0;

			if (verb)
				printf("Measuring VideoLUT table entry precision.\n");

			/* Run a small state machine until we come to a conclusion */
			sigbits = 8;
			for (issig = 0; issig < 2; ) {

				DBG((dbgo,"Trying %d bits\n",sigbits));
				/* Do the test */
				for (n = 0; n < res_samps; n++) {
					double v;
#if defined(__APPLE__) && defined(__POWERPC__)
					gcc_bug_fix(sigbits);
#endif
					/* Notional test value */
					v = (7 << (sigbits-3))/((1 << sigbits) - 1.0);
					/* And -1, 0 , +1 bit test values */
					if ((n % 3) == 2)
						v += 1.0/((1 << sigbits) - 1.0);
					else if ((n % 3) == 1)
						v += 0.0/((1 << sigbits) - 1.0);
					else
						v += -1.0/((1 << sigbits) - 1.0);
					ttt[n].r = ttt[n].g = ttt[n].b = v; 
				}
				if ((rv = dr->read(dr, ttt, res_samps, 1, res_samps, 1, 0, instNoClamp)) != 0) {
					dr->del(dr);
					error("display read failed with '%s'\n",disprd_err(rv));
				}
				/* Average the readings for each test value */
				a0 = a1 = a2 = 0.0;
				for (n = 0; n < res_samps; n++) {
					double v = ttt[n].XYZ[1];
					if ((n % 3) == 2) {
						a2 += v;
					} else if ((n % 3) == 1) {
						a1 += v;
					} else {
						a0 += v;
					}
				}
				a0 /= (res_samps / 3.0);
				a1 /= (res_samps / 3.0);
				a2 /= (res_samps / 3.0);
				DBG((dbgo,"Bits %d: -1: %f 0: %f +1 %f\n",sigbits, a0, a1, a2));
				/* Judge significance of any differences */
				dd = 0.0;
				for (n = 0; n < res_samps; n++) {
					double tt;
					if ((n % 3) == 2)
						tt = fabs(a2 - ttt[n].XYZ[1]);
					else if ((n % 3) == 1)
						tt = fabs(a1 - ttt[n].XYZ[1]);
					else
						tt = fabs(a0 - ttt[n].XYZ[1]);
					dd += tt * tt;
				}
				dd /= res_samps;
				dd = sqrt(dd);
				if (fabs(a1 - a0) > (2.0 * dd) && fabs(a2 - a1) > (2.0 * dd))
					issig = 1;		/* Noticable difference */
				else
					issig = 0;		/* No noticable difference */
				DBG((dbgo,"Bits %d: Between = %f, %f within = %f, sig = %s\n",sigbits, fabs(a1 - a0), fabs(a2 - a1), dd, issig ? "yes" : "no"));

				switch(sigbits) {
					case 8:				/* Do another trial */
						if (issig) {
							sigbits = 10;
							res_samps = 9;
						} else {
							sigbits = 6;
						}
						break;
					case 6:				/* Do another trial or give up */
						if (issig) {
							sigbits = 7;
							res_samps = 6;
						} else {
							sigbits = 0;
							issig = 2;	/* Give up */
						}
						break;
					case 7:				/* Terminal */
						if (!issig)
							sigbits = 6;
						issig = 2;		/* Stop here */
						break;
					case 10:			/* Do another trial */
						if (issig) {
							sigbits = 12;
							res_samps = 12;
						} else {
							sigbits = 9;
						}
						break;
					case 12:			/* Do another trial or give up */
						if (issig) {
							issig = 2;	/* Stop here */
						} else {
							sigbits = 11;
						}
						break;
					case 11:			/* Terminal */
						if (!issig)
							sigbits = 10;
						issig = 2;		/* Stop here */
						break;
					case 9:				/* Terminal */
						if (!issig)
							sigbits = 8;
						issig = 2;		/* Stop here */
						break;

					default:
						error("Unexpected number of bits in depth test (bits)",sigbits);
				}
			}
		}
# else	/* ! NEVER */
		/* See if we can deternine what sort of precision the LUT entries */
		/* have. Our ability to detect this may be limited by the instrument */
		/* (ie. Huey and Spyder 2) */
		/* We assume that we need to just detect 8 to 12 bits */
		if (doreport == 1) {
			sigbits = meas_ramdac_prec(8, dr);
		}
#endif /* NEVER */
#endif	/* MEAS_RES */

		if (doreport == 2)
			printf("Current calibration response:\n");
		else
			printf("Uncalibrated response:\n");
		printf("Black level = %.4f cd/m^2\n",tcols[0].XYZ[1]);
		printf("50%%   level = %.2f cd/m^2\n",tcols[1].XYZ[1]);
		printf("White level = %.2f cd/m^2\n",tcols[2].XYZ[1]);
		printf("Aprox. gamma = %.2f\n",cgamma);
		printf("Contrast ratio = %.0f:1\n",tcols[2].XYZ[1]/tcols[0].XYZ[1]);
		printf("White chromaticity coordinates %.4f, %.4f\n",wp[0],wp[1]);
		printf("White    Correlated Color Temperature = %.0fK, DE 2K to locus = %4.1f\n",cct,cct_de);
		printf("White Correlated Daylight Temperature = %.0fK, DE 2K to locus = %4.1f\n",cdt,cdt_de);
		printf("White        Visual Color Temperature = %.0fK, DE 2K to locus = %4.1f\n",vct,vct_de);
		printf("White     Visual Daylight Temperature = %.0fK, DE 2K to locus = %4.1f\n",vdt,vdt_de);
#ifdef	MEAS_RES
		if (doreport == 1) {
			if (sigbits == 0) {
				warning("Unable to determine effective Video LUT entry bit depth");
			} else {
				printf("Effective Video LUT entry depth seems to be %d bits\n",sigbits);
			}
		}
#endif	/* MEAS_RES */
		dr->del(dr);
		exit(0);
	}

	/* - - - - - - - - - - - - - - - - - - - - - - - - - - - - */
	/* If we're updating, retrieve the previously used settings, */
	/* and device model */
	if (doupdate) {
		cgats *icg;						/* output cgats structure */
		int nsamp;
		mcvco *rdv[3];					/* Scattered data for ramdac curves */
		int fi;							/* Field index */
		int si[4];						/* Set fields */

		if (verb) {
			if (doprofile)
				printf("Updating previous calibration and profile\n");
			else
				printf("Updating previous calibration\n");
		}

		icg = new_cgats();				/* Create a CGATS structure */
		icg->add_other(icg, "CAL"); 	/* our special type is Calibration file */
		
		if (icg->read_name(icg, outname)) {
			dr->del(dr);
			error("Can't update '%s' - read error : %s",outname, icg->e.m);
		}

		if (icg->ntables == 0
		 || icg->t[0].tt != tt_other || icg->t[0].oi != 0
		 || icg->t[1].tt != tt_other || icg->t[1].oi != 0) {
			dr->del(dr);
			error("Can't update '%s' - wrong type of file",outname);
		}
		if (icg->ntables < 2) {
			dr->del(dr);
			error("Can't update '%s' - there aren't two tables",outname);
		}

		out_tvenc = 0;
		if ((fi = icg->find_kword(icg, 0, "TV_OUTPUT_ENCODING")) >= 0) {
			if (strcmp(icg->t[0].kdata[fi], "YES") == 0
			 || strcmp(icg->t[0].kdata[fi], "yes") == 0)
				out_tvenc = 1;
		}

//printf("~1 reading previous cal, got 2 tables\n");

		/* Read in the setup, user and model values */

		if (ditype == 0) {	/* If the use hasn't set anything */
			if ((fi = icg->find_kword(icg, 0, "DEVICE_TYPE")) >= 0) {
				if (strcmp(icg->t[0].kdata[fi], "CRT") == 0)
					ditype = 'c';
				else if (strcmp(icg->t[0].kdata[fi], "LCD") == 0)
					ditype = 'l';
				else {
					ditype = icg->t[0].kdata[fi][0];	// Hmm. not handling '_' ...
				}
			}
		}
//printf("~1 dealt with device type\n");
	
		if ((fi = icg->find_kword(icg, 0, "TARGET_WHITE_XYZ")) < 0) {
			dr->del(dr);
			error("Can't update '%s' - can't find field 'TARGET_WHITE_XYZ'",outname);
		}
		if (sscanf(icg->t[0].kdata[fi], "%lf %lf %lf", &x.twh[0], &x.twh[1], &x.twh[2]) != 3) {
			dr->del(dr);
			error("Can't update '%s' - reading field 'TARGET_WHITE_XYZ' failed",outname);
		}
		x.nwh[0] = x.twh[0] / x.twh[1];
		x.nwh[1] = x.twh[1] / x.twh[1];
		x.nwh[2] = x.twh[2] / x.twh[1];

		if ((fi = icg->find_kword(icg, 0, "NATIVE_TARGET_WHITE")) >= 0) {
			wpx = wpy = 0.0;
			temp = 0.0;
			tbright = 0.0;
		} else {
			wpx = wpy = 0.0001;
			temp = 1.0;
			tbright = 1.0;
		}
//printf("~1 dealt with target white\n");

		if ((fi = icg->find_kword(icg, 0, "TARGET_GAMMA")) < 0) {
			dr->del(dr);
			error("Can't update '%s' - can't find field 'TARGET_GAMMA'",outname);
		}
		if (strcmp(icg->t[0].kdata[fi], "L_STAR") == 0)
			x.gammat = gt_Lab;
		else if (strcmp(icg->t[0].kdata[fi], "sRGB") == 0)
			x.gammat = gt_sRGB;
		else if (strcmp(icg->t[0].kdata[fi], "REC709") == 0)
			x.gammat = gt_Rec709;
		else if (strcmp(icg->t[0].kdata[fi], "SMPTE240M") == 0)
			x.gammat = gt_SMPTE240M;
		else {
			x.gammat = 0;
			gamma = atof(icg->t[0].kdata[fi]);
			
			if (fabs(gamma) < 0.1 || fabs(gamma) > 5.0) {
				dr->del(dr);
				error("Can't update '%s' - field 'TARGET_GAMMA' has bad value %f",outname,fabs(gamma));
			}
			if (gamma < 0.0) {	/* Effective gamma = actual power value */
				egamma = -gamma;
			}
		}
		if ((fi = icg->find_kword(icg, 0, "DEGREE_OF_BLACK_OUTPUT_OFFSET")) < 0) {
			/* Backward compatibility if value is not present */
			if (x.gammat == gt_Lab || x.gammat == gt_sRGB)
				x.oofff = 1.0;
			else
				x.oofff = 0.0;
		} else {
			x.oofff = atof(icg->t[0].kdata[fi]);
		}

		if ((fi = icg->find_kword(icg, 0, "BLACK_POINT_HACK")) < 0) {
			bkhack = 0;
		} else {
			if (strcmp(icg->t[0].kdata[fi], "YES") == 0
			 || strcmp(icg->t[0].kdata[fi], "yes") == 0) {
				bkhack = 1;
			} else {
				bkhack = 0;
			}
		}

		if ((fi = icg->find_kword(icg, 0, "TARGET_BLACK_BRIGHTNESS")) < 0) {
			bkbright = 0.0;		/* Native */
		} else {
			bkbright = atof(icg->t[0].kdata[fi]);
		}
		if (bkhack && bkbright > 0.0) {
			error("Can't update '%s' - BLACK_POINT_HACK and TARGET_BLACK_BRIGHTNESS conflict",outname);
		}
		if ((fi = icg->find_kword(icg, 0, "BLACK_POINT_CORRECTION")) < 0) {
			dr->del(dr);
			error("Can't update '%s' - can't find field 'BLACK_POINT_CORRECTION'",outname);
		}
		bkcorrect = atof(icg->t[0].kdata[fi]);
		if (bkcorrect < 0.0 || bkcorrect > 1.0) {
			dr->del(dr);
			error("Can't update '%s' - field 'BLACK_POINT_CORRECTION' has bad value %f",outname,bkcorrect);
		}

		if (bkhack && bkcorrect != 0.0) {
			if (bkcorrect > 0.0)
				warning("Due to -b flag, -k factor will be set to 0.0");
			bkcorrect = 0.0;
		}

		if ((fi = icg->find_kword(icg, 0, "BLACK_NEUTRAL_BLEND_RATE")) < 0) {
			x.nbrate = 8.0;		/* Backwards compatibility value */
		} else {
			x.nbrate = atof(icg->t[0].kdata[fi]);
			if (x.nbrate < 0.05 || x.nbrate > 20.0) {
				dr->del(dr);
				error("Can't update '%s' - field 'BLACK_NEUTRAL_BLEND_RATE' has bad value %f",outname,x.nbrate);
			}
		}

		if ((fi = icg->find_kword(icg, 0, "QUALITY")) < 0) {
			dr->del(dr);
			error("Can't update '%s' - can't find field 'QUALITY'",outname);
		}
		if (quality < -50) {	/* User hasn't overridden quality */
			if (strcmp(icg->t[0].kdata[fi], "ultra low") == 0)
				quality = -3;
			else if (strcmp(icg->t[0].kdata[fi], "very low") == 0)
				quality = -2;
			else if (strcmp(icg->t[0].kdata[fi], "low") == 0)
				quality = -1;
			else if (strcmp(icg->t[0].kdata[fi], "medium") == 0)
				quality = 0;
			else if (strcmp(icg->t[0].kdata[fi], "high") == 0)
				quality = 1;
			else if (strcmp(icg->t[0].kdata[fi], "ultra high") == 0)
				quality = 2;
			else {
				dr->del(dr);
				error("Can't update '%s' - field 'QUALITY' has unrecognised value '%s'",
				        outname,icg->t[0].kdata[fi]);
			}
		}
//printf("~1 dealt with quality\n");

		/* Read in the last set of calibration curves used */
		if ((nsamp = icg->t[0].nsets) < 2) {
			dr->del(dr);
			error("Can't update '%s' - %d not enough data points in calibration curves",
			        outname,nsamp);
		}
//printf("~1 got %d points in calibration curves\n",nsamp);

		for (k = 0; k < 3; k++) {
			if ((x.rdac[k] = new_mcv()) == NULL) {
				dr->del(dr);
				error("new_mcv x.rdac[%d] failed",k);
			}
			if ((rdv[k] = malloc(sizeof(mcvco) * nsamp)) == NULL) {
				dr->del(dr);
				error("Malloc of scattered data points failed");
			}
		}
//printf("~1 allocated calibration curve objects\n");

		/* Read the current calibration curve points (usually CAL_RES of them) */
		for (k = 0; k < 4; k++) {
			char *fnames[4] = { "RGB_I", "RGB_R", "RGB_G", "RGB_B" };

			if ((si[k] = icg->find_field(icg, 0, fnames[k])) < 0) {
				dr->del(dr);
				error("Can't updata '%s' - can't find field '%s'",outname,fnames[k]);
			}
			if (icg->t[0].ftype[si[k]] != r_t) {
				dr->del(dr);
				error("Can't updata '%s' - field '%s' is wrong type",outname,fnames[k]);
			}
		}
//printf("~1 Found calibration curve fields\n");

		for (i = 0; i < nsamp; i++) {
			rdv[0][i].p =
			rdv[1][i].p =
			rdv[2][i].p =
			                *((double *)icg->t[0].fdata[i][si[0]]);
			for (k = 0; k < 3; k++) {		/* RGB */
				rdv[k][i].v = *((double *)icg->t[0].fdata[i][si[k + 1]]);
			}
			rdv[0][i].w = rdv[1][i].w = rdv[2][i].w = 1.0;
		}
//printf("~1 Read calibration curve data points\n");
		for (k = 0; k < 3; k++) {
			x.rdac[k]->fit(x.rdac[k], 0, fitord, rdv[k], nsamp, RDAC_SMOOTH);
			free (rdv[k]);
		}
//printf("~1 Fitted calibration curves\n");

		/* Read in the per channel forward model curves */
		for (k = 0; k < 3; k++) {
			char *fnames[3] = { "R_P", "G_P", "B_P" };
			double *pp;

//printf("~1 Reading device curve channel %d\n",k);
			if ((si[k] = icg->find_field(icg, 1, fnames[k])) < 0) {
				dr->del(dr);
				error("Can't updata '%s' - can't find field '%s'",outname,fnames[k]);
			}
			if (icg->t[1].ftype[si[k]] != r_t) {
				dr->del(dr);
				error("Can't updata '%s' - field '%s' is wrong type",outname,fnames[k]);
			}
			/* Create the model curves */
			if ((pp = (double *)malloc(icg->t[1].nsets * sizeof(double))) == NULL) {
				dr->del(dr);
				error("Malloc of device curve parameters");
			}
			for (i = 0; i < icg->t[1].nsets; i++)
				pp[i] = *((double *)icg->t[1].fdata[i][si[k]]);

			if ((x.dcvs[k] = new_mcv_p(pp, icg->t[1].nsets)) == NULL) {
				dr->del(dr);
				error("new_mcv x.dcvs[%d] failed",k);
			}
			free(pp);
		}
		
		icg->del(icg);
//printf("~1 read in previous settings and device model\n");
	}

	/* Be nice - check we can read the iccprofile before calibrating the display */
	if (verify != 2 && doupdate && doprofile) {
		icmFile *ic_fp;
		icc *icco;
		icmErr err = { 0, { '\000'} };

		if ((icco = new_icc(&err)) == NULL) {
			dr->del(dr);
			error("Creation of ICC object to read profile '%s' failed (0x%x, '%s')",iccoutname,err.c,err.m);
		}

		/* Open up the profile for reading */
		if ((ic_fp = new_icmFileStd_name(&err, iccoutname,"r")) == NULL) {
			dr->del(dr);
			error("Can't open file '%s' (0x%x, '%s')",iccoutname,err.c,err.m);
		}

		/* Read header etc. */
		if ((rv = icco->read(icco,ic_fp,0)) != 0) {
			dr->del(dr);
			error("Reading profile '%s' failed with %d, %s",iccoutname, rv,icco->e.m);
		}

		ic_fp->del(ic_fp);

		if (icco->find_tag(icco, icSigVideoCardGammaTag) != 0) {
			dr->del(dr);
			error("Can't find VideoCardGamma tag in file '%s': %d, %s",
			      iccoutname, icco->e.c,icco->e.m);
		}
		icco->del(icco);
	}
	/* - - - - - - - - - - - - - - - - - - - - - - - - - - - - */
	/* Convert quality level to iterations etc. */
	/* Note that final tolerance is often double the */
	/* final errth, because one more corrections is always */
	/* performed after the last reading. */
    switch (quality) {
		case -3:				/* Test value */
			isteps = 3;
			rsteps = 9;
			mxits = 1;
			mxrpts = 8;
			errthr = 2.0;
			break;
		case -2:				/* Very low */
			isteps = 10;
			rsteps = 16;
			errthr = 1.5;
			if (doupdate)
				mxits = 1;
			else
				mxits = 1;
			mxrpts = 10;
			break;
		case -1:				/* Low */
			if (verify != 2 && doprofile && !doupdate)
				isteps = 24;	/* Use more steps if we're creating a profile */
			else
				isteps = 12;
			rsteps = 32;
			errthr = 0.9;
			if (doupdate)
				mxits = 1;
			else
				mxits = 2;
			mxrpts = 10;
			break;
		default:
		case 0:					/* Medum */
			quality = 0;		/* In case it wasn't set */
			if (verify != 2 && doprofile && !doupdate)
				isteps = 32;	/* Use more steps if we're creating a profile */
			else
				isteps = 16;
			rsteps = 64;
			errthr = 0.6;
			if (doupdate)
				mxits = 1;
			else
				mxits = 3;
			mxrpts = 12;
			break;
		case 1:					/* High */
			if (verify != 2 && doprofile && !doupdate)
				isteps = 40;	/* Use more steps if we're creating a profile */
			else
				isteps = 20;
			rsteps = 96;
			errthr = 0.4;
			if (doupdate)
				mxits = 1;
			else
				mxits = 4;
			mxrpts = 16;
			break;
		case 2:					/* Ultra */
			if (verify != 2 && doprofile && !doupdate)
				isteps = 48;	/* Use more steps if we're creating a profile */
			else
				isteps = 24;
			rsteps = 128;
			errthr = 0.25;
			if (doupdate)
				mxits = 1;
			else
				mxits = 5;
			mxrpts = 24;
			break;
	}

	/* Set native white target flag in calx so that other things can play the game.. */
	if (wpx == 0.0 && wpy == 0.0 && temp == 0.0 && tbright == 0.0)
		x.nat = 1;
	else
		x.nat = 0;

	x.bkhack = bkhack;

	/* Say something about what we're doing */
	if (verb) {
		if (out_tvenc)
			printf("Using TV encoding range of (16-235)/255\n");

		if (ditype > 0)
			printf("Display type is '%s'\n",inst_distr(ditype));

		if (doupdate) {
			if (x.nat)
				printf("Target white = native white point & brightness\n");
			else
				printf("Target white = XYZ %f %f %f\n",
				       x.twh[0], x.twh[1], x.twh[2]);
		} else {
			if (wpx > 0.0 || wpy > 0.0)
				printf("Target white = xy %f %f\n",wpx,wpy);
			else if (temp > 0.0) {
				if (planckian)
					printf("Target white = %f degrees kelvin Planckian (black body) spectrum\n",temp);
				else
					printf("Target white = %f degrees kelvin Daylight spectrum\n",temp);
			} else
				printf("Target white = native white point\n");
	
			if (tbright > 0.0)
				printf("Target white brightness = %f cd/m^2\n",tbright);
			else
				printf("Target white brightness = native brightness\n");
			if (bkbright > 0.0)
				printf("Target black brightness = %f cd/m^2\n",bkbright);
			else
				printf("Target black brightness = native brightness\n");
			if (bkhack)
				printf("Black point device hack is enabled\n");
		}

		switch(x.gammat) {
			case gt_power:
				if (egamma > 0.0)
					printf("Target effective gamma = %f\n",egamma);
				else
					printf("Target advertised gamma = %f\n",gamma);
				break;
			case gt_Lab:
				printf("Target gamma = L* curve\n");
				break;
			case gt_sRGB:
				printf("Target gamma = sRGB curve\n");
				break;
			case gt_Rec709:
				printf("Target gamma = REC 709 curve\n");
				break;
			case gt_SMPTE240M:
				printf("Target gamma = SMPTE 240M curve\n");
				break;
			default:
				error("Unknown gamma type");
		}
	}

	/* - - - - - - - - - - - - - - - - - - - - - - - - - - - - */
	/* Go through the procedure of adjusting monitor controls */
	if (docontrols) {
		int rgbch = 0;			/* Got RBG Yxy ? */
		double rgbXYZ[3][3];	/* The RGB XYZ */

		/* Make sure drift comp. is off for interactive adjustment */
		dr->change_drift_comp(dr, 0, 0);

		/* Until the user is done */
		printf("\nDisplay adjustment menu:");
		for (;;) {
			int c;

			/* Print the menue of adjustments */
			printf("\nPress 1 .. 7\n");
			printf("1) Black level (CRT: Offset/Brightness)\n");
			printf("2) White point (Color temperature, R,G,B, Gain/Contrast)\n");
			printf("3) White level (CRT: Gain/Contrast, LCD: Brightness/Backlight)\n");
			printf("4) Black point (R,G,B, Offset/Brightness)\n");
			printf("5) Check all\n");
			printf("6) Measure and set ambient for viewing condition adjustment\n");
			printf("7) Continue on to calibration\n");
			printf("8) Exit\n");

			empty_con_chars();
			c = next_con_char();

			/* Black level adjustment */
			/* Due to the possibility of the channel offsets not being even, */
			/* we use the largest of the XYZ values after they have been */
			/* scaled to be even according to the white XYZ balance. */
			/* It's safer to set the black level a bit low, and then the */
			/* calibration curves can bump the low ones up. */
			if (c == '1') {
				col tcols[3] = {	/* Base set of test colors */
					{ 0.0, 0.0, 0.0 },
					{ 0.5, 0.5, 0.5 },		/* And 1% values */
					{ 1.0, 1.0, 1.0 }
				};
				int ff;
				double mgamma, tar1, dev1;
		
				printf("Doing some initial measurements\n");
				/* Do an initial set of readings to set 1% output mark */
				if ((rv = dr->read(dr, tcols, 3, 0, 0, 1, 0, instClamp, 0)) != 0) {
					dr->del(dr);
					error("display read failed with '%s'\n",disprd_err(rv));
				} 

				if (verb) {
					printf("Black = XYZ %6.4f %6.4f %6.4f\n",tcols[0].XYZ[0],
					                                         tcols[0].XYZ[1], tcols[0].XYZ[2]);
					printf("Grey  = XYZ %6.3f %6.3f %6.3f\n",tcols[1].XYZ[0],
					                                         tcols[1].XYZ[1], tcols[1].XYZ[2]);
					printf("White = XYZ %6.3f %6.3f %6.3f\n",tcols[2].XYZ[0],
					                                         tcols[2].XYZ[1], tcols[2].XYZ[2]);
				}

				/* Advertised Gamma - Gross curve shape */
				mgamma = pop_gamma(tcols[0].XYZ[1], tcols[1].XYZ[1], tcols[2].XYZ[1]);

				dev1 = pow(0.01, 1.0/mgamma);
//printf("~1 device level for 1%% output = %f\n",dev1);
				tcols[1].r = tcols[1].g = tcols[1].b = dev1;
				tar1 = 0.01 * tcols[2].XYZ[1];

				printf("\nAdjust CRT brightness to get target level. Press space when done.\n");
				printf("   Target %.2f\n",tar1);
				for (ff = 0;; ff ^= 1) {
					double dir;			/* Direction to adjust brightness */
					double sv[3], val1;				/* Scaled values */
					if ((rv = dr->read(dr, tcols+1, 1, 0, 0, 1, ' ',instClamp, 0)) != 0) {
						if (rv == 4)
							break;			/* User is done with this adjustment */
						dr->del(dr);
						error("display read failed with '%s'\n",disprd_err(rv));
					} 
					/* Scale 1% values by ratio of Y to white XYZ */
					sv[0] = tcols[1].XYZ[0] * tcols[2].XYZ[1]/tcols[2].XYZ[0];
					sv[1] = tcols[1].XYZ[1];
					sv[2] = tcols[1].XYZ[2] * tcols[2].XYZ[1]/tcols[2].XYZ[2];
//printf("~1 scaled readings = %f %f %f\n",sv[0],sv[1],sv[2]);
					val1 = sv[1];
					if (sv[0] > val1)
						val1 = sv[0];
					if (sv[2] > val1)
						val1 = sv[2];
					dir = tar1 - val1;
					if (fabs(dir) < 0.01)
						dir = 0.0;
					printf("%c%c Current %.2f  %c",
					       cr_char,
					       ff == 0 ? '/' : '\\',
					       val1,
					       dir < 0.0 ? '-' : dir > 0.0 ? '+' : '=');
					fflush(stdout);
				}
				printf("\n");

			/* White point adjustment */
			} else if (c == '2') {
				int nat = 0;		/* NZ if using native white as target */
				col tcols[1] = {	/* Base set of test colors */
					{ 1.0, 1.0, 1.0 }
				};
				int ff;
				double tYxy[3];				/* Target white chromaticities */
				icmXYZNumber tXYZ;			/* Target white as XYZ */
				double tLab[3];				/* Target white as Lab or UCS */
				double tarw;				/* Target brightness */
				double Lab[3];				/* Last measured point Lab or UCS */
				double ct = 0.0, ct_de;		/* Color temperature & delta E to white locus */
		
				printf("Doing some initial measurements\n");

				if (rgbch == 0) {	/* Figure the RGB chromaticities */
					col ccols[3] = {
						{ 1.0, 0.0, 0.0 },
						{ 0.0, 1.0, 0.0 },
						{ 0.0, 0.0, 1.0 }
					};
					if ((rv = dr->read(dr, ccols, 3, 0, 0, 1, 0, instClamp, 0)) != 0) {
						dr->del(dr);
						error("display read failed with '%s'\n",disprd_err(rv));
					}
					if (verb) {
						printf("Red   = XYZ %6.3f %6.3f %6.3f\n",ccols[0].XYZ[0],
						                       ccols[0].XYZ[1], ccols[0].XYZ[2]);
						printf("Green = XYZ %6.3f %6.3f %6.3f\n",ccols[1].XYZ[0],
						                       ccols[1].XYZ[1], ccols[1].XYZ[2]);
						printf("Blue  = XYZ %6.3f %6.3f %6.3f\n",ccols[2].XYZ[0],
						                       ccols[2].XYZ[1], ccols[2].XYZ[2]);
					}
					for (i = 0; i < 3; i++)
						icmAry2Ary(rgbXYZ[i], ccols[i].XYZ);
					rgbch = 1;
				}
				/* Do an initial set of readings to set full output mark */
				if ((rv = dr->read(dr, tcols, 1, 0, 0, 1, 0, instClamp, 0)) != 0) {
					dr->del(dr);
					error("display read failed with '%s'\n",disprd_err(rv));
				}
				if (verb) {
					printf("White = XYZ %6.3f %6.3f %6.3f\n",tcols[0].XYZ[0],
					                       tcols[0].XYZ[1], tcols[0].XYZ[2]);
				}
				
				/* Figure out the target white chromaticity */
				if (wpx > 0.0 || wpy > 0.0) {	/* xy coordinates */
					tYxy[0] = 1.0;
					tYxy[1] = wpx;
					tYxy[2] = wpy;
				} else if (temp > 0.0) {		/* Daylight color temperature */
					double XYZ[3];
					if (planckian)
						rv = icx_ill_sp2XYZ(XYZ, icxOT_default, NULL, icxIT_Ptemp, temp, NULL, 0);
					else
						rv = icx_ill_sp2XYZ(XYZ, icxOT_default, NULL, icxIT_Dtemp, temp, NULL, 0);
					if (rv != 0)
						error("Failed to compute XYZ of target color temperature %f\n",temp);
					icmXYZ2Yxy(tYxy, XYZ);
				} else {						/* Native white */
					icmXYZ2Yxy(tYxy, tcols[0].XYZ);
					nat = 1;
				}

				/* Figure out the target white brightness */
				/* Note we're not taking the device gamut into account here. */
				/* ~~~~ this seems to go a bit crazy if the brightness target is */
				/* ~~~~ beyond what the display can achieve ???? */
				/* ~~~~~ Note that dir is computed but not used !!! */
				if (tbright > 0.0) {			/* Given brightness */
					tarw = tbright;
//printf("~1 set tarw %f from tbright\n",tarw);
				} else {						/* Native/maximum brightness */
					tarw = tcols[0].XYZ[1];
//printf("~1 set tarw %f from tcols[1]\n",tarw);
				}

				if (!nat) {	/* Target is a specified white */
					printf("\nAdjust R,G & B gain to get target x,y. Press space when done.\n");
					printf("   Target Br %.2f, x %.4f , y %.4f \n",
					        tarw, tYxy[1],tYxy[2]);

				} else {	/* Target is native white */
					printf("\nAdjust R,G & B gain to desired white point. Press space when done.\n");
					/* Compute the CT and delta E to white locus of target */
					ct = comp_ct(&ct_de, NULL, planckian, dovct, obType, custObserver, tcols[0].XYZ);
					printf("  Initial Br %.2f, x %.4f , y %.4f , %c%cT %4.0fK DE 2K %4.1f\n",
					        tarw, tYxy[1],tYxy[2],
				            dovct ? 'V' : 'C', planckian ? 'C' : 'D', ct,ct_de);
				}
				for (ff = 0;; ff ^= 1) {
					double dir;			/* Direction to adjust brightness */
					double Yxy[3];		/* Yxy of current reading */
					double rgbdir[3];	/* Direction to adjust RGB */
					double rgbxdir[3];	/* Biggest to move */
					double bdir, terr;	/* Bigest direction magnitude, total error */
					int bx = 0;			/* Bigest direction axis */
					
					if ((rv = dr->read(dr, tcols, 1, 0, 0, 1, ' ', instClamp, 0)) != 0) {
						if (rv == 4)
							break;			/* User is done with this adjustment */
						dr->del(dr);
						error("display read failed with '%s'\n",disprd_err(rv));
					} 
					dir = tarw - tcols[0].XYZ[1];
					if (fabs(dir) < 0.01)
						dir = 0.0;

					icmXYZ2Yxy(Yxy, tcols[0].XYZ);

					if (!nat) {	/* Target is a specified white */
						/* Compute values we need for delta E and RGB direction */
						icmYxy2XYZ(tLab, tYxy);
						tLab[0] /= tLab[1];
						tLab[2] /= tLab[1];
						tLab[1] /= tLab[1];
						icmAry2XYZ(tXYZ, tLab);					/* Lab white reference */
						icmXYZ2Lab(&tXYZ, tLab, tLab);			/* Target Lab */
	
						icmAry2Ary(Lab, tcols[0].XYZ);
						Lab[0] /= Lab[1];
						Lab[2] /= Lab[1];
						Lab[1] /= Lab[1];
						icmXYZ2Lab(&tXYZ, Lab, Lab);			/* Current Lab */

					} else {	/* Target is native white */
						double lxyz[3];	/* Locus XYZ */
						ct = comp_ct(&ct_de, lxyz, planckian, dovct, obType, custObserver, tcols[0].XYZ);

						icmXYZ2Yxy(tYxy, lxyz);
						/* lxyz is already normalised */
						icmAry2XYZ(tXYZ, lxyz);					/* Lab white reference */
						if (dovct)
							icmXYZ2Lab(&tXYZ, tLab, lxyz);		/* Target Lab */
						else
							icmXYZ21960UCS(tLab, lxyz);			/* Target UCS */
	
						icmAry2Ary(Lab, tcols[0].XYZ);
						Lab[0] /= Lab[1];
						Lab[2] /= Lab[1];
						Lab[1] /= Lab[1];
						if (dovct)
							icmXYZ2Lab(&tXYZ, Lab, Lab);		/* Current Lab */
						else
							icmXYZ21960UCS(Lab, Lab);			/* Current UCS */
					}

					/* Compute dot products */
					bdir = 0.0;
					for (i = 0; i < 3; i++) {
						double rgbLab[3];
						
						if (dovct)
							icmXYZ2Lab(&tXYZ, rgbLab, rgbXYZ[i]);
						else
							icmXYZ21960UCS(rgbLab, rgbXYZ[i]);
						rgbdir[i] = (tLab[1] - Lab[1]) * (rgbLab[1] - Lab[1])
						          + (tLab[2] - Lab[2]) * (rgbLab[2] - Lab[2]);
						rgbxdir[i] = 0.0;
						if (fabs(rgbdir[i]) > fabs(bdir)) {
							bdir = rgbdir[i];
							bx = i;
						}
					}

					/* See how close to the target we are */
					terr = sqrt((tLab[1] - Lab[1]) * (tLab[1] - Lab[1])
					          + (tLab[2] - Lab[2]) * (tLab[2] - Lab[2]));
					if (terr < 0.1)
						rgbdir[0] = rgbdir[1] = rgbdir[2] = 0.0;
					rgbxdir[bx] = rgbdir[bx];

	
					if (!nat) {
						printf("%c%c Current Br %.2f, x %.4f%c, y %.4f%c  DE %4.1f  R%c%c G%c%c B%c%c ",
						   cr_char,
					       ff == 0 ? '/' : '\\',
					       tcols[0].XYZ[1],
					       Yxy[1],
						   Yxy[1] > tYxy[1] ? '-' : Yxy[1] < tYxy[1] ? '+' : '=',
					       Yxy[2],
						   Yxy[2] > tYxy[2] ? '-' : Yxy[2] < tYxy[2] ? '+' : '=',
					       icmCIE2K(tLab, Lab),
					       rgbdir[0] < 0.0 ? '-' : rgbdir[0] > 0.0 ? '+' : '=',
					       rgbxdir[0] < 0.0 ? '-' : rgbxdir[0] > 0.0 ? '+' : ' ',
					       rgbdir[1] < 0.0 ? '-' : rgbdir[1] > 0.0 ? '+' : '=',
					       rgbxdir[1] < 0.0 ? '-' : rgbxdir[1] > 0.0 ? '+' : ' ',
					       rgbdir[2] < 0.0 ? '-' : rgbdir[2] > 0.0 ? '+' : '=',
					       rgbxdir[2] < 0.0 ? '-' : rgbxdir[2] > 0.0 ? '+' : ' ');
					} else {
						printf("%c%c Current Br %.2f, x %.4f%c, y %.4f%c  %c%cT %4.0fK DE 2K %4.1f  R%c%c G%c%c B%c%c ",
						   cr_char,
					       ff == 0 ? '/' : '\\',
					       tcols[0].XYZ[1],
					       Yxy[1],
						   Yxy[1] > tYxy[1] ? '-': Yxy[1] < tYxy[1] ? '+' : '=',
					       Yxy[2],
						   Yxy[2] > tYxy[2] ? '-': Yxy[2] < tYxy[2] ? '+' : '=',
				           dovct ? 'V' : 'C', planckian ? 'C' : 'D', ct,ct_de,
					       rgbdir[0] < 0.0 ? '-' : rgbdir[0] > 0.0 ? '+' : '=',
					       rgbxdir[0] < 0.0 ? '-' : rgbxdir[0] > 0.0 ? '+' : ' ',
					       rgbdir[1] < 0.0 ? '-' : rgbdir[1] > 0.0 ? '+' : '=',
					       rgbxdir[1] < 0.0 ? '-' : rgbxdir[1] > 0.0 ? '+' : ' ',
					       rgbdir[2] < 0.0 ? '-' : rgbdir[2] > 0.0 ? '+' : '=',
					       rgbxdir[2] < 0.0 ? '-' : rgbxdir[2] > 0.0 ? '+' : ' ');
					}
					fflush(stdout);
				}
				printf("\n");

			/* White level adjustment */
			} else if (c == '3') {
				col tcols[1] = {	/* Base set of test colors */
					{ 1.0, 1.0, 1.0 }
				};
				int ff;
				double tarw;
		
				printf("Doing some initial measurements\n");
				/* Do an initial set of readings to set full output mark */
				if ((rv = dr->read(dr, tcols, 1, 0, 0, 1, 0, instClamp, 0)) != 0) {
					dr->del(dr);
					error("display read failed with '%s'\n",disprd_err(rv));
				}
				if (verb) {
					printf("White = XYZ %6.3f %6.3f %6.3f\n",tcols[0].XYZ[0],
					                       tcols[0].XYZ[1], tcols[0].XYZ[2]);
				}
				
				/* Figure out the target white brightness */
				/* Note we're not taking the device gamut into account here */
				if (tbright > 0.0)			/* Given brightness */
					tarw = tbright;
				else						/* Native/maximum brightness */
					tarw = tcols[0].XYZ[1];

				if (tbright > 0.0) {
					printf("\nAdjust CRT Contrast or LCD Brightness to get target level. Press space when done.\n");
					printf("   Target %.2f\n", tarw);
				} else {
					printf("\nAdjust CRT Contrast or LCD Brightness to desired level. Press space when done.\n");
					printf("  Initial %.2f\n", tarw);
				}
				for (ff = 0;; ff ^= 1) {
					double dir;			/* Direction to adjust brightness */
					
					if ((rv = dr->read(dr, tcols, 1, 0, 0, 1, ' ', instClamp, 0)) != 0) {
						if (rv == 4)
							break;			/* User is done with this adjustment */
						dr->del(dr);
						error("display read failed with '%s'\n",disprd_err(rv));
					} 
					dir = tarw - tcols[0].XYZ[1];
					if (fabs(dir) < 0.01)
						dir = 0.0;

					if (tbright > 0.0)
						printf("%c%c Current %.2f  %c",
						   cr_char,
					       ff == 0 ? '/' : '\\',
					       tcols[0].XYZ[1],
					       dir < 0.0 ? '-' : dir > 0.0 ? '+' : '=');
					else
						printf("%c%c Current %.2f   ",
						   cr_char,
					       ff == 0 ? '/' : '\\',
					       tcols[0].XYZ[1]);
					fflush(stdout);
				}
				printf("\n");

			/* Black point adjustment */
			} else if (c == '4') {
				col tcols[3] = {	/* Base set of test colors */
					{ 0.0, 0.0, 0.0 },
					{ 0.5, 0.5, 0.5 },		/* And 1% values */
					{ 1.0, 1.0, 1.0 }
				};
				int ff;
				double tYxy[3];				/* Target white chromaticities */
				icmXYZNumber tXYZ;			/* Target white as XYZ */
				double tLab[3];				/* Target white as Lab or UCS */
				double mgamma, tar1, dev1;
				double Lab[3];				/* Last measured point Lab */
		
				printf("Doing some initial measurements\n");

				if (rgbch == 0) {	/* Figure the RGB chromaticities */
					col ccols[3] = {
						{ 1.0, 0.0, 0.0 },
						{ 0.0, 1.0, 0.0 },
						{ 0.0, 0.0, 1.0 }
					};
					if ((rv = dr->read(dr, ccols, 3, 0, 0, 1, 0, instClamp, 0)) != 0) {
						dr->del(dr);
						error("display read failed with '%s'\n",disprd_err(rv));
					}
					if (verb) {
						printf("Red   = XYZ %6.3f %6.3f %6.3f\n",ccols[0].XYZ[0],
						                       ccols[0].XYZ[1], ccols[0].XYZ[2]);
						printf("Green = XYZ %6.3f %6.3f %6.3f\n",ccols[1].XYZ[0],
						                       ccols[1].XYZ[1], ccols[1].XYZ[2]);
						printf("Blue  = XYZ %6.3f %6.3f %6.3f\n",ccols[2].XYZ[0],
						                       ccols[2].XYZ[1], ccols[2].XYZ[2]);
					}
					for (i = 0; i < 3; i++)
						icmAry2Ary(rgbXYZ[i], ccols[i].XYZ);
					rgbch = 1;
				}
				/* Do an initial set of readings to set 1% output mark */
				if ((rv = dr->read(dr, tcols, 3, 0, 0, 1, 0, instClamp, 0)) != 0) {
					dr->del(dr);
					error("display read failed with '%s'\n",disprd_err(rv));
				}
				if (verb) {
					printf("Black = XYZ %6.4f %6.4f %6.4f\n",tcols[0].XYZ[0],
					                                         tcols[0].XYZ[1], tcols[0].XYZ[2]);
					printf("Grey  = XYZ %6.3f %6.3f %6.3f\n",tcols[1].XYZ[0],
					                                         tcols[1].XYZ[1], tcols[1].XYZ[2]);
					printf("White = XYZ %6.3f %6.3f %6.3f\n",tcols[2].XYZ[0],
					                                         tcols[2].XYZ[1], tcols[2].XYZ[2]);
				}
				
				/* Advertised Gamma - Gross curve shape */
				mgamma = pop_gamma(tcols[0].XYZ[1], tcols[1].XYZ[1], tcols[2].XYZ[1]);

				dev1 = pow(0.01, 1.0/mgamma);
				tcols[1].r = tcols[1].g = tcols[1].b = dev1;
				tar1 = 0.01 * tcols[2].XYZ[1];

				/* Figure out the target white chromaticity */
				if (wpx > 0.0 || wpy > 0.0) {	/* xy coordinates */
					tYxy[0] = 1.0;
					tYxy[1] = wpx;
					tYxy[2] = wpy;
		
				} else if (temp > 0.0) {		/* Daylight color temperature */
					double XYZ[3];
					if (planckian)
						rv = icx_ill_sp2XYZ(XYZ, icxOT_default, NULL, icxIT_Ptemp, temp, NULL, 0);
					else
						rv = icx_ill_sp2XYZ(XYZ, icxOT_default, NULL, icxIT_Dtemp, temp, NULL, 0);
					if (rv != 0)
						error("Failed to compute XYZ of target color temperature %f\n",temp);
					icmXYZ2Yxy(tYxy, XYZ);
				} else {						/* Native white */
					icmXYZ2Yxy(tYxy, tcols[2].XYZ);
				}

				printf("\nAdjust R,G & B offsets to get target x,y. Press space when done.\n");
				printf("   Target Br %.4f, x %.4f , y %.4f \n", tar1, tYxy[1],tYxy[2]);
				for (ff = 0;; ff ^= 1) {
					double dir;			/* Direction to adjust brightness */
					double sv[3], val1;				/* Scaled values */
					double Yxy[3];		/* Yxy of current reading */
					double rgbdir[3];	/* Direction to adjust RGB */
					double rgbxdir[3];	/* Biggest to move */
					double bdir, terr;
					int bx = 0;
					
					if ((rv = dr->read(dr, tcols+1, 1, 0, 0, 1, ' ', instClamp, 0)) != 0) {
						if (rv == 4)
							break;			/* User is done with this adjustment */
						dr->del(dr);
						error("display read failed with '%s'\n",disprd_err(rv));
					} 
				
					/* Scale 1% values by ratio of Y to white XYZ */
					sv[0] = tcols[1].XYZ[0] * tcols[2].XYZ[1]/tcols[2].XYZ[0];
					sv[1] = tcols[1].XYZ[1];
					sv[2] = tcols[1].XYZ[2] * tcols[2].XYZ[1]/tcols[2].XYZ[2];
					val1 = sv[1];
					if (sv[0] > val1)
						val1 = sv[0];
					if (sv[2] > val1)
						val1 = sv[2];

					/* Compute 1% direction */
					dir = tar1 - val1;
					if (fabs(dir) < 0.01)
						dir = 0.0;

					/* Compute numbers for black point error and direction */
					icmYxy2XYZ(tLab, tYxy);
					tLab[0] /= tLab[1];
					tLab[2] /= tLab[1];
					tLab[1] /= tLab[1];
					icmAry2XYZ(tXYZ, tLab);				/* Lab white reference */
					icmXYZ2Lab(&tXYZ, tLab, tLab);

					icmXYZ2Yxy(Yxy, tcols[1].XYZ);
					icmAry2Ary(Lab, tcols[1].XYZ);
					Lab[0] /= Lab[1];
					Lab[2] /= Lab[1];
					Lab[1] /= Lab[1];
					icmXYZ2Lab(&tXYZ, Lab, Lab);
	
					/* Compute dot products */
					bdir = 0.0;
					for (i = 0; i < 3; i++) {
						double rgbLab[3];
						
						icmXYZ2Lab(&tXYZ, rgbLab, rgbXYZ[i]);
						rgbdir[i] = (tLab[1] - Lab[1]) * (rgbLab[1] - Lab[1])
						          + (tLab[2] - Lab[2]) * (rgbLab[2] - Lab[2]);
						rgbxdir[i] = 0.0;
						if (fabs(rgbdir[i]) > fabs(bdir)) {
							bdir = rgbdir[i];
							bx = i;
						}
					}

					/* See how close to the target we are */
					terr = sqrt((tLab[1] - Lab[1]) * (tLab[1] - Lab[1])
					          + (tLab[2] - Lab[2]) * (tLab[2] - Lab[2]));
					if (terr < 0.1)
						rgbdir[0] = rgbdir[1] = rgbdir[2] = 0.0;
					rgbxdir[bx] = rgbdir[bx];

			 		printf("%c%c Current Br %.4f, x %.4f%c, y %.4f%c  DE %4.1f  R%c%c G%c%c B%c%c ",
					       cr_char,
					       ff == 0 ? '/' : '\\',
					       val1,
					       Yxy[1],
						   Yxy[1] > tYxy[1] ? '-': Yxy[1] < tYxy[1] ? '+' : '=',
					       Yxy[2],
						   Yxy[2] > tYxy[2] ? '-': Yxy[2] < tYxy[2] ? '+' : '=',
					       icmCIE2K(tLab, Lab),
					       rgbdir[0] < 0.0 ? '-' : rgbdir[0] > 0.0 ? '+' : '=',
					       rgbxdir[0] < 0.0 ? '-' : rgbxdir[0] > 0.0 ? '+' : ' ',
					       rgbdir[1] < 0.0 ? '-' : rgbdir[1] > 0.0 ? '+' : '=',
					       rgbxdir[1] < 0.0 ? '-' : rgbxdir[1] > 0.0 ? '+' : ' ',
					       rgbdir[2] < 0.0 ? '-' : rgbdir[2] > 0.0 ? '+' : '=',
					       rgbxdir[2] < 0.0 ? '-' : rgbxdir[2] > 0.0 ? '+' : ' ');
					fflush(stdout);
				}
				printf("\n");

			/* Report on how well we current meet the targets */
			} else if (c == '5') {
				int nat = 0;		/* NZ if using native white as target */
				col tcols[4] = {	/* Set of test colors */
					{ 0.0, 0.0, 0.0 },
					{ 0.5, 0.5, 0.5 },
					{ 1.0, 1.0, 1.0 },
					{ 0.0, 0.0, 0.0 }	/* 1% test value */
				};
				double tYxy[3];				/* Target white chromaticities */
				double sv[3], val1;			/* Scaled values */
				double mgamma, tarw, tar1, dev1, tarh;
				double gooff;				/* Aproximate output offset needed */
				icmXYZNumber tXYZ;
				double tLab[3], wYxy[3], wLab[3], bYxy[3], bLab[3];
				double ct, ct_de;			/* Color temperature & delta E to white locus */
		
				printf("Doing check measurements\n");

				/* Do an initial set of readings to set 1% output mark */
				if ((rv = dr->read(dr, tcols, 3, 0, 0, 1, 0, instClamp, 0)) != 0) {
					dr->del(dr);
					error("display read failed with '%s'\n",disprd_err(rv));
				}
				if (verb) {
					printf("Black = XYZ %6.4f %6.4f %6.4f\n",tcols[0].XYZ[0],
					                                         tcols[0].XYZ[1], tcols[0].XYZ[2]);
					printf("Grey  = XYZ %6.3f %6.3f %6.3f\n",tcols[1].XYZ[0],
					                                         tcols[1].XYZ[1], tcols[1].XYZ[2]);
					printf("White = XYZ %6.3f %6.3f %6.3f\n",tcols[2].XYZ[0],
					                                         tcols[2].XYZ[1], tcols[2].XYZ[2]);
				}
				
				/* Approximate Gamma - use the gross curve shape for robustness */
				mgamma = pop_gamma(tcols[0].XYZ[1], tcols[1].XYZ[1], tcols[2].XYZ[1]);

				dev1 = pow(0.01, 1.0/mgamma);
				tcols[3].r = tcols[3].g = tcols[3].b = dev1;
				tar1 = 0.01 * tcols[2].XYZ[1];

				/* Read the 1% value */
				if ((rv = dr->read(dr, tcols+3, 1, 0, 0, 1, 0, instClamp, 0)) != 0) {
					dr->del(dr);
					error("display read failed with '%s'\n",disprd_err(rv));
				}
				if (verb) {
					printf("1%%    = XYZ %6.3f %6.3f %6.3f\n",tcols[3].XYZ[0],
					                                         tcols[3].XYZ[1], tcols[3].XYZ[2]);
				}
				
				/* Scale 1% values by ratio of Y to white XYZ */
				/* (Note we're assuming -k1 here, which may not be true...) */
				sv[0] = tcols[3].XYZ[0] * tcols[2].XYZ[1]/tcols[2].XYZ[0];
				sv[1] = tcols[3].XYZ[1];
				sv[2] = tcols[3].XYZ[2] * tcols[2].XYZ[1]/tcols[2].XYZ[2];
				val1 = sv[1];
				if (sv[0] > val1)
					val1 = sv[0];
				if (sv[2] > val1)
					val1 = sv[2];

				/* Figure out the target white brightness */
				/* Note we're not taking the device gamut into account here */
				if (tbright > 0.0)			/* Given brightness */
					tarw = tbright;
				else						/* Native/maximum brightness */
					tarw = tcols[2].XYZ[1];

				/* Figure out the target white chromaticity */
				if (wpx > 0.0 || wpy > 0.0) {	/* xy coordinates */
					tYxy[0] = 1.0;
					tYxy[1] = wpx;
					tYxy[2] = wpy;
		
				} else if (temp > 0.0) {		/* Daylight color temperature */
					double XYZ[3];
					if (planckian)
						rv = icx_ill_sp2XYZ(XYZ, icxOT_default, NULL, icxIT_Ptemp, temp, NULL, 0);
					else
						rv = icx_ill_sp2XYZ(XYZ, icxOT_default, NULL, icxIT_Dtemp, temp, NULL, 0);
					if (rv != 0)
						error("Failed to compute XYZ of target color temperature %f\n",temp);
					icmXYZ2Yxy(tYxy, XYZ);
				} else {						/* Native white */
					icmXYZ2Yxy(tYxy, tcols[2].XYZ);
					nat = 1;
				}

				/* Figure out the target 50% device output value */
				gooff = tcols[0].XYZ[1]/tcols[2].XYZ[1];	/* Aprox. normed black output offset */

				/* Use tech_gamma() to do the hard work */
				tarh = tech_gamma(&x, NULL, NULL, NULL, egamma, gamma, gooff);

				/* Convert from Y fraction to absolute Y */
				tarh = tarh * tcols[2].XYZ[1];

				/* Compute various white point values */
				icmYxy2XYZ(tLab, tYxy);
				tLab[0] /= tLab[1];
				tLab[2] /= tLab[1];
				tLab[1] /= tLab[1];
				icmAry2XYZ(tXYZ, tLab);
				icmXYZ2Lab(&tXYZ, tLab, tLab);

				icmXYZ2Yxy(wYxy, tcols[2].XYZ);
				icmAry2Ary(wLab, tcols[2].XYZ);
				wLab[0] /= wLab[1];
				wLab[2] /= wLab[1];
				wLab[1] /= wLab[1];
				icmXYZ2Lab(&tXYZ, wLab, wLab);

				icmXYZ2Yxy(bYxy, tcols[3].XYZ);
				icmAry2Ary(bLab, tcols[3].XYZ);
				bLab[0] /= bLab[1];
				bLab[2] /= bLab[1];
				bLab[1] /= bLab[1];
				icmXYZ2Lab(&tXYZ, bLab, bLab);

				/* And color temperature */
				ct = comp_ct(&ct_de, NULL, planckian, dovct, obType, custObserver, tcols[2].XYZ);

				printf("\n");

				if (tbright > 0.0)			/* Given brightness */
					printf("  Target Brightness = %.3f, Current = %.3f, error = % .1f%%\n",
				       tarw, tcols[2].XYZ[1], 
				       100.0 * (tcols[2].XYZ[1] - tarw)/tarw);
				else
					printf("  Current Brightness = %.2f\n", tcols[2].XYZ[1]);
				
				printf("  Target 50%% Level  = %.3f, Current = %.3f (Aprox. Gamma %.2f), error = % .1f%%\n",
				       tarh, tcols[1].XYZ[1], 
				       mgamma,
				       100.0 * (tcols[1].XYZ[1] - tarh)/tarw);
				
				printf("  Target Near Black = %.4f, Current = %.4f, error = % .1f%%\n",
				       tar1, val1, 
				       100.0 * (val1 - tar1)/tarw);

				if (!nat)
					printf("  Target white = x %.4f, y %.4f, Current = x %.4f, y %.4f, error = %5.2f DE\n",
				       tYxy[1], tYxy[2], wYxy[1], wYxy[2], icmCIE2K(tLab, wLab));
				else
					printf("  Current white = x %.4f, y %.4f, %c%cT %4.0fK DE 2K %4.1f\n",
				       wYxy[1], wYxy[2], dovct ? 'V' : 'C', planckian ? 'C' : 'D', ct,ct_de);

				printf("  Target black = x %.4f, y %.4f, Current = x %.4f, y %.4f, error = %5.2f DE\n",
				       tYxy[1], tYxy[2], bYxy[1], bYxy[2], icmCIE2K(tLab, bLab));

			
			/* Measure and set ambient for viewing condition adjustment */
			} else if (c == '6') {
				if ((rv = dr->meas_ambient(dr, &ambientl, 1)) != 0) {
					if (rv == 8) {
						printf("Instrument doesn't have an ambient reading capability\n");
					} else {
						dr->del(dr);
						error("ambient measure failed with '%s'\n",disprd_err(rv));
					}
				} else {
					printf("Measured ambient level = %.1f Lux\n",ambientl);
				}

			} else if (c == '7') {
				if (!verb) {		/* Tell user command has been accepted */
					if (verify == 2)
						printf("Commencing display verification\n");
					else
						printf("Commencing display calibration\n");
				}
				break;
			} else if (c == '8' || c == 0x03 || c == 0x1b) {
				printf("Exiting\n");
				dr->del(dr);
				exit(0);
			}
		}

		/* Make sure drift comp. is set to the command line options */
		dr->change_drift_comp(dr, bdrift, wdrift);
	}

	/* - - - - - - - - - - - - - - - - - - - - - - - - - - - - */

	/* Take a small number of readings, and compute basic */
	/* informations such as black & white, white target, */
	/* aproximate matrix based display forward and reverse model. */
	/* If bkcorrect is auto, determine a level. */

	/* Read the base test set */
	{
		double mrd[3];		/* Number for matrix */
		double mgn[3];
		double mbl[3];
		double mwh[3];
		ramdac *or = NULL;

		col base[9] = {	/* Base set of test colors */
			{ 0.0, 0.0, 0.0 },		/* 0 - Black */
			{ 1.0, 0.0, 0.0 },		/* 1 - Red */
			{ 1.0, 1.0, 1.0 },		/* 2 - White */
			{ 0.0, 0.0, 0.0 },		/* 3 - Black */
			{ 0.0, 1.0, 0.0 },		/* 4 - Green */
			{ 0.0, 0.0, 0.0 },		/* 5 - Black */
			{ 0.0, 0.0, 1.0 },		/* 6 - Blue */
			{ 1.0, 1.0, 1.0 },		/* 7 - White */
			{ 0.0, 0.0, 0.0 }		/* 8 - Black */
		};
		int ix_k1 = 0;
		int ix_r  = 1;
		int ix_w1 = 2;
		int ix_k2 = 3;
		int ix_g  = 4;
		int ix_k3 = 5;
		int ix_b  = 6;
		int ix_w2 = 7;
		int ix_k4 = 8;

		if (verb) {
			if (verify == 2)
				printf("Commencing display verification\n");
			else
				printf("Commencing display calibration\n");
		}

		/* Switch to native for this, so the black calc is realistic. */
		/* (Should we really get black aim from previous .cal though ???) */
		if (verify == 2) {
			if (fake)
				error("Can't verify against current curves using fake device");

			if ((or = dr->dw->get_ramdac(dr->dw)) != NULL) {
				ramdac *r;
				if (verb) printf("Switching to native response for base measurements\n");
				r = or->clone(or);
				r->setlin(r);
				dr->dw->set_ramdac(dr->dw, r, 0);
				r->del(r);
			}
		}

		/* Read the patches without clamping */
		if ((rv = dr->read(dr, base, 9, 1, 9, 1, 0, instNoClamp, 0)) != 0) {
			dr->del(dr);
			error("display read failed with '%s'\n",disprd_err(rv));
		} 

		/* Restore the cal we're verifying */
		if (verify == 2 && or != NULL) {
			if (verb) printf("Switching back to calibration being verified\n");
			dr->dw->set_ramdac(dr->dw, or, 0);
			or->del(or);
		}

		if (base[ix_k1].XYZ_v == 0) {
			dr->del(dr);
			error("Failed to get an XYZ value from the instrument!\n");
		}

		if (verb >= 3) {
			for (i = 0; i < 9; i++)
				printf("Meas %d XYZ = %f %f %f\n",i,base[i].XYZ[0], base[i].XYZ[1], base[i].XYZ[2]);
		}

		/* Average black relative from 4 readings */
		x.bk[0] = 0.25 * (base[ix_k1].XYZ[0] + base[ix_k2].XYZ[0]
		                + base[ix_k3].XYZ[0] + base[ix_k4].XYZ[0]);
		x.bk[1] = 0.25 * (base[ix_k1].XYZ[1] + base[ix_k2].XYZ[1]
		                + base[ix_k3].XYZ[1] + base[ix_k4].XYZ[1]);
		x.bk[2] = 0.25 * (base[ix_k1].XYZ[2] + base[ix_k2].XYZ[2]
		                + base[ix_k3].XYZ[2] + base[ix_k4].XYZ[2]);
		icmClamp3(x.bk, x.bk);	/* And clamp them */

		/* Average white reading from 2 readings */
		base[ix_w1].XYZ[0] = 0.5 * (base[ix_w1].XYZ[0] + base[ix_w2].XYZ[0]);
		base[ix_w1].XYZ[1] = 0.5 * (base[ix_w1].XYZ[1] + base[ix_w2].XYZ[1]);
		base[ix_w1].XYZ[2] = 0.5 * (base[ix_w1].XYZ[2] + base[ix_w2].XYZ[2]);
	
		for (i = 0; i < 9; i++)
			icmClamp3(base[i].XYZ, base[i].XYZ);

		/* Copy other readings into place */
		dispLum = base[ix_w1].XYZ[1];				/* White Y */
		icmAry2Ary(x.wh, base[ix_w1].XYZ);
		icmAry2XYZ(x.twN, x.wh);	/* Use this as Lab reference white until we establish target */

		icmAry2Ary(mrd, base[ix_r].XYZ);
		icmAry2Ary(mgn, base[ix_g].XYZ);
		icmAry2Ary(mbl, base[ix_b].XYZ);
		icmAry2Ary(mwh, base[ix_w1].XYZ);

		if (verb) {
			printf("Black = XYZ %6.4f %6.4f %6.4f\n",x.bk[0],x.bk[1],x.bk[2]);
			printf("Red   = XYZ %6.3f %6.3f %6.3f\n",base[ix_r].XYZ[0], base[ix_r].XYZ[1], base[ix_r].XYZ[2]);
			printf("Green = XYZ %6.3f %6.3f %6.3f\n",base[ix_g].XYZ[0], base[ix_g].XYZ[1], base[ix_g].XYZ[2]);
			printf("Blue  = XYZ %6.3f %6.3f %6.3f\n",base[ix_b].XYZ[0], base[ix_b].XYZ[1], base[ix_b].XYZ[2]);
			printf("White = XYZ %6.3f %6.3f %6.3f\n",base[ix_w1].XYZ[0], base[ix_w1].XYZ[1], base[ix_w1].XYZ[2]);
		}

		/* Setup forward matrix */
		if (icmRGBXYZprim2matrix(mrd, mgn, mbl, mwh, x.fm)) {
			dr->del(dr);
			error("Aprox. fwd matrix unexpectedly singular\n");
		}

#ifdef DEBUG
		if (verb) {
			printf("Forward matrix is:\n");
			printf("%f %f %f\n", x.fm[0][0], x.fm[0][1], x.fm[0][2]);
			printf("%f %f %f\n", x.fm[1][0], x.fm[1][1], x.fm[1][2]);
			printf("%f %f %f\n", x.fm[2][0], x.fm[2][1], x.fm[2][2]);
		}
#endif

		/* Compute bwd matrix */
		if (icmInverse3x3(x.bm, x.fm)) {
			dr->del(dr);
			error("Inverting aprox. fwd matrix failed");
		}

		/* Decide on the level of black correction. */
		if (bkcorrect < 0.0) {
			double rat;

			/* rat is 0 for displays with a good black, */
			/* and 1 for displays with a bad black level. */
			/* (Not sure if this should be scaled by the white, */
			/*  making it contrast ratio sensitive?) */
			rat = (x.bk[1] - 0.02)/(0.3 - 0.02);
			if (rat < 0.0)
				rat = 0.0;
			else if (rat > 1.0)
				rat = 1.0;
			/* Make transition more perceptual */
			rat = sqrt(rat);
			bkcorrect = 1.0 - rat;
			if (verb)
				printf("Automatic black point hue correction level = %1.2f\n", bkcorrect);
		}
	}

	/* Now do some more readings, to compute the basic per channel */
	/* transfer characteristics, and then a device model. */
	if (verify != 2 && !doupdate) {
		col *cols;				/* Read 4 x isteps patches from display */
		sxyz *asrgb[4];			/* samples for r, g, b & w */

		if ((cols = (col *)malloc(isteps * 4 * sizeof(col))) == NULL) {
			dr->del(dr);
			error("Malloc of array of readings failed");
		}
		for (j = 0; j < 4; j++) {
			if ((asrgb[j] = (sxyz *)malloc(isteps * sizeof(sxyz))) == NULL) {
				free(cols);
				dr->del(dr);
				error("Malloc of array of readings failed");
			}
		}

		/* Set the device colors to read */
		for (i = 0; i < isteps; i++) {
			double vv;

#if defined(__APPLE__) && defined(__POWERPC__)
			gcc_bug_fix(i);
#endif
			vv = i/(isteps - 1.0);
			vv = pow(vv, MOD_DIST_POW);
			for (j = 0; j < 4; j++) {
				cols[i * 4 + j].r = cols[i * 4 + j].g = cols[i * 4 + j].b = 0.0;
				if (j == 0)
					cols[i * 4 + j].r = vv;
				else if (j == 1)
					cols[i * 4 + j].g = vv;
				else if (j == 2)
					cols[i * 4 + j].b = vv;
				else
					cols[i * 4 + j].r = cols[i * 4 + j].g = cols[i * 4 + j].b = vv;
			}
		}

		/* Read the patches */
		if ((rv = dr->read(dr, cols, isteps * 4, 1, isteps * 4, 1, 0, instClamp, 0)) != 0) {
			free(cols); free(asrgb[0]); free(asrgb[1]); free(asrgb[2]); free(asrgb[3]);
			dr->del(dr);
			error("display read failed with '%s'\n",disprd_err(rv));
		} 

		/* Transfer readings to asrgb[] */
		for (i = 0; i < isteps; i++) {
			double vv =  cols[i * 4 + 0].r;
			for (j = 0; j < 4; j++) {
//printf("~1 R = %f, G = %f, B = %f, XYZ = %f %f %f\n",
//cols[i * 4 + j].r, cols[i * 4 + j].g, cols[i * 4 + j].b, cols[i * 4 + j].XYZ[0], cols[i * 4 + j].XYZ[1], cols[i * 4 + j].XYZ[2]);
				asrgb[j][i].v = vv;
				asrgb[j][i].xyz[0] = cols[i * 4 + j].XYZ[0];
				asrgb[j][i].xyz[1] = cols[i * 4 + j].XYZ[1];
				asrgb[j][i].xyz[2] = cols[i * 4 + j].XYZ[2];
			}
		}

		/* Convert RGB channel samples to curves */
		{
			mcvco *sdv;		/* Points used to create cvs[], RGB */
			double blrgb[3];
			double *op;		/* Parameters to optimise */
			double *sa;		/* Search area */
			double re;		/* Residual error */

			/* Transform measured black back to linearised RGB values */
			icmMulBy3x3(blrgb, x.bm, x.bk);
//printf("~1 model black should be %f %f %f\n", x.bk[0], x.bk[1], x.bk[2]);
//printf("~1 linearised RGB should be %f %f %f\n", blrgb[0], blrgb[1], blrgb[2]);

			if ((sdv = malloc(sizeof(mcvco) * isteps)) == NULL) {
				free(cols); free(asrgb[0]); free(asrgb[1]); free(asrgb[2]); free(asrgb[3]);
				dr->del(dr);
				error("Malloc of scattered data points failed");
			}
			for (k = 0; k < 3; k++) {	/* Create the model curves */
				for (i = 0; i < isteps; i++) {
					sdv[i].p = asrgb[k][i].v;
					sdv[i].v = ICMNORM3(asrgb[k][i].xyz);
					sdv[i].w = 1.0;
//printf("~1 chan %d, entry %d, p = %f, v = %f from XYZ %f %f %f\n",
//k,i,x.sdv[k][i].p,x.sdv[k][i].v, asrgb[k][i].xyz[0], asrgb[k][i].xyz[1], asrgb[k][i].xyz[2]);
				}
				if ((x.dcvs[k] = new_mcv()) == NULL) {
					free(cols); free(asrgb[0]); free(asrgb[1]); free(asrgb[2]); free(asrgb[3]);
					dr->del(dr);
					error("new_mcv x.dcvs[%d] failed",k);
				}
				x.dcvs[k]->fit(x.dcvs[k], 0, fitord, sdv, isteps, 5.0);

				/* Scale the whole curve so the output is scaled to 1.0 */
				x.dcvs[k]->force_scale(x.dcvs[k], 1.0);

				/* Force curves to produce this lrgb for 0.0 */
				x.dcvs[k]->force_0(x.dcvs[k], blrgb[k]);
			}
			free(sdv);

#ifdef OPTIMIZE_MODEL	
			/* Setup list of reference points ready for optimisation */
			x.nrp = 4 * isteps;
			if ((x.rp = (optref *)malloc(sizeof(optref) * x.nrp)) == NULL) {
				free(cols); free(asrgb[0]); free(asrgb[1]); free(asrgb[2]); free(asrgb[3]);
				dr->del(dr);
				error("Malloc of measurement reference points failed");
			}
			for (k = 0; k < 4; k++) {
				for (i = 0; i < isteps; i++) {
					int ii = k * isteps + i;
					double v[3];
				
					v[0] = v[1] = v[2] = 0.0;
					if (k == 0)
						v[k] = asrgb[k][i].v;
					else if (k == 1)
						v[k] = asrgb[k][i].v;
					else if (k == 2)
						v[k] = asrgb[k][i].v;
					else
						v[0] = v[1] = v[2] = asrgb[k][i].v;
					icmAry2Ary(x.rp[ii].dev, v);
					icmXYZ2Lab(&x.twN, x.rp[ii].lab, asrgb[k][i].xyz);
					if (k == 3)		/* White */
						x.rp[ii].w = 0.5;
					else
						x.rp[ii].w = 0.16667;
				}
			}

			/* Get parameters and setup for optimisation */
			op = dev_get_params(&x);
			if ((sa = malloc(x.np * sizeof(double))) == NULL) {
				free(cols); free(asrgb[0]); free(asrgb[1]); free(asrgb[2]); free(asrgb[3]);
				dr->del(dr);
				error("Malloc of scattered data points failed");
			}
			
			for (i = 0; i < x.np; i++)
				sa[i] = 0.1;

			/* Do optimisation */
#ifdef NEVER
			if (powell(&re, x.np, op, sa, 1e-5, 3000, dev_opt_func, (void *)&x, NULL, NULL) != 0) {
				free(cols); free(asrgb[0]); free(asrgb[1]); free(asrgb[2]); free(asrgb[3]);
				dr->del(dr);
				error("Model powell failed, re = %f",re);
			}
#else
			if (conjgrad(&re, x.np, op, sa, 1e-5, 3000,
			                         dev_opt_func, dev_dopt_func, (void *)&x, NULL, NULL) != 0) {
				if (re > 1e-2) {
					free(cols); free(asrgb[0]); free(asrgb[1]); free(asrgb[2]); free(asrgb[3]);
					dr->del(dr);
					error("Model conjgrad failed, residual error = %f",re);
				} else
					warning("Model conjgrad failed, residual error = %f",re);
			}
#endif

			/* Put optimised parameters in place */
			dev_put_params(&x, op);

			free(x.rp);
			x.rp = NULL;
			x.nrp = 0;
			free(x.dtin_iv);		/* Free temporary arrays */
			x.dtin_iv = NULL;
			free(sa);
			free(op);
#endif /* OPTIMIZE_MODEL */	
		}

#ifdef DEBUG_PLOT
		/* Plot the current calc curves */
		{
			#define	XRES 256
			double xx[XRES];
			double yy[3][XRES];
			double xyz[3];
			for (i = 0; i < XRES; i++) {
				xx[i] = i/(XRES-1.0);
				for (j = 0; j < 3; j++)
					yy[j][i] = x.dcvs[j]->interp(x.dcvs[j], xx[i]);
			}
			printf("Channel curves\n");
			do_plot(xx,yy[0],yy[1],yy[2],XRES);
			#undef XRES
		}
#endif

		/* We're done with cols[] and asrgb[] */
		free(cols); free(asrgb[0]); free(asrgb[1]); free(asrgb[2]); free(asrgb[3]);
	}

#ifdef CHECK_MODEL
	/* Check how well our fwd model agrees with the device */
	if (verify != 2) {
		col set[3];					/* Variable to read up to 3 values from the display */
		int nn = 27;
		double alab[3], mxyz[3], mlab[3];	/* Actual and model Lab */
		double mnerr;		/* Maximum neutral error */
		double mnv;			/* Value where maximum error is */
		double anerr;		/* Average neutral error */

		mnerr = anerr = 0.0;
		/* !!! Should change this to single batch to work better with drift comp. !!! */
		for (i = 0; i < (nn + 3); i++) {
			double vv, v[3];
			double de;

			if (i < nn) {
				vv = i/(nn - 1.0);
				vv = pow(vv, CHECK_DIST_POW);
				v[0] = v[1] = v[2] = vv;
				set[0].r = set[0].g = set[0].b = vv;

			} else {	/* Do R, G, B */
				v[0] = v[1] = v[2] = 0.0;
				v[i - nn] = 1.0;
				set[0].r = v[0];
				set[0].g = v[1];
				set[0].b = v[2];
			}

			if ((rv = dr->read(dr, set, 1, i+1, nn+3, 1, 0, instClamp, 0)) != 0) {
				dr->del(dr);
				error("display read failed with '%s'\n",disprd_err(rv));
			} 
			icmXYZ2Lab(&x.twN, alab, set[0].XYZ);
		
			fwddev(&x, mxyz, v);
			icmXYZ2Lab(&x.twN, mlab, mxyz);

			de = icmCIE2K(mlab, alab);
			if (de > mnerr) {
				mnerr = de;
				mnv = vv;
			}
			anerr += de;
			
			printf("RGB %.3f %.3f %.3f -> XYZ %.2f %.2f %.2f, model %.2f %.2f %.2f\n",
			set[0].r, set[0].g, set[0].b,
			set[0].XYZ[0], set[0].XYZ[1],
			set[0].XYZ[2], mxyz[0], mxyz[1], mxyz[2]);

			printf("RGB %.3f %.3f %.3f -> Lab %.2f %.2f %.2f, model %.2f %.2f %.2f, DE %f\n",
			set[0].r, set[0].g, set[0].b, alab[0], alab[1], alab[2], mlab[0], mlab[1], mlab[2],de);
		}
		anerr /= (double)(nn+3);
		printf("Model maximum error(@ %f) = %f deltaE\n",mnv, mnerr);
		printf("Model average error = %f deltaE\n",anerr);
	}
#endif /* CHECK_MODEL */

	/* Figure out our calibration curve parameter targets */
	if (!doupdate) {

		/* Figure out the target white point */
		if (wpx > 0.0 || wpy > 0.0) {	/* xy coordinates */
			double Yxy[3];
			Yxy[0] = 1.0;
			Yxy[1] = wpx;
			Yxy[2] = wpy;
			icmYxy2XYZ(x.twh, Yxy);

		} else if (temp > 0.0) {		/* Daylight color temperature */
			if (planckian)
				rv = icx_ill_sp2XYZ(x.twh, icxOT_default, NULL, icxIT_Ptemp, temp, NULL, 0);
			else
				rv = icx_ill_sp2XYZ(x.twh, icxOT_default, NULL, icxIT_Dtemp, temp, NULL, 0);
			if (rv != 0)
				error("Failed to compute XYZ of target color temperature %f\n",temp);
//printf("~1 Raw target from temp %f XYZ = %f %f %f\n",temp,x.twh[0],x.twh[1],x.twh[2]);
		} else {						/* Native white */
			x.twh[0] = x.wh[0]/x.wh[1];
			x.twh[1] = x.wh[1]/x.wh[1];
			x.twh[2] = x.wh[2]/x.wh[1];
		}
		x.nwh[0] = x.twh[0];
		x.nwh[1] = x.twh[1];
		x.nwh[2] = x.twh[2];

		/* Convert it to absolute white target */
		if (tbright > 0.0) {			/* Given brightness */
			x.twh[0] *= tbright;
			x.twh[1] *= tbright;
			x.twh[2] *= tbright;
		} else {						/* Native/maximum brightness */
			x.twh[0] *= x.wh[1];
			x.twh[1] *= x.wh[1];
			x.twh[2] *= x.wh[1];
			if (verb)
				printf("\nInitial native brightness target = %f cd/m^2\n", x.twh[1]);
		}

		/* Now make sure the target white will fit in gamut. */
		if (verify != 2 &&
		   ((tbright > 0.0 && invlindev(&x, NULL, x.twh) > 0.0) /* Defined brightness and clips */
		 || (tbright <= 0.0 && x.nat == 0))) {		/* Max non-native white */
			double rgb[3];
			double scale = 0.5;
			double sa = 0.1;

			if (powell(NULL, 1, &scale, &sa, 1e-7, 500, wp_opt_func, (void *)&x, NULL, NULL) != 0)
				error("WP scale powell failed");

			x.twh[0] *= scale;
			x.twh[1] *= scale;
			x.twh[2] *= scale;
			invdev(&x, rgb, x.twh);
			if (verb) {
				printf("Had to scale brightness from %f to %f to fit within gamut,\n",x.twh[1]/scale, x.twh[1]);
				printf("corresponding to aprox. RGB %f %f %f\n",rgb[0],rgb[1],rgb[2]);
			}
		}

		icmXYZ2Yxy(x.twYxy, x.twh);		/* For information */

		if (verb)
			printf("Target white value is XYZ %f %f %f [xy %f %f]\n",x.twh[0],x.twh[1],x.twh[2],
			                                                          x.twYxy[1], x.twYxy[2]);
	}

	/* Need this for Lab conversions */
	icmAry2XYZ(x.twN, x.twh);

	/* Figure out the black point target */
	{
		double tbL[3];
		double tbkLab[3]; 

		icmXYZ2Lab(&x.twN, tbkLab, x.bk);	/* Convert measured black to Lab */
	
//printf("~1 black point Lab = %f %f %f\n", tbkLab[0], tbkLab[1], tbkLab[2]);

		/* Now blend the a* b* with that of the target white point */
		/* according to how much to try and correct the hue. */
		tbL[0] = tbkLab[0];
		tbL[1] = bkcorrect * 0.0 + (1.0 - bkcorrect) * tbkLab[1];
		tbL[2] = bkcorrect * 0.0 + (1.0 - bkcorrect) * tbkLab[2];

//printf("~1 blended black Lab = %f %f %f\n", tbL[0], tbL[1], tbL[2]);
	
		if (bkbright > 0.0 && (bkbright <= x.bk[1] || (2.0 * bkbright) >= x.twh[1]))
			warning("Black brigtness %f ignored because it is out of range",bkbright);
		else if (bkbright > 0.0) {
			double bkbxyz[3], bkbLab[3], vv, bl;
			/* Figure out the L value of the target brightness */
			bkbxyz[0] = 0.0;
			bkbxyz[1] = bkbright;
			bkbxyz[2] = 0.0;
			icmXYZ2Lab(&x.twN, bkbLab, bkbxyz);
			
			/* Do crossover to white neutral */
			vv = bkbLab[0] / (100.0 - tbL[0]);
			bl = pow((1.0 - vv), x.nbrate);		/* Crossover near the black */
			tbL[0] = bkbLab[0];
			tbL[1] = (1.0 - bl) * 0.0 + bl * tbL[1];
			tbL[2] = (1.0 - bl) * 0.0 + bl * tbL[2];
//printf("~1 brighted black Lab = %f %f %f\n", tbL[0], tbL[1], tbL[2]);
		}

		/* And make this the black hue to aim for */
		icmLab2XYZ(&x.twN, x.tbk, tbL);
		icmAry2XYZ(x.tbN, x.tbk);
		if (verb)
			printf("Adjusted target black XYZ %.4f %.4f %.4f, Lab %.3f %.3f %.3f\n",
	        x.tbk[0], x.tbk[1], x.tbk[2], tbL[0], tbL[1], tbL[2]);
	}

	/* Figure out the gamma curve black offset value */
	/* that will give us the black level we actually have. */
	{
		double yy, tby;			/* Target black y */
		
		/* Make target black Y as high as necessary */
		/* to get the black point hue */
		/* ????? should do this by increasing L* until XYZ > x.bk ????? */
		tby = x.bk[1];
//printf("Target Y from Y = %f\n",tby);
		yy = x.bk[0] * x.tbk[1]/x.tbk[0];
//printf("Target Y from X = %f\n",yy);
		if (yy > tby)
			tby = yy;
		yy = x.bk[2] * x.tbk[1]/x.tbk[2];
//printf("Target Y from Z = %f\n",yy);
		if (yy > tby)
			tby = yy;

		if (x.tbk[1] > tby)		/* If target is already high enough */
			tby = x.tbk[1];

		if (verb) {
			double tbp[3], tbplab[3];
			if (fabs(x.tbk[1]) > 1e-9)
		        tbp[0] = x.tbk[0] * tby/x.tbk[1];
			else
		        tbp[0] = x.tbk[0];
			tbp[1] = tby;
			if (fabs(x.tbk[1]) > 1e-9)
				tbp[2] = x.tbk[2] * tby/x.tbk[1];
			else
				tbp[2] = x.tbk[2];
			icmXYZ2Lab(&x.twN, tbplab, tbp);
			printf("Target black after min adjust: XYZ %.4f %.4f %.4f, Lab %.3f %.3f %.3f\n",
			        tbp[0], tbp[1], tbp[2], tbplab[0], tbplab[1], tbplab[2]);
		}

		/* Figure out the x.gioff and egamma needed to get this x.gooff and gamma */
		x.gooff = tby / x.twh[1];			/* Convert to relative */

		/* tech_gamma() does the hard work */
		tech_gamma(&x, &x.egamma, &x.gooff, &x.gioff, egamma, gamma, x.gooff);
	}

	if (verb)
		printf("Gamma curve input offset = %f, output offset = %f, power = %f\n",x.gioff,x.gooff,x.egamma);

	/* For ambient light compensation, we make use of CIECAM02 */
	if (ambientl > 0.0) {
		double xyz[3], Jab[3];
		double t1, t0, a1, a0;

		/* Setup default source viewing conditions */
		if ((x.svc = new_icxcam(cam_default)) == NULL
		 || (x.dvc = new_icxcam(cam_default)) == NULL) {
			error("Failed to create source and destination CAMs");
		}
	
		switch(x.gammat) {
			case gt_power:	/* There's nothing obvious for these cases, */
			case gt_Lab:	/* So default to a computerish source viewing condition */

			case gt_sRGB:	/* sRGB standard viewing conditions */
				x.svc->set_view(x.svc, vc_none,
					x.nwh,					/* Display normalised white point */
					0.2 * 80.0,				/* Adapting luminence, 20% of display 80 cd/m^2 */
					0.2,					/* Background relative to reference white */
					80.0,					/* Display is 80 cd/m^2 */
			        0.0, 0.01, x.nwh,		/* 0% flare and 1% glare same white point */
					0, 1.0, 0.0, NULL);		/* No HK */
				break;

			case gt_Rec709:
			case gt_SMPTE240M:	/* Television studio conditions */
				x.svc->set_view(x.svc, vc_none,
					x.nwh,					/* Display normalised white point */
					0.2 * 1000.0/3.1415,	/* Adapting luminence, 20% of 1000 lux in cd/m^2 */
					0.2,					/* Background relative to reference white */
					1000.0/3.1415,			/* Luminance of white in the Image field (cd/m^2) */
			        0.0, 0.01, x.nwh,		/* 0% flare and 1% glare same white point */
					0, 1.0, 0.0, NULL);		/* No HK */
				break;

			default:
				error("Unknown gamma type");
		}
		/* The display we're calibratings situation */
		x.dvc->set_view(x.dvc, vc_none,
			x.nwh,				/* Display normalised white point */
			0.2 * ambientl/3.1415,	/* Adapting luminence, 20% of ambient in cd/m^2 */
			0.2,				/* Background relative to reference white */
			x.twh[1],			/* Target white level (cd/m^2) */
	        0.0, 0.01, x.nwh,	/* 0% flare and 1% glare same white point */
			0, 1.0, 0.0, NULL);	/* No HK */

		/* Compute the normalisation values */
		x.svc->XYZ_to_cam(x.svc, Jab, x.nwh);		/* Relative white point */
		x.dvc->cam_to_XYZ(x.dvc, xyz, Jab);
		t1 = x.nwh[1];
		a1 = xyz[1];

		xyz[0] = x.tbk[1]/x.twh[1] * x.nwh[0];
		xyz[1] = x.tbk[1]/x.twh[1] * x.nwh[1];
		xyz[2] = x.tbk[1]/x.twh[1] * x.nwh[2];
		t0 = xyz[1];
		x.svc->XYZ_to_cam(x.svc, Jab, xyz);		/* Relative black Y */
		x.dvc->cam_to_XYZ(x.dvc, xyz, Jab);
		a0 = xyz[1];

//printf("~1 t1 = %f, t0 = %f\n",t1,t0);
//printf("~1 a1 = %f, a0 = %f\n",a1,a0);
		x.vn1 = (t1 - t0)/(a1 - a0);		/* Scale factor */
		x.vn0 = t0 - (a0 * x.vn1);			/* Then offset */
//printf("~1 vn1 = %f, vn0 = %f\n",x.vn1, x.vn0);
//printf("~1 fix a1 = %f, should be = %f\n",a1 * x.vn1 + x.vn0, t1);
//printf("~1 fix a0 = %f, should be = %f\n",a0 * x.vn1 + x.vn0, t0);

		x.vc = 1;

		/* Compute aproximate power of viewing transform */
		if (verb) {
			double v;
			v = view_xform(&x, 0.5);
			v = log(v) / log(0.5);
			printf("Viewing conditions adjustment aprox. power = %f\n",v);
		}
#ifdef NEVER
{
	int i;

	printf("~1 viewing xtranform:\n");
	for (i = 0; i <= 10; i++) {
		double w, v = i/10.0;
		
		w = view_xform(&x, v);
		printf("~1 in %f -> %f\n",v,w); 
	}
}
#endif /* NEVER */
	}

	/* - - - - - - - - - - - - - - - - - - - - - */
	/* Make sure nver has a sane value */
	if (verify == 0)
		nver = 0;			/* 0 verify count if no verify */
	else if (nver == 0)
		nver = 1;			/* min 1 count if verify */

	/* Start with a scaled down number of test points and refine threshold, */
	/* and double/halve these on each iteration. */
	if (verb && verify != 2)
		printf("Total Iteration %d, Final Samples = %d Final Repeat threshold = %f\n",
		        mxits, rsteps, errthr);
	if (verify == 2) {
		rsteps = VER_RES;
		errthr = 0.0;
	} else {
		rsteps /= (1 << (mxits-1));
		errthr *= pow((double)(1 << (mxits-1)), THRESH_SCALE_POW);
	}

	/* Setup the initial calibration test point values */
	init_csamp(&asgrey, &x, doupdate, verify, verify == 2 ? 1 : 0, rsteps, verb);
	
	/* Calculate the initial calibration curve values */
	if (verify != 2 && !doupdate) {
		int nsamp = 128;
		mcvco *sdv[3];				/* Scattered data for creating mcv */

		for (j = 0; j < 3; j++) {
			if ((x.rdac[j] = new_mcv()) == NULL) {
				dr->del(dr);
				error("new_mcv x.rdac[%d] failed",j);
			}
		}

		for (j = 0; j < 3; j++) {
			if ((sdv[j] = malloc(sizeof(mcvco) * rsteps)) == NULL) {
				dr->del(dr);
				error("Malloc of scattered data points failed");
			}
		}

		if (verb)
			printf("Creating initial calibration curves...\n");

		/* Copy the sample points */
		for (i = 0; i < rsteps; i++) {
			for (j = 0; j < 3; j++) {
				sdv[j][i].p = asgrey.s[i].v;
				sdv[j][i].v = asgrey.s[i].rgb[j];
				sdv[j][i].w = 1.0;
			}
		}
		if (x.nat)		/* Make curve go thought white if possible by setting a weighting */
			sdv[0][rsteps-1].w = sdv[1][rsteps-1].w = sdv[2][rsteps-1].w = 50.0;

		if (x.bkhack)	/* Make curve go thought black if possible by setting a weighting */
			sdv[0][0].w = sdv[1][0].w = sdv[2][0].w = 50.0;

		/* Create an initial set of RAMDAC curves */
		for (j = 0; j < 3; j++)
			x.rdac[j]->fit(x.rdac[j], 0, fitord, sdv[j], rsteps, RDAC_SMOOTH);

		/* Make sure that if we are using native brightness and white point, */
		/* that the curves go to a perfect 1.0 ... */
		if (x.nat) {
			for (j = 0; j < 3; j++)
				x.rdac[j]->force_1(x.rdac[j], 1.0);
		}

		/* Make sure that if we are using black point hack, */
		/* that the curves go to a perfect 0.0 ... */
		if (x.bkhack) {
			for (j = 0; j < 3; j++)
				x.rdac[j]->force_0(x.rdac[j], 0.0);
		}

		for (j = 0; j < 3; j++)
			free (sdv[j]);
	}

#ifdef DEBUG_PLOT
	/* Plot the initial curves */
	if (verify != 2) {
		#define	XRES 255
		double xx[XRES];
		double y1[XRES];
		double y2[XRES];
		double y3[XRES];
		double rgb[3];
		for (i = 0; i < XRES; i++) {
			double drgb[3], rgb[3];
			xx[i] = i/(XRES-1.0);
			rgb[0] = rgb[1] = rgb[2] = xx[i];
			for (j = 0; j < 3; j++)
				drgb[j] = x.rdac[j]->interp(x.rdac[j], rgb[j]);
			y1[i] = drgb[0];
			y2[i] = drgb[1];
			y3[i] = drgb[2];
		}
		printf("Initial ramdac curves\n");
		do_plot(xx,y1,y2,y3,XRES);
		#undef XRES
	}
#endif

	/* If native white and white drift compensation enabled, */
	/* reset white drift target at start of main cal. */
	if (x.nat && asgrey.s[0].v == 1.0 && wdrift)
		dr->reset_targ_w(dr);

	/* Now we go into the main verify & refine loop */
	for (it = (verify == 2) ? mxits : 0;
	     it < (mxits + nver);
	     rsteps *= 2, errthr /= (it < mxits) ? pow(2.0,THRESH_SCALE_POW) : 1.0, it++) {
		int totmeas = 0;		/* Total number of measurements in this pass */
		col set[3];				/* Variable to read one to three values from the display */

		/* Verify pass ? */
		if (it >= mxits)
			rsteps = VER_RES;	/* Fixed verification resolution */
		else
			thrfail = 0;		/* Not verify pass */

		/* re-init asgrey if the number of test points has changed */
		reinit_csamp(&asgrey, &x, verify, it >= mxits ? 1 : 0, rsteps, verb);

		if (verb) {
			if (it >= mxits)
				printf("\nDoing verify pass %d/%d with %d sample points\n",
				                              it - mxits+1, nver, rsteps);
			else
				printf("\nDoing iteration %d/%d with %d sample points and repeat threshold of %f DE\n",
				                                                      it+1,mxits, rsteps, errthr);
		}

		/* Read and adjust each step */
		/* Do this white to black to track drift in native white point */
		for (i = rsteps-1; i >= 0; i--) {
			int rpt;
			double peqXYZ[3];		/* Previous steps equivalent aim point */
			double bestrgb[3];		/* In case we fail */
			double bestxyz[3];
			double prevde = 1e7;
			double best_de = 1e7;
			double bestde = 1e7;
			double bestdc = 1e7;
			double bestpeqde = 1e7;
			double besthde = 1e7;
			double rgain = REFINE_GAIN;	/* Scale down if lots of repeats */
			int mjac = 0;				/* We measured the Jacobian */
			double ierrth = errthr;		/* This points error threshold */
			
			icmCpy3(asgrey.s[i].prgb, asgrey.s[i].rgb);		/* Init previous */

			/* Setup a second termination threshold criteria based on */
			/* the delta E to the previous step point for the last pass. */
			/* This is to try and steer towards neutral axis consistency ? */
			if (i == (rsteps-1) || it < (mxits-1)) {
				icmAry2Ary(peqXYZ, asgrey.s[i].tXYZ);	/* Its own aim point */
			} else {
				double Lab1[3], Lab2[3], Lab3[3];
				icmXYZ2Lab(&x.twN, Lab1, asgrey.s[i+1].XYZ);
				icmXYZ2Lab(&x.twN, Lab2, asgrey.s[i].tXYZ);
				Lab1[0] = Lab2[0];		/* L of current target with ab of previous as 2nd threshold */
				icmLab2XYZ(&x.twN, peqXYZ, Lab1);		/* Previous equivalent */
			}

#ifdef ADJ_THRESH
			/* Adjust the termination threshold to make sure it is less than */
			/* half a step */
			if (it == (mxits-1)) {		/* If final pass */
				double de;
				if (i < (rsteps-1)) {
					de = 0.5 * icmXYZLabDE(&x.twN, asgrey.s[i].tXYZ, asgrey.s[i+1].tXYZ);
					if (de < MIN_THRESH)		/* Don't be silly */
						de = MIN_THRESH;
					if (de < ierrth)
						ierrth = de;
				}
				if (i > 0) {
					de = 0.5 * icmXYZLabDE(&x.twN, asgrey.s[i].tXYZ, asgrey.s[i-1].tXYZ);
					if (de < MIN_THRESH)		/* Don't be silly */
						de = MIN_THRESH;
					if (de < ierrth)
						ierrth = de;
				}
			}
#endif /* ADJ_THRESH */

			/* Until we meet the necessary accuracy or give up */
			for (rpt = 0; rpt < mxrpts; rpt++) {
				double hlew = 1.0;	/* high L* error weight */
				int gworse = 0;		/* information flag */
				double w_de, wde;	/* informational */
				double pjadj[3][3] = { 0.0 };	/* Previous jacobian adjustment */

				set[0].r = asgrey.s[i].rgb[0];
				set[0].g = asgrey.s[i].rgb[1];
				set[0].b = asgrey.s[i].rgb[2];
				set[0].id = NULL;

				/* Read patches (no auto cr in case we repeat last patch) */
				if ((rv = dr->read(dr, set, 1, rsteps-i, rsteps, 0, 0, instClamp, 0)) != 0) {
					dr->del(dr);
					error("display read failed with '%s'\n",disprd_err(rv));
				} 
				totmeas++;
	
				icmAry2Ary(asgrey.s[i].pXYZ, asgrey.s[i].XYZ);	/* Remember previous XYZ */
				icmAry2Ary(asgrey.s[i].XYZ, set[0].XYZ);		/* Transfer current reading */

				/* If native white and we've just measured it, */
				/* and we're not doing a verification, */
				/* adjust all the other point targets txyz to track the white. */
				if (x.nat && i == (rsteps-1) && it < mxits && asgrey.s[i].v == 1.0) {
					icmAry2Ary(x.twh, asgrey.s[i].XYZ);		/* Set current white */
					icmAry2XYZ(x.twN, x.twh);				/* Need this for Lab conversions */
					init_csamp_txyz(&asgrey, &x, 1, verb);	/* Recompute txyz's */
					icmAry2Ary(peqXYZ, asgrey.s[i].tXYZ);	/* Fix peqXYZ */
//printf("~1 Just reset target white to native white\n");
					if (wdrift) {	/* Make sure white drift is reset on next read. */
						dr->reset_targ_w(dr);			/* Reset this */
					}
				}

				/* If black point hack and we've just measured it, */
				/* and we're not doing a verification, */
				if (x.bkhack && i == 0 && it < mxits && asgrey.s[i].v == 0.0) {
					icmAry2Ary(x.tbk, asgrey.s[i].XYZ);		/* Set current black */
					icmAry2XYZ(x.tbN, x.tbk);
					init_csamp_txyz(&asgrey, &x, 1, verb);	/* Recompute txyz's */
					icmAry2Ary(peqXYZ, asgrey.s[i].tXYZ);	/* Fix peqXYZ */
				}

				/* Compute the next change wanted to hit target */
				icmSub3(asgrey.s[i].deXYZ, asgrey.s[i].tXYZ, asgrey.s[i].XYZ);

				/* For the darkest 5% of targets, weight the L* delta E so that */
				/* we err on the darker side */
				if (asgrey.s[i].v < POWERR_THR)
					hlew = 1.0 + POWERR_WEIGHT * pow((POWERR_THR - asgrey.s[i].v)/POWERR_THR,
					                                                       POWERR_WEIGHT_POW);
				else
					hlew = 1.0;
//printf("~1 i %d, v %f, hlew %f\n",i,asgrey.s[i].v,hlew);
				asgrey.s[i]._de = icmXYZLabDE(&x.twN, asgrey.s[i].tXYZ, asgrey.s[i].XYZ);
				asgrey.s[i].de = bwXYZLabDE(&x.twN, asgrey.s[i].tXYZ, asgrey.s[i].XYZ, hlew);
				asgrey.s[i].peqde = bwXYZLabDE(&x.twN, peqXYZ, asgrey.s[i].XYZ, hlew);
				asgrey.s[i].hde = 0.8 * asgrey.s[i].de + 0.2 * asgrey.s[i].peqde;
				/* Eudclidian difference of XYZ, because this doesn't always track Lab */
				asgrey.s[i].dc = icmLabDE(asgrey.s[i].tXYZ, asgrey.s[i].XYZ);

				/* Compute actual change from last XYZ */
				icmSub3(asgrey.s[i].dXYZ, asgrey.s[i].XYZ, asgrey.s[i].pXYZ);

				w_de = asgrey.s[i]._de;
				wde = asgrey.s[i].de;

				if (verb >= 3) {
					printf("\n\nTest point %d, v = %f, rpt %d\n",rsteps - i,asgrey.s[i].v,rpt);
					printf("Current rgb %f %f %f -> XYZ %f %f %f, de %f, dc %f\n", 
					asgrey.s[i].rgb[0], asgrey.s[i].rgb[1], asgrey.s[i].rgb[2],
					asgrey.s[i].XYZ[0], asgrey.s[i].XYZ[1], asgrey.s[i].XYZ[2],
					asgrey.s[i]._de, asgrey.s[i].dc);
					printf("Target XYZ %f %f %f, delta needed %f %f %f\n", 
					asgrey.s[i].tXYZ[0], asgrey.s[i].tXYZ[1], asgrey.s[i].tXYZ[2],
					asgrey.s[i].deXYZ[0], asgrey.s[i].deXYZ[1], asgrey.s[i].deXYZ[2]);
					if (rpt > 0) {
						printf("Last intended XYZ change %f %f %f, actual change %f %f %f\n", 
						asgrey.s[i].pdXYZ[0], asgrey.s[i].pdXYZ[1], asgrey.s[i].pdXYZ[2],
						asgrey.s[i].dXYZ[0], asgrey.s[i].dXYZ[1], asgrey.s[i].dXYZ[2]);
					}
				}

				if (it < mxits) {		/* Not verify, apply correction */
					int impj = 0;		/* We adjusted the Jacobian */
					int dclip = 0;		/* We clipped the new RGB */
#ifdef ADJ_JACOBIAN
					int isclipped = 0;

#ifndef CLIP		/* Check for cliping */
					/* Don't try and update the Jacobian if the */	
					/* device values are going out of gamut, */
					/* and being clipped without Jac correction being aware. */
					for (j = 0; j < 3; j++) {
						if (asgrey.s[i].rgb[j] <= 0.0 || asgrey.s[i].rgb[j] >= 1.0) {
							isclipped = 1;
							break;
						}
					}
#endif /* !CLIP */

#ifdef REMEAS_JACOBIAN
					/* If the de hasn't improved, try and measure the Jacobian */
//					if (it < (rsteps-1) && mjac == 0 && asgrey.s[i].de > (0.8 * prevde))
					if (mjac == 0 && asgrey.s[i].de > (0.8 * prevde))
					{
						double dd;
						if (asgrey.s[i].v < 0.5)
							dd = 0.05;
						else
							dd= -0.05;
						set[0].r = asgrey.s[i].rgb[0] + dd;
						set[0].g = asgrey.s[i].rgb[1];
						set[0].b = asgrey.s[i].rgb[2];
						set[0].id = NULL;
						set[1].r = asgrey.s[i].rgb[0];
						set[1].g = asgrey.s[i].rgb[1] + dd;
						set[1].b = asgrey.s[i].rgb[2];
						set[1].id = NULL;
						set[2].r = asgrey.s[i].rgb[0];
						set[2].g = asgrey.s[i].rgb[1];
						set[2].b = asgrey.s[i].rgb[2] + dd;
						set[2].id = NULL;

						if ((rv = dr->read(dr, set, 1, rsteps-i, rsteps, 0, 0, instClamp, 0)) != 0
						 || (rv = dr->read(dr, set+1, 1, rsteps-i, rsteps, 0, 0, instClamp, 0)) != 0
						 || (rv = dr->read(dr, set+2, 1, rsteps-i, rsteps, 0, 0, instClamp, 0)) != 0) {
							dr->del(dr);
							error("display read failed with '%s'\n",disprd_err(rv));
						} 
						totmeas += 3;

//printf("\n~1 remeasured jacobian\n");
						/* Matrix organization is J[XYZ][RGB] for del RGB->del XYZ*/
						for (j = 0; j < 3; j++) {
							asgrey.s[i].j[0][j] = (set[j].XYZ[0] - asgrey.s[i].XYZ[0]) / dd;
							asgrey.s[i].j[1][j] = (set[j].XYZ[1] - asgrey.s[i].XYZ[1]) / dd;
							asgrey.s[i].j[2][j] = (set[j].XYZ[2] - asgrey.s[i].XYZ[2]) / dd;
						}

						/* Clear pjadj */
						for (j = 0; j < 3; j++)
							pjadj[j][0] = pjadj[j][1] = pjadj[j][2] = 0.0;

						if (icmInverse3x3(asgrey.s[i].ij, asgrey.s[i].j)) {
							/* Should repeat with bigger dd ? */
							if (verb)
								printf("dispcal: inverting Jacobian failed (3) - falling back\n");

							/* Revert to the initial Jacobian */
							icmCpy3x3(asgrey.s[i].ij, asgrey.s[i].fb_ij);
						}
						/* Restart at the best we've had */
						if (asgrey.s[i].hde > besthde) {
							asgrey.s[i]._de = best_de;
							asgrey.s[i].de = bestde;
							asgrey.s[i].dc = bestdc;
							asgrey.s[i].peqde = bestpeqde;
							asgrey.s[i].hde = besthde;
							asgrey.s[i].rgb[0] = bestrgb[0];
							asgrey.s[i].rgb[1] = bestrgb[1];
							asgrey.s[i].rgb[2] = bestrgb[2];
							asgrey.s[i].XYZ[0] = bestxyz[0];
							asgrey.s[i].XYZ[1] = bestxyz[1];
							asgrey.s[i].XYZ[2] = bestxyz[2];
							icmSub3(asgrey.s[i].deXYZ, asgrey.s[i].tXYZ, asgrey.s[i].XYZ);
						}
						mjac = 1;
						impj = 1;		/* Have remeasured */
					}
#endif /* REMEAS_JACOBIAN */

					/* Compute a correction to the Jacobian if we can. */
					/* (Don't do this unless we have a solid previous */
					/* reading for this patch, and we haven't remeasured it) */ 
					if (impj == 0 && rpt > 0 && isclipped == 0) {
						double nsdrgb;			/* Norm squared of pdrgb */
						double spdrgb[3];		/* Scaled previous delta rgb */
						double dXYZerr[3];		/* Error in previous prediction */
						double jadj[3][3];		/* Adjustment to Jacobian */
						double tj[3][3];		/* Temp Jacobian */
						double itj[3][3];		/* Temp inverse Jacobian */

//printf("~1 Jacobian was: %f %f %f\n", asgrey.s[i].j[0][0], asgrey.s[i].j[0][1], asgrey.s[i].j[0][2]);
//printf("~1               %f %f %f\n", asgrey.s[i].j[1][0], asgrey.s[i].j[1][1], asgrey.s[i].j[1][2]);
//printf("~1               %f %f %f\n", asgrey.s[i].j[2][0], asgrey.s[i].j[2][1], asgrey.s[i].j[2][2]);

						/* Use Broyden's Formula */
						icmSub3(dXYZerr, asgrey.s[i].dXYZ, asgrey.s[i].pdXYZ);
//printf("~1 Jacobian error = %f %f %f\n", dXYZerr[0], dXYZerr[1], dXYZerr[2]);
						nsdrgb = icmNorm3sq(asgrey.s[i].pdrgb);
						/* If there was sufficient change in device values */
						/* to be above any noise: */ 
						if (nsdrgb >= (0.005 * 0.005)) {
							icmScale3(spdrgb, asgrey.s[i].pdrgb, 1.0/nsdrgb);
							icmTensMul3(jadj, dXYZerr, spdrgb);

#ifdef DEBUG
							/* Check that new Jacobian predicts previous delta XYZ */
							{
								double eXYZ[3];
	
//printf("~1 del RGB %f %f %f got del XYZ %f %f %f\n", asgrey.s[i].pdrgb[0], asgrey.s[i].pdrgb[1], asgrey.s[i].pdrgb[2], asgrey.s[i].dXYZ[0], asgrey.s[i].dXYZ[1], asgrey.s[i].dXYZ[2]);

								/* Make a full adjustment to temporary Jac */
								icmAdd3x3(tj, asgrey.s[i].j, jadj);

//printf("~1 Full Jacobian: %f %f %f\n", tj[0][0], tj[0][1], tj[0][2]);
//printf("~1                %f %f %f\n", tj[1][0], tj[1][1], tj[1][2]);
//printf("~1                %f %f %f\n", tj[2][0], tj[2][1], tj[2][2]);

								icmMulBy3x3(eXYZ, tj, asgrey.s[i].pdrgb);
								icmSub3(eXYZ, eXYZ, asgrey.s[i].dXYZ);
								printf("Jac check resid %f %f %f\n", eXYZ[0], eXYZ[1], eXYZ[2]);
							}
#endif	/* DEBUG */

							/* Add to portion of previous adjustment */
							/* to counteract undershoot & overshoot */
							icmScale3x3(pjadj, pjadj, JAC_COMP_FACT);
							icmAdd3x3(jadj, jadj, pjadj);
							icmCpy3x3(pjadj, jadj);

							/* Add part of our correction to actual Jacobian */
							/* to smooth out correction to counteract noise */
							icmScale3x3(jadj, jadj, JAC_COR_FACT);
							icmAdd3x3(tj, asgrey.s[i].j, jadj);

							if (icmInverse3x3(itj, tj) == 0) {		/* Invert OK */
								icmCpy3x3(asgrey.s[i].j, tj);		/* Use adjusted */
								icmCpy3x3(asgrey.s[i].ij, itj);
								impj = 1;

#ifdef NEVER
					/* Check how close new Jacobian predicts previous delta XYZ */
					{
						double eXYZ[3];
						double ergb[3];

						icmMulBy3x3(eXYZ, asgrey.s[i].j, asgrey.s[i].pdrgb);
						icmSub3(eXYZ, eXYZ, asgrey.s[i].dXYZ);
						printf("Jac check2 resid %f %f %f\n", eXYZ[0], eXYZ[1], eXYZ[2]);

						icmMulBy3x3(ergb, asgrey.s[i].ij, asgrey.s[i].pdXYZ);
						printf("Jac check2 drgb would have been %f %f %f\n", ergb[0], ergb[1], ergb[2]);
						icmAdd3(ergb, ergb, asgrey.s[i].prgb);
						printf("Jac check2 rgb would have been %f %f %f\n", ergb[0], ergb[1], ergb[2]);
					}
#endif
							}
//else printf("~1 ij failed - reverted\n");
						}
//else printf("~1 nsdrgb was below threshold\n");
					}
//else if (isclipped) printf("~1 no j update: rgb is clipped\n");
//printf("~1 Jacobian now: %f %f %f\n", asgrey.s[i].j[0][0], asgrey.s[i].j[0][1], asgrey.s[i].j[0][2]);
//printf("~1               %f %f %f\n", asgrey.s[i].j[1][0], asgrey.s[i].j[1][1], asgrey.s[i].j[1][2]);
//printf("~1               %f %f %f\n", asgrey.s[i].j[2][0], asgrey.s[i].j[2][1], asgrey.s[i].j[2][2]);

#endif	/* ADJ_JACOBIAN */

					/* Track the best solution we've found */
					if (asgrey.s[i].hde <= besthde) {
						best_de = asgrey.s[i]._de;
						bestde = asgrey.s[i].de;
						bestdc = asgrey.s[i].dc;
						bestpeqde = asgrey.s[i].peqde;
						besthde = asgrey.s[i].hde;
						bestrgb[0] = asgrey.s[i].rgb[0];
						bestrgb[1] = asgrey.s[i].rgb[1];
						bestrgb[2] = asgrey.s[i].rgb[2];
						bestxyz[0] = asgrey.s[i].XYZ[0];
						bestxyz[1] = asgrey.s[i].XYZ[1];
						bestxyz[2] = asgrey.s[i].XYZ[2];

//printf("~1 new best\n");
					} else if (asgrey.s[i].dc > bestdc) {
						/* we got worse in Lab and XYZ ! */

						/* If we've wandered too far, return to best we found */
						if (asgrey.s[i].hde > (3.0 * besthde)) {
//printf("~1 resetting to last best\n");
							asgrey.s[i]._de = best_de;
							asgrey.s[i].de = bestde;
							asgrey.s[i].dc = bestdc;
							asgrey.s[i].peqde = bestpeqde;
							asgrey.s[i].hde = besthde;
							asgrey.s[i].rgb[0] = bestrgb[0];
							asgrey.s[i].rgb[1] = bestrgb[1];
							asgrey.s[i].rgb[2] = bestrgb[2];
							asgrey.s[i].XYZ[0] = bestxyz[0];
							asgrey.s[i].XYZ[1] = bestxyz[1];
							asgrey.s[i].XYZ[2] = bestxyz[2];
							icmSub3(asgrey.s[i].deXYZ, asgrey.s[i].tXYZ, asgrey.s[i].XYZ);
						}

						/* If the Jacobian hasn't changed, moderate the gain */
						if (impj == 0) {
							rgain *= 0.8;		/* We might be overshooting */
//printf("~1 reducing rgain to %f\n",rgain);
						}
						gworse = 1;
					}

					/* See if we need to repeat */
					if (asgrey.s[i].de <= ierrth && asgrey.s[i].peqde < ierrth) {	
						if (verb > 1) {
							if (it < (mxits-1))
								printf("Point %d DE %f, W.DE %f, OK ( < %f)\n",rsteps - i,asgrey.s[i]._de, asgrey.s[i].de, ierrth);
							else
								printf("Point %d DE %f, W.DE %f, W.peqDE %f, OK ( < %f)\n",rsteps - i,asgrey.s[i]._de,asgrey.s[i].de, asgrey.s[i].peqde, ierrth);
						}
						break;	/* No more retries */
					}
					if ((rpt+1) >= mxrpts) {
						asgrey.s[i]._de = best_de;			/* Restore to best we found */
						asgrey.s[i].de = bestde;
						asgrey.s[i].dc = bestdc;
						asgrey.s[i].peqde = bestpeqde;		/* Restore to best we found */
						asgrey.s[i].hde = besthde;			/* Restore to best we found */
						asgrey.s[i].rgb[0] = bestrgb[0];
						asgrey.s[i].rgb[1] = bestrgb[1];
						asgrey.s[i].rgb[2] = bestrgb[2];
						asgrey.s[i].XYZ[0] = bestxyz[0];
						asgrey.s[i].XYZ[1] = bestxyz[1];
						asgrey.s[i].XYZ[2] = bestxyz[2];
						if (verb > 1) {
							if (it < (mxits-1))
								printf("Point %d DE %f, W.DE %f, Fail ( > %f)\n",rsteps - i,asgrey.s[i]._de, asgrey.s[i].de, ierrth);
							else
								printf("Point %d DE %f, W.DE %f, W.peqDE %f, Fail ( > %f)\n",rsteps - i,asgrey.s[i]._de,asgrey.s[i].de,asgrey.s[i].peqde,ierrth);
						}
						thrfail = 1;						/* Failed to meet target */
						if (bestde > failerr)
							failerr = bestde;				/* Worst failed delta E */
						break;	/* No more retries */
					}
					if (verb > 1) {
						if (gworse)
							if (it < (mxits-1))
								printf("Point %d DE %f, W.DE %f, Repeat (got worse)\n", rsteps - i, w_de, wde);
							else
								printf("Point %d DE %f, W.DE %f, peqDE %f, Repeat (got worse)\n", rsteps - i, w_de, wde,asgrey.s[i].peqde);
						else
							if (it < (mxits-1))
								printf("Point %d DE %f, W.DE %f, Repeat\n", rsteps - i,asgrey.s[i]._de,asgrey.s[i].de);
							else
								printf("Point %d DE %f, W.DE %f, peqDE %f, Repeat\n", rsteps - i,asgrey.s[i]._de,asgrey.s[i].de,asgrey.s[i].peqde);
					}
				
//printf("~1 RGB Jacobian: %f %f %f\n", asgrey.s[i].j[0][0], asgrey.s[i].j[0][1], asgrey.s[i].j[0][2]);
//printf("~1               %f %f %f\n", asgrey.s[i].j[1][0], asgrey.s[i].j[1][1], asgrey.s[i].j[1][2]);
//printf("~1               %f %f %f\n", asgrey.s[i].j[2][0], asgrey.s[i].j[2][1], asgrey.s[i].j[2][2]);
					/* Compute refinement of rgb */
					icmMulBy3x3(asgrey.s[i].pdrgb, asgrey.s[i].ij, asgrey.s[i].deXYZ);
//printf("~1 XYZ delta needed %f %f %f -> delta RGB %f %f %f\n",
//asgrey.s[i].deXYZ[0], asgrey.s[i].deXYZ[1], asgrey.s[i].deXYZ[2],
//asgrey.s[i].pdrgb[0], asgrey.s[i].pdrgb[1], asgrey.s[i].pdrgb[2]);

					/* Gain scale */
					icmScale3(asgrey.s[i].pdrgb, asgrey.s[i].pdrgb, rgain);
//printf("~1 delta RGB after gain scale %f %f %f\n", asgrey.s[i].pdrgb[0], asgrey.s[i].pdrgb[1], asgrey.s[i].pdrgb[2]);

#ifdef CLIP
					/* Component wise clip */
					for (j = 0; j < 3; j++) {		/* Check for clip */
						if ((-asgrey.s[i].pdrgb[j]) > asgrey.s[i].rgb[j]) {
							asgrey.s[i].pdrgb[j] = -asgrey.s[i].rgb[j];
							dclip = 1;
						}
						if (asgrey.s[i].pdrgb[j] > (1.0 - asgrey.s[i].rgb[j])) {
							asgrey.s[i].pdrgb[j] = (1.0 - asgrey.s[i].rgb[j]);
							dclip = 1;
						}
					}
					if (verb >= 3 && dclip) printf("delta RGB after clip %f %f %f\n",
					       asgrey.s[i].pdrgb[0], asgrey.s[i].pdrgb[1], asgrey.s[i].pdrgb[2]);
#endif	/* CLIP */
					/* Compute next on the basis of this one RGB */
					icmCpy3(asgrey.s[i].prgb, asgrey.s[i].rgb);		/* Save previous */
					icmAdd3(asgrey.s[i].rgb, asgrey.s[i].rgb, asgrey.s[i].pdrgb);

					/* Save expected change in XYZ */
					icmMulBy3x3(asgrey.s[i].pdXYZ, asgrey.s[i].j, asgrey.s[i].pdrgb);
					if (verb >= 3) {
						printf("New rgb %f %f %f from expected del XYZ %f %f %f\n",
						       asgrey.s[i].rgb[0], asgrey.s[i].rgb[1], asgrey.s[i].rgb[2],
						       asgrey.s[i].pdXYZ[0], asgrey.s[i].pdXYZ[1], asgrey.s[i].pdXYZ[2]);
					}
				} else {	/* Verification, so no repeat */
					break;
				}

				prevde = asgrey.s[i].de;
			}	/* Next repeat */

			if (verb >= 3) {
				printf("After adjustment:\n");
				printf("Current rgb %f %f %f -> XYZ %f %f %f, de %f, dc %f\n", 
				asgrey.s[i].rgb[0], asgrey.s[i].rgb[1], asgrey.s[i].rgb[2],
				asgrey.s[i].XYZ[0], asgrey.s[i].XYZ[1], asgrey.s[i].XYZ[2],
				asgrey.s[i].de, asgrey.s[i].dc);
				printf("Target XYZ %f %f %f, delta needed %f %f %f\n", 
				asgrey.s[i].tXYZ[0], asgrey.s[i].tXYZ[1], asgrey.s[i].tXYZ[2],
				asgrey.s[i].deXYZ[0], asgrey.s[i].deXYZ[1], asgrey.s[i].deXYZ[2]);
			}

		}		/* Next patch/step */
		if (verb)
			printf("\n");			/* Final return for patch count */

#ifdef DEBUG_PLOT
		/* Plot the measured response XYZ */
		{
			#define	XRES 256
			double xx[XRES];
			double yy[3][XRES];
			double xyz[3];
			for (i = 0; i < XRES; i++) {
				xx[i] = i/(XRES-1.0);
				csamp_interp(&asgrey, xyz, xx[i]);
				for (j = 0; j < 3; j++)
					yy[j][i] = xyz[j];
			}
			printf("Measured neutral axis XYZ\n",k);
			do_plot(xx,yy[0],yy[1],yy[2],XRES);
			#undef XRES
		}
#endif

		/* Check out the accuracy of the results: */
		{
			double ctwh[3];		/* Current target white */
			icmXYZNumber ctwN;	/* Same as above as XYZNumber */
			double brerr;		/* Brightness error */
			double cterr;		/* Color temperature delta E */
			double mnerr;		/* Maximum neutral error */
			double mnv = 0.0;	/* Value where maximum error is */
			double anerr;		/* Average neutral error */
			double lab1[3], lab2[3];
			
			/* Brightness */
			brerr = asgrey.s[asgrey.no-1].XYZ[1] - x.twh[1];
		
			/* Compensate for brightness error */
			for (j = 0; j < 3; j++)
				ctwh[j] = x.twh[j] * asgrey.s[asgrey.no-1].XYZ[1]/x.twh[1];
			icmAry2XYZ(ctwN, ctwh);		/* Need this for Lab conversions */
			
			/* Color temperature error */
			icmXYZ2Lab(&ctwN, lab1, ctwh);		/* Should be 100,0,0 */
			icmXYZ2Lab(&ctwN, lab2, asgrey.s[asgrey.no-1].XYZ);
			cterr = icmLabDE(lab1, lab2);

			/* check delta E of all the sample points */
			/* We're checking against our given brightness and */
			/* white point target. */
			mnerr = anerr = 0.0;
			init_csamp_txyz(&asgrey, &x, 0, verb);	/* In case the targets were tweaked */
			for (i = 0; i < asgrey.no; i++) {
				double err;

				/* Re-compute de in case last pass had tweaked targets */
				asgrey.s[i].de = icmXYZLabDE(&x.twN, asgrey.s[i].tXYZ, asgrey.s[i].XYZ);
				err = asgrey.s[i].de;
//printf("RGB %.3f -> Lab %.2f %.2f %.2f, target %.2f %.2f %.2f, DE %f\n",
//asgrey.s[i].v, lab2[0], lab2[1], lab2[2], lab1[0], lab1[1], lab1[2], err);
				if (err > mnerr) {
					mnerr = err;
					mnv = asgrey.s[i].v;
				}
				anerr += err;
			}
			anerr /= (double)asgrey.no;

			if (verb || it >= mxits) {
				if (it >= mxits)
					printf("Verification results:\n");
				printf("Brightness error = %f cd/m^2 (is %f, should be %f)\n",brerr,asgrey.s[asgrey.no-1].XYZ[1],x.twh[1]);
				printf("White point error = %f deltaE\n",cterr);
				printf("Maximum neutral error (@ %f) = %f deltaE\n",mnv, mnerr);
				printf("Average neutral error = %f deltaE\n",anerr);
				if (it < mxits && thrfail)
					printf("Failed to meet target %f delta E, got worst case %f\n",errthr,failerr);
				printf("Number of measurements taken = %d\n",totmeas);
			}
		}

		/* Convert our test points into calibration curves. */
		/* The call to reinit_csamp() will then convert the */
		/* curves back to current test point values. */
		/* This applies some level of cohesion between the test points, */
		/* as well as forcing monotomicity */
		if (it < mxits) {		/* If not verify pass */
			mcvco *sdv[3];				/* Scattered data for mcv */

			for (j = 0; j < 3; j++) {
				if ((sdv[j] = malloc(sizeof(mcvco) * asgrey.no)) == NULL) {
					dr->del(dr);
					error("Malloc of scattered data points failed");
				}
			}

			if (verb)
				printf("Computing update to calibration curves...\n");

			/* Use fixed rgb's */
			for (j = 0; j < 3; j++) {
				for (i = 0; i < asgrey.no; i++) {
					sdv[j][i].p = asgrey.s[i].v;
					sdv[j][i].v = asgrey.s[i].rgb[j];
					sdv[j][i].w = 1.0;
#ifdef NEVER
					printf("rdac %d point %d = %f, %f\n",j,i,sdv[j][i].p,sdv[j][i].v);
#endif
				}
			}
			if (x.nat)		/* Make curve go thought white if possible */
				sdv[0][rsteps-1].w = sdv[1][rsteps-1].w = sdv[2][rsteps-1].w = 10.0;

			if (x.bkhack)		/* Make curve go thought black if possible */
				sdv[0][0].w = sdv[1][0].w = sdv[2][0].w = 10.0;

			for (j = 0; j < 3; j++)
				x.rdac[j]->fit(x.rdac[j], 0, fitord, sdv[j], asgrey.no, RDAC_SMOOTH);

			/* Make sure that if we are using native brightness and white point, */
			/* that the curves go to a perfect 1.0 ... */
			if (x.nat) {
				for (j = 0; j < 3; j++)
					x.rdac[j]->force_1(x.rdac[j], 1.0);
			}

			/* Make sure that if we are using black hack black point, */
			/* that the curves go to a perfect 0.0 ... */
			if (x.bkhack) {
				for (j = 0; j < 3; j++)
					x.rdac[j]->force_0(x.rdac[j], 0.0);
			}

			for (j = 0; j < 3; j++)
				free(sdv[j]);
#ifdef DEBUG_PLOT
			/* Plot the current curves */
			{
				#define	XRES 255
				double xx[XRES];
				double y1[XRES];
				double y2[XRES];
				double y3[XRES];
				double rgb[3];
				for (i = 0; i < XRES; i++) {
					double drgb[3], rgb[3];
					xx[i] = i/(XRES-1.0);
					rgb[0] = rgb[1] = rgb[2] = xx[i];
					for (j = 0; j < 3; j++)
						drgb[j] = x.rdac[j]->interp(x.rdac[j], rgb[j]);
					y1[i] = drgb[0];
					y2[i] = drgb[1];
					y3[i] = drgb[2];
				}
				printf("Current ramdac curves\n");
				do_plot(xx,y1,y2,y3,XRES);
				#undef XRES
			}
#endif
		}
	}	/* Next refine/verify loop */

	free_alloc_csamp(&asgrey);		/* We're done with test points */
	dr->del(dr);	/* Now we're done with test window */

	/* Write out the resulting calibration file */
	if (verify != 2) {
		int calres = CAL_RES;			/* steps in calibration table saved */
		cgats *ocg;						/* output cgats structure */
		time_t clk = time(0);
		struct tm *tsp = localtime(&clk);
		char *atm = asctime(tsp);		/* Ascii time */
		cgats_set_elem *setel;			/* Array of set value elements */
		int ncps;						/* Number of curve parameters */
		double *cps[3];					/* Arrays of curve parameters */
		char *bp = NULL, buf[100];		/* Buffer to sprintf into */

		ocg = new_cgats();				/* Create a CGATS structure */
		ocg->add_other(ocg, "CAL"); 	/* our special type is Calibration file */

		ocg->add_table(ocg, tt_other, 0);	/* Add a table for RAMDAC values */
		ocg->add_kword(ocg, 0, "DESCRIPTOR", "Argyll Device Calibration Curves",NULL);
		ocg->add_kword(ocg, 0, "ORIGINATOR", "Argyll dispcal", NULL);
		atm[strlen(atm)-1] = '\000';	/* Remove \n from end */
		ocg->add_kword(ocg, 0, "CREATED",atm, NULL);

		ocg->add_kword(ocg, 0, "DEVICE_CLASS","DISPLAY", NULL);
		ocg->add_kword(ocg, 0, "COLOR_REP","RGB", NULL);
		/* Tell downstream whether they can expect that this calibration */
		/* will be applied in hardware or not. */
		ocg->add_kword(ocg, 0, "VIDEO_LUT_CALIBRATION_POSSIBLE",noramdac ? "NO" : "YES", NULL);
		/* Tell downstream whether the device range was actually (16-235)/255 */
		ocg->add_kword(ocg, 0, "TV_OUTPUT_ENCODING",out_tvenc ? "YES" : "NO", NULL);

		/* Put the target parameters in the CGATS file too */
		if (ditype != 0) {
			if ((ditype & ~0xff) != 0)
				sprintf(buf,"%c%c",((ditype >> 8) & 0xff), ditype & 0xff);
			else
				sprintf(buf,"%c",ditype);
			ocg->add_kword(ocg, 0, "DEVICE_TYPE", buf, NULL);
		}
		
		if (wpx == 0.0 && wpy == 0.0 && temp == 0.0 && tbright == 0.0)
			ocg->add_kword(ocg, 0, "NATIVE_TARGET_WHITE","", NULL);

		sprintf(buf,"%f %f %f", x.twh[0], x.twh[1], x.twh[2]);
		ocg->add_kword(ocg, 0, "TARGET_WHITE_XYZ",buf, NULL);

		switch(x.gammat) {
			case gt_power:
				if (egamma > 0.0)
					sprintf(buf,"%f", -egamma);
				else
					sprintf(buf,"%f", gamma);
				break;
			case gt_Lab:
				strcpy(buf,"L_STAR");
				break;
			case gt_sRGB:
				strcpy(buf,"sRGB");
				break;
			case gt_Rec709:
				strcpy(buf,"REC709");
				break;
			case gt_SMPTE240M:
				strcpy(buf,"SMPTE240M");
				break;
			default:
				error("Unknown gamma type");
		}
		ocg->add_kword(ocg, 0, "TARGET_GAMMA",buf, NULL);

		sprintf(buf,"%f", x.oofff);
		ocg->add_kword(ocg, 0, "DEGREE_OF_BLACK_OUTPUT_OFFSET",buf, NULL);
		
		sprintf(buf,"%f", bkcorrect);
		ocg->add_kword(ocg, 0, "BLACK_POINT_CORRECTION", buf, NULL);

		sprintf(buf,"%f", x.nbrate);
		ocg->add_kword(ocg, 0, "BLACK_NEUTRAL_BLEND_RATE", buf, NULL);

		if (bkbright > 0.0) {
			sprintf(buf,"%f", bkbright);
			ocg->add_kword(ocg, 0, "TARGET_BLACK_BRIGHTNESS",buf, NULL);
		}

		if (bkhack) {
			ocg->add_kword(ocg, 0, "BLACK_POINT_HACK","YES", NULL);
		}

		/* Write rest of setup */
	    switch (quality) {
			case -3:				/* Test value */
				bp = "ultra low";
				break;
			case -2:				/* Very low */
				bp = "very low";
				break;
			case -1:				/* Low */
				bp = "low";
				break;
			case 0:					/* Medum */
				bp = "medium";
				break;
			case 1:					/* High */
				bp = "high";
				break;
			case 2:					/* Ultra */
				bp = "ultra high";
				break;
			default:
				error("unknown quality level %d",quality);
		}
		ocg->add_kword(ocg, 0, "QUALITY",bp, NULL);

		ocg->add_field(ocg, 0, "RGB_I", r_t);
		ocg->add_field(ocg, 0, "RGB_R", r_t);
		ocg->add_field(ocg, 0, "RGB_G", r_t);
		ocg->add_field(ocg, 0, "RGB_B", r_t);

		if ((setel = (cgats_set_elem *)malloc(sizeof(cgats_set_elem) * 4)) == NULL)
			error("Malloc failed!");

		/* Write the video lut curve values */
		for (i = 0; i < calres; i++) {
			double vv, rgb[3];

#if defined(__APPLE__) && defined(__POWERPC__)
			gcc_bug_fix(i);
#endif
			vv = i/(calres-1.0);
			for (j = 0; j < 3; j++) {
				double cc;
				cc = x.rdac[j]->interp(x.rdac[j], vv);
				if (cc < 0.0)
					cc = 0.0;
				else if (cc > 1.0)
					cc = 1.0;
				rgb[j] = cc;
			}

			setel[0].d = vv;
			setel[1].d = rgb[0];
			setel[2].d = rgb[1];
			setel[3].d = rgb[2];

			ocg->add_setarr(ocg, 0, setel);
		}

		free(setel);

		/* Write some of the device model information to a second */
		/* table, so that we can update the calibration latter on without */
		/* having to read R,G & B curves. */

		ocg->add_table(ocg, tt_other, 0);	/* Add a second table for setup and model */
		ocg->add_kword(ocg, 1, "DESCRIPTOR", "Argyll Calibration options and model",NULL);
		ocg->add_kword(ocg, 1, "ORIGINATOR", "Argyll dispcal", NULL);
		atm[strlen(atm)-1] = '\000';	/* Remove \n from end */
		ocg->add_kword(ocg, 1, "CREATED",atm, NULL);


		/* Write device model curves */
		ocg->add_field(ocg, 1, "R_P", r_t);
		ocg->add_field(ocg, 1, "G_P", r_t);
		ocg->add_field(ocg, 1, "B_P", r_t);

		if ((setel = (cgats_set_elem *)malloc(sizeof(cgats_set_elem) * 3)) == NULL)
			error("Malloc failed!");

		ncps = -1;
		for (i = 0; i < 3; i++) {
			int nn;
			nn = x.dcvs[i]->get_params(x.dcvs[i], &cps[i]);
			if (ncps != -1 && ncps != nn)
				error("Expect device model linearisation curves to have the same order");
			ncps = nn;
		}

		for (i = 0; i < ncps; i++) {
			setel[0].d = cps[0][i];
			setel[1].d = cps[1][i];
			setel[2].d = cps[2][i];
			ocg->add_setarr(ocg, 1, setel);
		}

		for (i = 0; i < 3; i++)
			free(cps[i]);
		free(setel);

		if (ocg->write_name(ocg, outname))
			error("Write error : %s",ocg->e.m);

		if (verb)
			printf("Written calibration file '%s'\n",outname);

		ocg->del(ocg);		/* Clean up */

	}

	/* Update the ICC file with the new 'vcgt' curves */
	if (verify != 2 && doupdate && doprofile) {
		icmFile *ic_fp;
		icc *icco;
		icmErr err = { 0, { '\000'} };
		int j, i;
		icmVideoCardGamma *wo;

		if ((icco = new_icc(&err)) == NULL)
			error("Creation of ICC object to read profile '%s' failed (0x%x, '%s')",
			                                                iccoutname,err.c,err.m);
		/* Open up the profile for reading */
		if ((ic_fp = new_icmFileStd_name(&err, iccoutname,"r")) == NULL)
			error("Can't open file '%s' (0x%x, '%s')",iccoutname,err.c,err.m);

		/* Read header etc. */
		if ((rv = icco->read(icco,ic_fp,0)) != 0)
			error("Reading profile '%s' failed with %d, %s",iccoutname, rv,icco->e.m);

		/* Read every tag */
		if (icco->read_all_tags(icco) != 0) {
			error("Unable to read all tags from '%s': %d, %s",iccoutname, icco->e.c,icco->e.m);
		}

		ic_fp->del(ic_fp);

		wo = (icmVideoCardGamma *)icco->read_tag(icco, icSigVideoCardGammaTag);
		if (wo == NULL)
			error("Can't find VideoCardGamma tag in file '%s': %d, %s",
			      iccoutname, icco->e.c,icco->e.m);

		wo->tagType = icVideoCardGammaTable;
		wo->u.table.channels = 3;			/* rgb */
		wo->u.table.entryCount = CAL_RES;	/* full lut */
		wo->u.table.entrySize = 2;			/* 16 bits */
		wo->allocate(wo);
		for (j = 0; j < 3; j++) {
			for (i = 0; i < CAL_RES; i++) {
				double cc, vv;
#if defined(__APPLE__) && defined(__POWERPC__)
				gcc_bug_fix(i);
#endif
				vv = i/(CAL_RES-1.0);

				cc = x.rdac[j]->interp(x.rdac[j], vv);

				if (cc < 0.0)
					cc = 0.0;
				else if (cc > 1.0)
					cc = 1.0;
				if (out_tvenc) {
					cc = (cc * (235.0-16.0) + 16.0)/255.0;

					/* For video encoding the extra bits of precision are created by bit shifting */
					/* rather than scaling, so we need to scale the fp value to account for this. */
					/* We assume the precision is the vcgt table size = 16 */
					/* ~~99 ideally we should tag the fact that this is video encoded, so that */
					/* the vcgt loader can adjust for a different bit precision ~~~~ */
					cc = (cc * 255 * (1 << (16 - 8)))/((1 << 16) - 1.0); 	
				}
				wo->u.table.data[j][i] = cc;
			}
		}

		/* Open up the profile again writing */
		if ((ic_fp = new_icmFileStd_name(&err, iccoutname,"w")) == NULL)
			error("Can't open file '%s' for writing (0x%x, '%s')",iccoutname,err.c,err.m);

		if ((rv = icco->write(icco,ic_fp,0)) != 0)
			error("Write to file '%s' failed: %d, %s",iccoutname, rv,icco->e.m);

		if (verb)
			printf("Updated profile '%s'\n",iccoutname);

		ic_fp->del(ic_fp);
		icco->del(icco);

	/* Create a fast matrix/shaper profile */
	/*
	     [ Another way of doing this would be to run all the
	     measured points through the calibration curves, and
	     then re-fit the curve/matrix to the calibrated points.
	     This might be more accurate ?]

		 Ideally we should also re-measure primaries through calibration
		 rather than computing the calibrated values ? 

	 */

	} else if (verify != 2 && doprofile) {
		icmFile *wr_fp;
		icc *wr_icco;
		icmErr err = { 0, { '\000'} };
		double uwp[3];		/* Absolute Uncalibrated White point in XYZ */
		double wp[3];		/* Absolute White point in XYZ */
		double bp[3];		/* Absolute Black point in XYZ */
		double mat[3][3];	/* Device to XYZ matrix */
		double calrgb[3];	/* 1.0 through calibration curves */
		double clrgb[3];	/* 1.0 through calibration and linearization */

		/* Open up the file for writing */
		if ((wr_fp = new_icmFileStd_name(&err, iccoutname,"w")) == NULL)
			error("Write: Can't open file '%s' (0x%x, '%s')",iccoutname,err.c,err.m);

		if ((wr_icco = new_icc(&err)) == NULL)
			error("Write: Creation of ICC object failed (0x%x, '%s')",err.c,err.m);

		if (wr_icco->set_version(wr_icco, iccver) != 0)
			error("set_version %d failed: %d, %s",iccver,wr_icco->e.c,wr_icco->e.m);

		/* Set the header: */
		{
			icmHeader *wh = wr_icco->header;

			/* Values that must be set before writing */
			wh->deviceClass     = icSigDisplayClass;
			wh->colorSpace      = icSigRgbData;				/* Display is RGB */
			wh->pcs         = icSigXYZData;					/* XYZ for matrix based profile */
			wh->renderingIntent = icRelativeColorimetric;	/* For want of something */

			wh->manufacturer = icmSigUnknownType;
	    	wh->model        = icmSigUnknownType;
#ifdef NT
			wh->platform = icSigMicrosoft;
#endif
#ifdef UNIX_APPLE
			wh->platform = icSigMacintosh;
#endif
#if defined(UNIX_X11)
			wh->platform = icmSig_nix;
#endif
		}

		/* Lookup white and black points */
		{
			int j;
			double rgb[3];

			calrgb[0] = calrgb[1] = calrgb[2] = 1.0;

			fwddev(&x, uwp, calrgb);		/* absolute uncalibrated WP (native white point) */

//printf("~1 native abs white point XYZ %f %f %f\n", uwp[0], uwp[1], uwp[2]);

			/* RGB 1.0 Through calibration */
			for (j = 0; j < 3; j++) {
				calrgb[j] = x.rdac[j]->interp(x.rdac[j], calrgb[j]);
				if (calrgb[j] < 0.0)
					calrgb[j] = 0.0;
				else if (calrgb[j] > 1.0)
					calrgb[j] = 1.0;
			}
			fwddev(&x, wp, calrgb);		/* absolute calibrated WP */
//printf("~1 calibrated rgb = %f %f %f\n", calrgb[0], calrgb[1], calrgb[2]);
//printf("~1 calibrated abs white point XYZ %f %f %f\n", wp[0], wp[1], wp[2]);

			for (j = 0; j < 3; j++)
				clrgb[j] = x.dcvs[j]->interp(x.dcvs[j], calrgb[j]);
//printf("~1 cal & lin rgb = %f %f %f\n", clrgb[0], clrgb[1], clrgb[2]);

			rgb[0] = rgb[1] = rgb[2] = 0.0;

			/* RGB 0.0 through calibration */
			for (j = 0; j < 3; j++) {
				rgb[j] = x.rdac[j]->interp(x.rdac[j], rgb[j]);
				if (rgb[j] < 0.0)
					rgb[j] = 0.0;
				else if (rgb[j] > 1.0)
					rgb[j] = 1.0;
			}
			fwddev(&x, bp, rgb);		/* Absolute calibrated BP */
		}

		/* Apply calibration to matrix, and then adjust it to be */
		/* relative to D50 white point, rather than absolute. */
		{
			double rgb[3];
			icmXYZNumber swp;

			/* Transfer from parameter to matrix */
			icmCpy3x3(mat, x.fm);

			/* Compute the calibrated matrix values so that the curves */
			/* device curves end at 1.0. */

			/* In the HW calibrated case this represents the lower XYZ due to */
			/* the HW calibrated lower RGB values of white compared to the raw */
			/* model response, so that the calibration curve concatentation with the */
			/* device curves can be scaled up to end at 1.0. */
			if (noramdac == 0) {
				for (j = 0; j < 3; j++) {
					for (i = 0; i < 3; i++)
						rgb[i] = 0.0;
					rgb[j] = clrgb[j];
					icmMulBy3x3(rgb, x.fm, rgb);	/* clrgb -> matrix -> RGB */
					for (i = 0; i < 3; i++)
						mat[i][j] = rgb[i];
				}
#ifdef NEVER
				{
					double rgb[3], xyz[3], lab[3];
				
					rgb[0] = rgb[1] = rgb[2] = 1.0;
					icmMulBy3x3(xyz, mat, rgb);	
					icmXYZ2Lab(&icmD50, lab, xyz);
				
					printf("RGB 1 through matrix = XYZ %f %f %f, Lab %f %f %f\n", xyz[0], xyz[1], xyz[2], lab[0], lab[1], lab[2]);
				}
#endif
				/* Chromatic Adaptation matrix */
				icmAry2XYZ(swp, wp);
				wr_icco->chromAdaptMatrix(wr_icco, ICM_CAM_MULMATRIX, NULL, mat, icmD50, swp);
#ifdef NEVER
				{
					double rgb[3], xyz[3], lab[3];
				
					rgb[0] = rgb[1] = rgb[2] = 1.0;
					icmMulBy3x3(xyz, mat, rgb);	
					icmXYZ2Lab(&icmD50, lab, xyz);
				
					printf("RGB 1 through chrom matrix = XYZ %f %f %f, Lab %f %f %f\n", xyz[0], xyz[1], xyz[2], lab[0], lab[1], lab[2]);
				}
#endif

			/* For the calibration incororated in profile case, we should boost the */
			/* XYZ by 1/calrgb[] so that the lower calibrated RGB values results in the native */
			/* white point, but we want to reduce it by callinrgb[] to move from the native */
			/* white point to the calibrated white point. */
			} else {
				icmCpy3x3(mat, x.fm);

				for (j = 0; j < 3; j++) {
					for (i = 0; i < 3; i++)
						rgb[i] = 0.0;
					rgb[j] = clrgb[j]/calrgb[j];
					icmMulBy3x3(rgb, x.fm, rgb);	/* 1/calrgb -> matrix -> RGB */
					for (i = 0; i < 3; i++)
						mat[i][j] = rgb[i];
				}
#ifdef NEVER
				{
					double rgb[3], xyz[3], lab[3];
				
					for (j = 0; j < 3; j++)
						rgb[j] = calrgb[j];
					icmMulBy3x3(xyz, mat, rgb);	
					icmXYZ2Lab(&icmD50, lab, xyz);
				
					printf("RGB cal through matrix = XYZ %f %f %f, Lab %f %f %f\n", xyz[0], xyz[1], xyz[2], lab[0], lab[1], lab[2]);
				}
#endif
				/* Chromatic Adaptation matrix */
				icmAry2XYZ(swp, wp);
				wr_icco->chromAdaptMatrix(wr_icco, ICM_CAM_MULMATRIX, NULL, mat, icmD50, swp);
#ifdef NEVER
				{
					double rgb[3], xyz[3], lab[3];
				
					icmMulBy3x3(xyz, mat, calrgb);	
					icmXYZ2Lab(&icmD50, lab, xyz);
				
					printf("RGB cal through chrom matrix = XYZ %f %f %f, Lab %f %f %f\n", xyz[0], xyz[1], xyz[2], lab[0], lab[1], lab[2]);
				}
#endif
			}
		}
		
		/* Add all the other tags required */

		/* Profile Description Tag: */
		{
			icmCommonTextDescription *wo;
			char *dst, dstm[200];			/* description */

			if (profDesc != NULL)
				dst = profDesc;
			else {
				dst = iccoutname;
			}

			if ((wo = (icmCommonTextDescription *)wr_icco->add_tag(
			           wr_icco, icSigProfileDescriptionTag,	icmSigCommonTextDescriptionType)) == NULL) 
				error("add_tag failed: %d, %s",wr_icco->e.c,wr_icco->e.m);

			wo->count = strlen(dst)+1; 	/* Allocated and used size of desc, inc null */
			wo->allocate(wo);/* Allocate space */
			strcpy(wo->desc, dst);		/* Copy the string in */
		}
		/* Copyright Tag: */
		{
			icmCommonTextDescription *wo;
			char *crt;

			if (copyright != NULL)
				crt = copyright;
			else
				crt = "Copyright, the creator of this profile";

			if ((wo = (icmCommonTextDescription *)wr_icco->add_tag(
			           wr_icco, icSigCopyrightTag,	icmSigCommonTextDescriptionType)) == NULL) 
				error("add_tag failed: %d, %s",wr_icco->e.c,wr_icco->e.m);

			wo->count = strlen(crt)+1; 	/* Allocated and used size of text, inc null */
			wo->allocate(wo);/* Allocate space */
			strcpy(wo->desc, crt);		/* Copy the text in */
		}
		/* Device Manufacturers Description Tag: */
		if (deviceMfgDesc != NULL) {
			icmCommonTextDescription *wo;
			char *dst = deviceMfgDesc;

			if ((wo = (icmCommonTextDescription *)wr_icco->add_tag(
			           wr_icco, icSigDeviceMfgDescTag,	icmSigCommonTextDescriptionType)) == NULL) 
				error("add_tag failed: %d, %s",wr_icco->e.c,wr_icco->e.m);

			wo->count = strlen(dst)+1; 	/* Allocated and used size of desc, inc null */
			wo->allocate(wo);/* Allocate space */
			strcpy(wo->desc, dst);		/* Copy the string in */
		}
		/* Model Description Tag: */
		if (modelDesc != NULL) {
			icmCommonTextDescription *wo;
			char *dst = modelDesc;

			if ((wo = (icmCommonTextDescription *)wr_icco->add_tag(
			           wr_icco, icSigDeviceModelDescTag,	icmSigCommonTextDescriptionType)) == NULL) 
				error("add_tag failed: %d, %s",wr_icco->e.c,wr_icco->e.m);

			wo->count = strlen(dst)+1; 	/* Allocated and used size of desc, inc null */
			wo->allocate(wo);/* Allocate space */
			strcpy(wo->desc, dst);		/* Copy the string in */
		}
		/* Luminance tag */
		{
			icmXYZArray *wo;;
	
			if ((wo = (icmXYZArray *)wr_icco->add_tag(
			           wr_icco, icSigLuminanceTag, icSigXYZArrayType)) == NULL) 
				error("add_tag failed: %d, %s",wr_icco->e.c,wr_icco->e.m);

			/* (Only Y is used according to the ICC spec.) */
			wo->count = 1;
			wo->allocate(wo);	/* Allocate space */
			wo->data[0].X = 0.0;
			wo->data[0].Y = dispLum * wp[1]/uwp[1];	/* Adjust for effect of calibration */
			wo->data[0].Z = 0.0;

			if (verb)
				printf("Luminance XYZ = %f %f %f\n", wo->data[0].X, wo->data[0].Y, wo->data[0].Z);
		}
		/* White Point Tag: */
		{
			icmXYZArray *wo;
			/* Note that tag types icSigXYZType and icSigXYZArrayType are identical */
			if ((wo = (icmXYZArray *)wr_icco->add_tag(
			           wr_icco, icSigMediaWhitePointTag, icSigXYZArrayType)) == NULL) 
				error("add_tag failed: %d, %s",wr_icco->e.c,wr_icco->e.m);

			wo->count = 1;
			wo->allocate(wo);	/* Allocate space */
			wo->data[0].X = wp[0] * 1.0/wp[1];
			wo->data[0].Y = wp[1] * 1.0/wp[1];
			wo->data[0].Z = wp[2] * 1.0/wp[1];

			if (verb)
				printf("White point XYZ = %f %f %f\n", wo->data[0].X, wo->data[0].Y, wo->data[0].Z);
		}
		/* Black Point Tag: */
		{
			icmXYZArray *wo;
			if ((wo = (icmXYZArray *)wr_icco->add_tag(
			           wr_icco, icSigMediaBlackPointTag, icSigXYZArrayType)) == NULL) 
				error("add_tag failed: %d, %s",wr_icco->e.c,wr_icco->e.m);

			wo->count = 1;
			wo->allocate(wo);	/* Allocate space */
			wo->data[0].X = bp[0] * 1.0/wp[1];
			wo->data[0].Y = bp[1] * 1.0/wp[1];
			wo->data[0].Z = bp[2] * 1.0/wp[1];

			if (verb)
				printf("Black point XYZ = %f %f %f\n", wo->data[0].X, wo->data[0].Y, wo->data[0].Z);
		}

		/* vcgt tag, if the display has an accessible VideoLUT */
		if (noramdac == 0) {
			int j, i;
			icmVideoCardGamma *wo;
			wo = (icmVideoCardGamma *)wr_icco->add_tag(wr_icco,
			                           icSigVideoCardGammaTag, icSigVideoCardGammaType);
			if (wo == NULL)
				error("add_tag failed: %d, %s",wr_icco->e.c,wr_icco->e.m);

			wo->tagType = icVideoCardGammaTable;
			wo->u.table.channels = 3;				/* rgb */
			wo->u.table.entryCount = CAL_RES;		/* full lut */
			wo->u.table.entrySize = 2;				/* 16 bits */
			wo->allocate(wo);
			for (j = 0; j < 3; j++) {
				for (i = 0; i < CAL_RES; i++) {
					double cc, vv = i/(CAL_RES-1.0);
					cc = x.rdac[j]->interp(x.rdac[j], vv);
					if (cc < 0.0)
						cc = 0.0;
					else if (cc > 1.0)
						cc = 1.0;
					if (out_tvenc) {
						cc = (cc * (235.0-16.0) + 16.0)/255.0;
						/* For video encoding the extra bits of precision are created by bit */
						/* shifting rather than scaling, so we need to scale the fp value to */
						/* account for this. We assume the precision is the vcgt table size = 16 */
						/* ~~99 ideally we should tag the fact that this is video encoded, so */
						/* that the vcgt loader can adjust for a different bit precision ~~~~ */
						cc = (cc * 255 * (1 << (16 - 8)))/((1 << 16) - 1.0); 	
					}
					wo->u.table.data[j][i] = cc;
				}
			}
		}

		/* Red, Green and Blue Colorant Tags: */
		{
			icmXYZArray *wor, *wog, *wob;
			if ((wor = (icmXYZArray *)wr_icco->add_tag(
			           wr_icco, icSigRedColorantTag, icSigXYZArrayType)) == NULL) 
				error("add_tag failed: %d, %s",rv,wr_icco->e.m);
			if ((wog = (icmXYZArray *)wr_icco->add_tag(
			           wr_icco, icSigGreenColorantTag, icSigXYZArrayType)) == NULL) 
				error("add_tag failed: %d, %s",rv,wr_icco->e.m);
			if ((wob = (icmXYZArray *)wr_icco->add_tag(
			           wr_icco, icSigBlueColorantTag, icSigXYZArrayType)) == NULL) 
				error("add_tag failed: %d, %s",rv,wr_icco->e.m);

			wor->count = wog->count = wob->count = 1;
			wor->allocate(wor);	/* Allocate space */
			wog->allocate(wog);
			wob->allocate(wob);

			/* Make sure rounding doesn't wreck white point */
			icmTranspose3x3(mat, mat);	/* Convert [XYZ][RGB] to [RGB][XYZ] */
			quantizeRGBprimsS15Fixed16(mat);

			wor->data[0].X = mat[0][0]; wor->data[0].Y = mat[0][1]; wor->data[0].Z = mat[0][2];
			wog->data[0].X = mat[1][0]; wog->data[0].Y = mat[1][1]; wog->data[0].Z = mat[1][2];
			wob->data[0].X = mat[2][0]; wob->data[0].Y = mat[2][1]; wob->data[0].Z = mat[2][2];
		}

		/* Red, Green and Blue Tone Reproduction Curve Tags: */
		{
			icmCurve *wor, *wog, *wob;
			int ui;

			if ((wor = (icmCurve *)wr_icco->add_tag(
			           wr_icco, icSigRedTRCTag, icSigCurveType)) == NULL) 
				error("add_tag failed: %d, %s",rv,wr_icco->e.m);
			if ((wog = (icmCurve *)wr_icco->add_tag(
			           wr_icco, icSigGreenTRCTag, icSigCurveType)) == NULL) 
				error("add_tag failed: %d, %s",rv,wr_icco->e.m);
			if ((wob = (icmCurve *)wr_icco->add_tag(
			           wr_icco, icSigBlueTRCTag, icSigCurveType)) == NULL) 
				error("add_tag failed: %d, %s",rv,wr_icco->e.m);

			wor->ctype = wog->ctype = wob->ctype = icmCurveSpec; 
			wor->count = wog->count = wob->count = 256;			/* Number of entries */
			wor->allocate(wor);	/* Allocate space */
			wog->allocate(wog);
			wob->allocate(wob);
	
			/* For the HW calibrated case, we have lowered the matrix */
			/* values to reflect the calibrated RGB through the native */
			/* device model, so now we can scale up the comcatenation */
			/* of the calibration and linearisation curves so that */
			/* 1.0 in maps to 1.0 out. */
			if (noramdac == 0) {

				for (ui = 0; ui < wor->count; ui++) {
					double in, rgb[3];
		
					for (j = 0; j < 3; j++) {
#if defined(__APPLE__) && defined(__POWERPC__)
						gcc_bug_fix(ui);
#endif
						in = (double)ui / (wor->count - 1.0);
			
						/* Transform through calibration curve */
						in = x.rdac[j]->interp(x.rdac[j], in);

						if (in < 0.0)
							in = 0.0;
						else if (in > 1.0)
							in = 1.0;

						/* Trandform though device model linearisation */
						in = x.dcvs[j]->interp(x.dcvs[j], in);

						/* Scale back so that 1.0 in gets 1.0 out */
						in /= clrgb[j];


						if (in < 0.0)
							in = 0.0;
						else if (in > 1.0)
							in = 1.0;
						rgb[j] = in;
//printf("Step %d, Chan %d, %f -> %f\n",ui,j,(double)ui / (wor->count - 1.0),in);
					}
					wor->data[ui] = rgb[0];	/* Curve values 0.0 - 1.0 */
					wog->data[ui] = rgb[1];
					wob->data[ui] = rgb[2];
				}

			/* For the calibration incororated in profile case, */
			/* we bypass the inverse calibration curve if it would */
			/* result in saturation, and then scale the overall output */
			/* back by the calrgb[] value so that the overall curve */
			/* maps 1.0 to 1.0. The scaled up values in the matrix */
			/* then result in a calibrated RGB input mapping to the */
			/* PCS white point. */
			} else {

				for (ui = 0; ui < wor->count; ui++) {
					double in, rgb[3];
		
					for (j = 0; j < 3; j++) {
#if defined(__APPLE__) && defined(__POWERPC__)
						gcc_bug_fix(ui);
#endif
						in = (double)ui / (wor->count - 1.0);
			
						/* If within the inversion range, */
						/* transform through the inverse calibration curve. */
						if (in < calrgb[j]) {
							in = x.rdac[j]->inv_interp(x.rdac[j], in);
							if (in < 0.0)
								in = 0.0;
							else if (in > 1.0)
								in = 1.0;
							/* Pass through device model linearisation. */
							in = x.dcvs[j]->interp(x.dcvs[j], in);

						/* Linearly extrapolate when outside inv range */
						} else {
							in /= calrgb[j];
						}

						/* Scale it back again to 0.0 to 1.0, */
						/* which is compensated for by matrix scale. */
						in *= calrgb[j];

						if (in < 0.0)
							in = 0.0;
						else if (in > 1.0)
							in = 1.0;
						rgb[j] = in;
//printf("Step %d, Chan %d, %f -> %f\n",ui,j,(double)ui / (wor->count - 1.0),in);
					}
					wor->data[ui] = rgb[0];	/* Curve values 0.0 - 1.0 */
					wog->data[ui] = rgb[1];
					wob->data[ui] = rgb[2];
				}
			}
		}

		/* Write the file (including all tags) out */
		if ((rv = wr_icco->write(wr_icco,wr_fp,0)) != 0) {
			error("Write file: %d, %s",rv,wr_icco->e.m);
		}

		if (verb)
			printf("Created fast shaper/matrix profile '%s'\n",iccoutname);

		/* Close the file */
		wr_icco->del(wr_icco);
		wr_fp->del(wr_fp);
	}

	if (verify != 2) {
		for (j = 0; j < 3; j++)
			x.rdac[j]->del(x.rdac[j]);
	
		for (k = 0; k < 3; k++)
			x.dcvs[k]->del(x.dcvs[k]);
	}

	if (x.svc != NULL) {
		x.svc->del(x.svc);
		x.svc = NULL;
	}
	if (x.dvc != NULL) {
		x.dvc->del(x.svc);
		x.dvc = NULL;
	}

	free_a_disppath(disp);
	free_ccids(ccids);

	return 0;
}




